title: Spectrophotometer UV/VIS with Integrated Water Analyses Programmed Methods:

description: Spectrophotometer for measurement in the visible and the UV fields with a wavelengths range of minimum 190 up to 1 100 mmThe accuracy of the wavelength should be a minimum of ±1nm with repetition of 0.1 nm and resolution of a minimum 0,1 nm (scan 0,1 nm) or better.The instrument’s resolution should be 0,001 Abs or better, with accuracy of absorbance < 0,5 %.The instrument should allow measurement of concertation, absorption, transmission, scanning of spectrums and kinetics in the absorption and transmission mode.The instrument should be programmed for a minimum of 180 previously programmed work methods and to perform automatic recognition of the tests through a barcode. It should have the possibility to update/scale up the programmed methods and the software.Additionally, the instrument should memorize methods that have been defined and programmed by the user.The apparatus should provide automatic reading of the analyses’ results.Zero adjustment with appropriate zero cell.There should be a possibility for a possible correction of the turbidity when reading the concentration.The instrument should memorize the measured data and allow for data transfer on UBS or PC.The instrument to be able to work with rectangular cuvettes of 10, 20, 50 mm and 100 mm, as well as round cuvettes with a 16 mm diameter, and allow for automatic recognition.The cuvettes holder should be removable for easier cleaning.When the instrument is turned on, it must run self-check tests of the instrument, including the memory, the lamp, the filters, the wavelength calibration.The instrument’s main menu should offer a possibility to use the methods programmed in advance, the methods defined and programmed by the user, direct/quick access to absorption/transmission measurement.The instrument should have a minimum of three USB ports for transfer of data and minimum one ethernet port work conditions of the instrument: temperature +10 up to 35 C, relative humidity 20 to 80 % optics and electronics protection class IP 31 or better.The instrument should be in compliance with EN 61326 and EN 61010.The instrument should be delivered with 2 quartz cuvettes of 100 mm.The instrument should be delivered with a valid calibration certificate in line with ISO 17025.The bidder should provide a minimum 1-year guarantee and a post-guarantee period of minimum 2 years, as well as supply of original spare parts.Consumable materials must be delivered accompanying the spectrophotometer:— at least 100 tests for phenols determination with a low limit of determination of 0,002 mg/L applicable on the offered instrument.

title: System for Determination of BOC Applying the Pressure Measurement Method:

description: • The apparatus for determination of the biological oxygen consumption (BOC) based on manometric pressure measurement (without mercury), electronic pressure sensor• The kit should contain:— at least 6 measurement stations;— inductive mixer with at least 6 magnets;— containers for the measurement stations (at least 6);— pouring flasks for sample measurement (for a range of 0-40 mg/l BOC, for a range of 0-400 mg/l BOC and for a range of 0-800 mg/l BOC.• Technical data— measurement range from 0-40 mg/l to 0-400 mg/l oxygen;— possibility to select the measurement period from 1 to 28 days;— to have an appropriate software installed that will directly read the value of the BOC and the measurement time;— to perform memorization of the values;— dimensions (length x width x height) maximum 400 mm x 400 mm x 300 mm, with a mixing unit with the mixer and the containers;— a clock: real time clock, works on batteries if the device is not charged with external power supply;— protection class: IP 54 (head sensor);— interface: USB host port (USB stick), USB port (computer), SD card;— power supply: through batteries or through a power supply device with the assistance of an S cable with the mixing unit;— automatic start function: after levelling the samples’ temperature, it can be turned off and be controlled by the pressure fall in the bottle;— storage interval: after 1 hour (1 day), at every 2 hours (2 days), daily (3-28 days);— automatic results storage: at least 500 results, depending on the measurement period and the amount of sample bottles;— large graphic luminescent screen: 128 x 40 pixels or equivalent;— environment conditions: 2 – 40 C, relative humidity 5-80 %, without condensation;— the device should be in compliance with DIN EN 61326 or equivalent.• The kit should contain:— nitrification inhibitor – solution x3;— absorbing solution x3;— BOC test of the controlling material x3.The bidder should provide a minimum 1-year guarantee and a post-guarantee period of a minimum 2 years, as well as supply of original spare parts.

title: Laboratory pH Meter:

description: — work measurement range of the pH from 0 to 14 pH and resolutions of 0.001/0.01/0.1 pH units. PH Measurement accuracy ≤0.02 pH;— to have the possibility for automatic and manual calibration in 1, 2 or 3 points, with the possibility the operator to manually set the buffer calibration value (i.e. to be able to use buffers whose values are set by the manufacturer);— automatic buffer recognition;— to have a luminescent back screen;— to be able to memorize the results;— the instrument should have the possibility to connect to a printer;— the instrument should be equipped with a glass pH electrode intended for the measurement of pH in liquids;— the electrode should measure in a minimal range from 0 to 14 pH, with an accuracy of ±0.01 pH or better;— the electrode should have a glass body and a liquid electrolyte with the possibility to refill;— the instrument should be equipped with a temperature sensor with a minimal range of 0 to 100 C, the resolution of 0.1 C or better and accuracy of ±0.5 % or better;— the device should be completed with a pH electrode – amount 2, a stand and an electrode holder and a temperature sensor, ATC probe, cable and instructions;— to have a performed calibration certificate in compliance with the ISO 17025 standard;— the bidder should provide a minimum 1-year guarantee and a post-guarantee period of at least 2 years, as well as supply of original spare parts.

title: Laboratory Conductometer:

description: — a work range for measurement of electroconductibility of a minimum 0,01 μS/cm-200,0 mS/cm with a minimal resolution of 0,01 μS/cm- 0,1 μS/cm adjustable to the measurement range, and accuracy of ± 0,5 % or better;— to have the possibility for automatic and manual calibration, with the possibility the operator to manually set the value of the calibration standards (i.e. to be able to use standards whose values are not set by the manufacturer);— the possibility to manually enter the constant’s value;— automatic buffer recognition;— to have a luminescent back screen;— to be able to memorize the results;— the instrument should have the option to connect to a printer;— the instrument should perform automatic temperature compensation with a temperature sensor that has a work range of a minimum – 10 C to + 105 C, resolution of 0.1 and accuracy of a minimum ± 0.5 %. The instrument should express the result at 20 C or 25 C (the analyst can select the result expression temperature by himself);— the instrument should be equipped with a Pt electrode for conductivity measurement, which has a glass protection (glass body conductivity cell) and whose minimal measurement range is from 0,0 μS/cm to 200 mS/cm with accuracy of ±0,5 % or better;— the device should be completed with a conductivity probe – amount 2, a stand and an electrode holder; cable and instructions; appropriate printer and paper to print the results;— to have a performed calibration certificate in compliance with the ISO 17025 standard;— the bidder should provide a one-year minimum guarantee and a post-guarantee period of at least 2 years, as well as supply of original spare parts.

title: Thermohydrometer:

description: — the thermohydrometer must be able to be set on a table under an appropriate angle and to be able to be hanged on the wall;— the box of the thermohydrometer should be manufactured by solid plastic and be water resistant in order to provide a long lifespan;— the thermohydrometer should not be larger than 110 x 70 x 20 mm;— it must have a large, easily readable LCD display with minimal dimensions of 40 x 30 mm, in order to simultaneously show the measured values for humidity and temperature;— the thermohydrometer must memorize the maximal and the minimal read temperature with the possibility to alarm;— the measurement range of the thermohydrometer for humidity should be 10 % RH to 99 % RH with a minimal reading accuracy of ±1 % RH;— the measurement range of the thermohydrometer for temperature should be a minimum – 50 C to + 70 C, with a reading accuracy of ± 0,1 C;— the temperature values of the thermohydrometer should be followable in F or C, depending on the choice of the user with easy switch through the control in the front panel;— it should have a performed calibration certificate in compliance with the ISO 17025 standard.The bidder should provide a one-year minimum guarantee and a post-guarantee period of at least 2 years, as well as supply of original spare parts

title: Thermometer with Two Probes to Track the Temperature in the Refrigerator and the Freezer:

description: — the thermometer must have two separate temperature probes;— each of the temperature probes must be sank in a solution bottle that will help in the temperature compensation when opening the door of the refrigerator, in order to avoid fluctuation during measurement;— the bottles where the probes are to be sank should be resistant to chemicals and water;— the thermometer must have a three-part display that will show the momentary temperature and the minimal and maximal temperatures measured by each of the probes;— to have the possibility to set alarms for the upper and lower temperature limits;— the temperature range of the thermometer should be a minimum of -50 C to maximum + 70 C;— the temperature accuracy of the thermometer should be a minimum of ± 0.3 C;— the thermometer’s resolution should be a minimum of 0,01 C;— the cable of each of the probes should be at least three meters long;— the thermometer should come with a conductivity certificate to NIST;— it must have holder for placing it on surfaces and an additional part that would allow it to be hanged on the wall;— two appropriate spare batteries must be delivered with the thermometer;— it should have a performed calibration certificate in compliance with the ISO 17025 standard.The bidder should provide a 1-year minimum guarantee and a post-guarantee period of at least 2 years, as well as supply of original spare parts.

title: Muffled Embering Oven:

description: — the oven should reach a maximal temperature of 1 100 C;— the heating should take place by two lateral ceramic plates with an integrated heater, protected from evaporations and bursts;— the oven should have internal dimension not smaller than 28 cm x 34 cm x 25 cm;— the door of the oven should open downwards on a flap principle in order to be used as a work surface additionally;— the exhaust vent of the oven should be on the back side and it should have a chimney (exhaust vent) installed in order to collect and direct the exhaust gases;— air circulation – natural ventilation;— the oven should have an LCD screen controller that will enable the memorization of a minimum 10 programs with a minimum of 10 segments each, i.e. programs with a minimum of 4 heating ramps and a minimum of 4 levels of temperature maintenance (hold times);— the controller should have a timer, alarm and a real time clock;— the controller should provide pre-heating protection (internal over-temperature protection);— measurement accuracy of a maximum +/- 1 C;— the oven should be in compliance with EN 61010;— the oven should be delivered with a valid performed calibration certificate in compliance with ISO 17025;— the bidder should provide a one-year minimum guarantee and a post-guarantee period of at least 2 years, as well as supply of original spare parts.

title: Water Bath:

description: — a water bath with a minimum of eight positions with easily adjustable openings where different dimensions glass items can be placed;— the bath’s openings should be covered with ring covers made from thermoresistant material;— the temperature range of 10 C minimum up to 95 C and a boiling point;— the temperature sensor should have a timer;— the bath’s heating element should be equipped with a protection from over-heating;— the bath should be made from a non-rusting steel;— the exterior dimensions of the bath: minimum 800 x 500 x 280 mm;— the bath’s water volume: 45 l minimum.The bidder should provide a 1-year minimum guarantee and a post-guarantee period of at least 2 years, as well as supply of original spare parts.

title: Manual Colorimeter for free Chloride Measurement:

description: — range: 0,00 go a minimum 2,50 ppm (mg/L);— resolution: 0,01 ppm (mg/L) minimum;— accuracy: 3 % of the reading ± 0,03 ppm (mg/L) @ 25 C;— light source: LED @25 nm;— light detector: silicon photocell;— ambiental conditions: 0 to 50 C (32 to 122 F); RH max 95 % non-condensing;— method: EPA approved DPD method Standard method 4500 – CI G;— the device should be delivered with two cuvettes, a cuvettes cover, reagents for at least 25 tests and a battery.

title: System Carrier for Counting of Coliforms, Legionella, E. Coli, Entherococci, Pseudomonas and Heterotrophs in Shallow and Surface Waters with MPH Method with Detection and Enumeration

description: — a minimum level of detection of 1 microorganism on 100 ml;— no need of a glass consumable material for analysis;— results in 24 hours;— no need of confirmation;— no need of dilution;— no need of media;— to have the possibility for enumeration from 1 to 2, 419 without dilution;— ISO 9308 – 2:2012.

title: Fully Automatized System for Identification of Microorganisms and Testing of Antimicrobe Sensitivity of Bacteria and Yeasts:

description: Asic Features:— expert software for determination of regular and irregular resistances;— work reagents should be in the form of a hermetically closed card in which all the necessary work reagents are incorporated (the system should not need additional reagents);— a capacity of a minimum of 30 analyses/tests with the possibility of continuous adding and removing of samples;— the device should contain all the necessary parts for the performance of the analysis and with the following minimum:• independent computer unit;• Barcode reader;• Reagents’ carrier;• Pipets for suspension transfer;• An apparatus to determine the mcfarland.The device’s work principle:— identification and testing of bacteria and yeasts’ sensitivity;— colorimetric work technique for bacteria or the equivalent;— turbidimetric work technique for yeasts or the equivalent;— automatic inoculation of bacteria inoculate/suspension in the test analysis that will be performed inside the device itself in order to avoid possible contamination;— the device should identify:• gram negative bacteria of 2 hours;• gram positive bacteria of 2 hours;• bacillus species of 14 hours;• anaerobes and Corynebacterium of 6 hours;• yeasts of 18 hours;• Neisseria, Haemophilus;— the device should perform sensitivity test of:• at least 5 different tests with the selection of 18 antibiotics minimum for gram negative bacteria;• at least 5 different tests with the selection of 18 antibiotics minimum for gram positive bacteria;• special card to test Streptococcus Pneumoniae;• yeasts;— the device should use hermetically closed reagents in order to avoid the risk for the employees and contamination;— the device should have the ability for independent work in a vacuum chamber and reader/incubator in case of defect of the computer unit;— the device should not have liquid waste.Results’ analysis software:— the device should perform autonomous results’ interpretation through a phenotypical recognition of the results or an equivalent with MIK in accordance with EUCAST directions or equivalent;— the device should have a validation package (IQ, OQ, PQ) in order to be able to be accredited at a microbiological laboratory;• IQ – Qualitative control of the results;• OQ – Device validation;• PQ – Final results validation;— to have a database with more than 100 000 references;— to have more than 2000 described phenotypes;— to have more than 20 000 MIK concentrations;— to have more than 100 resistance mechanisms.

title: Laboratory Apparatuses and Equipment (Second Time)

description: More details in the tender documentation.

PHO Center for Public Health — Skopje

street: Nr.18, Third Macedonian Brigade str.


title: Supply of Electricity for the Ministry of Environment and Physical Planning

currency: MKD

description: Supply of electricity for the Ministry of environment and physical planning.

title: Supply of Electricity for the State Automatic Ambient Air Quality Monitoring System

description: Electricity supply for all the automated air quality monitoring stations in the Republic of North Macedonia.

title: Supply of Electricity for the Gjavato Hydrosystem in the Dojran Lake

description: Supply of electricity for the Gjavato Hydrosystem in the Dojran Lake.

title: Supply of Electricity for the Central Environmental Laboratory

description: Supply of electricity for the Central Environmental Laboratory.

title: Supply of Electricity for the Wastewater Treatment Plant in Glumovo village, Municipality of Saraj

description: Supply of electricity for the Wastewater Treatment Plant in Glumovo village, Municipality of Saraj.

scheme: CPV

title: Procurement of Electricity

documentType: tenderNotice

description: P1: Supply of electricity for the Ministry of environment and physical planning;P2: Supply of electricity for the state automatic ambient Air Quality Monitoring System;P3: Supply of electricity for the Gjavato Hydrosystem in the Dojran Lake;P4: Supply of electricity for the Central Enivronmental Laboratory;P5: Supply of electricity for the Wastewater treatment plant in Glumovo village, Municipality of Saraj.

awardCriteria: priceOnly

procurementMethod: open

awardCriteriaDetails: PRICE: 100

mainProcurementCategory: goods

street: Square Presveta Bogorodica n.3, Skopje

countryName: MK

initiationType: tender


documentType: awardNotice

title: Telecommunication Services for Connection to the Retail Outlets

description: Telecommunication services for connection to the retail outlets.

procurementMethod: selective

mainProcurementCategory: services

National VideoLottery of Macedonia

street: bul. Kiro Gligorov 2, kat 2

name: Mr. Kostadin Karakabakov, General manager


name: Ministry of Interior of the Republic of Macedonia

title: Service, Maintenance, Calibration, Validation/Verification of Instruments by the Applied Biosystems Manufacturer

description: Service, maintenance, calibration, validation/verification of instruments by the Applied Biosystems manufacturer.

title: Service, Maintenance, Calibration, Validation/Verification of Instruments by the Manufacturer Qiagen

description: Service, maintenance, calibration, validation/verification of instruments by the manufacturer Qiagen.

title: Service, Maintenance/Verification of Instruments by the Manufacturer Sartroius

description: Service, maintenance/verification of instruments by the manufacturer Sartroius.

title: Service, Maintenance, Validation/Verification of Instruments from the Manufacturer Biorad

description: Service, maintenance, validation/verification of instruments from the manufacturer Biorad.

title: Service, Maintenance, Calibration, Instrument Validation/Verification: GC-FID 2010 PLUS-Shimadzu with AOC 20i and AOC-20s Automatic Sample Input Device

description: Service, maintenance, calibration, instrument validation/verification: GC-FID 2010 PLUS-Shimadzu with AOC 20i and AOC-20s automatic sample input device.

title: Service and Maintenance of Comparative Microscopes and Stereo Microscope from the Manufacturer Leica

description: Service and maintenance of comparative microscopes and stereo microscope from the manufacturer Leica.

title: Service and Maintenance of the Stereo Microscope from the Manufacturer Wild Heerbrug

description: Service and maintenance of the stereo microscope from the manufacturer Wild Heerbrug.

title: Digital Equipment Service from Sony, Canon and Nikon

description: Digital equipment service from Sony, Canon and Nikon.

title: Service, Maintenance, Calibration, Validation/Verification of Scanning Electron Microscope

description: Service, maintenance, calibration, validation/verification of scanning electron microscope.

title: Service, Maintenance, Calibration, Validation/Verification of Instruments by the Manufacturer Perkin Elmer

description: Service, maintenance, calibration, validation/verification of instruments by the manufacturer Perkin Elmer.

title: Service, Maintenance of Gas Installation in Laboratory and Gas Station and Procurement of Technical Gases

description: Service, maintenance of gas installation in laboratory and gas station and procurement of technical gases.

title: Service, Maintenance of Laboratory Dishwasher Steelco

description: Service, maintenance of laboratory dishwasher Steelco.

title: Service, Maintenance, Calibration, Validation/Verification of Instruments: GC/MS QP 2010 PLUS-Shimadzu

description: Service, maintenance, calibration, validation/verification of instruments: GC/MS QP 2010 PLUS-Shimadzu.

title: Service, Maintenance, Calibration, Validation/Verification of Instruments: GC/MS QP 2020 PLUS-Shimadzu

description: Service, maintenance, calibration, validation/verification of instruments: GC/MS QP 2020 PLUS-Shimadzu.

title: Service, maintenance, Calibration, Validation/Verification of Instruments by the Manufacturer Foster and Freeman

description: Service, maintenance, calibration, validation/verification of instruments by the manufacturer Foster and Freeman.

title: Service and Maintenance of Security Cabinet — Laminar

description: Service and maintenance of security cabinet — laminar.

title: Service and Maintenance of Germicidal UV Chamber with UV Lamp from the Manufacturer Germy Gima Plus

description: Service and maintenance of Germicidal UV chamber with UV lamp from the manufacturer Germy Gima Plus.

title: Service, Maintenance and Verification of Instruments from the Manufacturer Eppendorf

description: Service, maintenance and verification of instruments from the manufacturer Eppendorf.

title: Spin Service, Maintenance and Verification

description: Spin service, maintenance and verification.

title: Service, Maintenance and Verification of Centrifuges from the Manufacturer Hettich — Germany

description: Service, maintenance and verification of centrifuges from the manufacturer Hettich — Germany.

title: Service and Maintenance of Ultra Low Temperature Refrigerators

description: Service and maintenance of ultra low temperature refrigerators.

title: Service and Maintenance of Refrigerators with Temperature up to + 8 ºC

description: Service and maintenance of refrigerators with temperature up to + 8 ºC.

title: Calibration of Digital Temperature and Humidity Meters

description: Calibration of digital temperature and humidity meters.

title: Calibration on Measuring Flasks

description: Calibration on measuring flasks.

title: Electronic Automatic Scale Calibration (0.00001 g — 220 g)

description: Electronic automatic scale calibration (0.00001 g — 220 g).

title: Electronic Automatic Scale Calibration (0.2 g — 30 000 g)

description: Electronic automatic scale calibration (0.2 g — 30 000 g).

title: Calibration on Electronic Non-automatic Scales (0.01 g — 310 g)

description: Calibration on electronic non-automatic scales (0.01 g — 310 g).

title: Calibration on Electronic Non-automatic scale (0.0001 g — 210 g)

description: Calibration on electronic non-automatic scale (0.0001 g — 210 g).

title: Calibration on Electronic Non-automatic Scale (5 g - 150 000 g)

description: Calibration on electronic non-automatic scale (5 g - 150 000 g).

title: Electronic Analytical Balance Calibration

description: Electronic analytical balance calibration.

title: Calibration of Weights

description: Calibration of weights.

title: Calibration on Pipettes

description: Calibration on pipettes.

title: Calibration of Thermomixers

description: Calibration of thermomixers.

title: Maintenance of Software System eQMS:: DNA (DNA database) from the Manufacturer Pardus doo

description: Maintenance of software system eQMS:: DNA (DNA database) from the manufacturer Pardus doo.

title: Service, Maintenance, Calibration and Validation of Instruments, Measuring Equipment and Laboratory Machines for OKTIV

description: Service, maintenance, calibration and validation of instruments, measuring equipment and laboratory machines for OKTIV.

street: Dimce Mircev 9


description: Procurement of new servers and other ICT equipment for continuous operation of the AKMIS system with all upgrades

The Appellate Court Bitola

street: Bul. 1Maj num.63

Primary Court Gevgelija

street: ul.Dimitar Vlahov

Primary Court Kratovo

street: ul.Josif Daskalov

Primary Court Kriva Palanka

street: ul.Sv.Joakim Osogovski num.142

Primary Court Struga

street: ul.Sthruski Kepenci

Primary Court Berovo

street: ul.Kej na Mladina

Higher Administrative Court

street: ul.Nikola Parapunov num.31

The Appellate Court Skopje

street: Blvd Krste Misirkov num.1

Basic Civil Court Skopje

street: Blvd Goce Delchev num.4

Primary Court Kavadarci

street: ul.Nov Belgrad num.16

Primary Court Radovis

street: ul.22-ri Oktomvri

Judicial Council of RNS

street: ul.Makedonija num.5

Administrative Court

street: ul.Orce Nikolov

The Appellate Court Gostivar

street: ul.Zhivko Brajkovski num.1

The Appellate Court Stip

street: ul.General Mihajlo Apostolski num.16

Primary Court Veles

street: ul.8-mi Septemvri num.53

Primary Court Kumanovo

street: ul.11 Oktomvri

Primary Court Bitola

street: bul.1 Maj num.63

Primary Court Strumica

street: ul.Leninova num.34

name: Supreme Court of the Republic of Macedonia

street: Blvd Krste Misirkov num.8

Primary Court Prilep

street: bul.Goce Delchev

Primary Court Stip

street: ul.General Mihajlo Apostolski num.18

Primary Court Vinnitsa

street: ul.Bel Kamen

Primary Court Kocani

street: ul.Nikola Karev num.4

Primary Court Sveti Nikole

street: ul.23-ti Oktomvri num.59

Primary Court Debar

street: ul.Velko Vlahovikj

Primary Court Negotino

street: ul.Marshal Tito num.84

Primary Court Ohrid

street: bul.Makedonski prosvetiteli num.10

Primary Court Resen

street: ul.Tashe Miloshevski num.3

Primary Court Kicevo

street: bul.Osloboduvanje num.17


name: Railways of Repablik of Nort Macedonija Transport JSC Skopje

title: Dizelsko gorivo ( D-E V)

description: U tehnickoj specifikacii iz dokumentacije za nadmetanje

title: Ekstra lagano gorivo ( EL- )

description: U tehnickoj specifikacii iz dokumentacii za nadmetanje

title: Dizel gorivo i ekstra lagano gorivo

description: U tehnicke specifikacije iz dokumentacije za ndametanje

street: Bul.3-ta Makedonska brigada no.66


name: PHI Institute of Public Health of Republic of Macedonia Skopje

title: 1

description: 54 'Tests for immunoglobulin M rubella antibodies (RUBELLA IgM).

title: 2

description: 55 'Tests for immunoglobulin G Toxoplasma antibodies (TOXO IgG).

title: 3

description: 56 'Tests for immunoglobulin M Toxoplasma antibodies (TOXO IgM).

title: 4

description: 51 ' Tests for Immunoglobulin G cytomegalovirus antibodies (CMV IgG).

title: 5

description: 53 'Tests for immunoglobulin G antibodies to rubella (RUBELLA IgG).

title: 6

description: 9428 '' Tests for hepatitis B virus surface antigen (HBsAg).

title: 7

description: 9425 '' Tests for Immunoglobulin M Antibodies to Hepatitis A Virus (HAV IgM).

title: 8

description: 9431 '' Test for detection of immunoglobulin M antibodies to hepatitis B virus antigen c (ANTI HBc IgM).

title: 9

description: 9578 'Test for detection of HCV antigen.

title: 10

description: 9448 'Test for detection of IgG antibodies to hepatitis A virus (HAV AB).

title: 11

description: 9451 'Test for detection of total antibodies to HIV-1 + HIV-2 (HIV Combo).

title: 12

description: 9445 'Test for detection of total antibodies to hepatitis B virus (HBS AB).

title: 13

description: 9437 'Test for detection of total antibodies to hepatitis C virus (Anti HCV).

title: 14

description: 9901 'Tests for immunoglobulin G antibodies to cytomegalovirus (CMV IgM).

title: 15

description: 9434 'Test for detection of total antibodies against hepatitis B internal antigen (ANTI HBc).

title: 16

description: 9579 'Epstein Barr virus (EBV VCAIgG) antibody detection test applicable to CLIA apparatus or equivalent to 100 T.

title: 17

description: 9577 'Test for detection of Epstein Barr virus antibody (EBV VCAIgM) applicable to CLIA apparatus or equivalent to 100 T.

title: 18

description: 9887 Test for qualitative and quantitative determination of IgG antibodies to SARS CoV 2 IgG.

title: 19

description: 9888 Tests for Immunoglobulin M antibody for SARS COV 2 IgM.

title: Procurement of Enzyme Hemiluminescence Tests (CLIA metod)

description: Procurement of enzyme hemiluminescence tests (CLIA metod).

street: 50 Divizija Br.6


name: Ss. Cyril and Methodius University Faculty of Medicine Skopje

title: Reagent Test Kit for Molecular Identification and Detection of SARS CoV 2 from Biological Samples based on the Principle of Multiplex RT PCR Technology or Equivalent with Appropriate Equipment

description: Reagent test kit for molecular identification and detection of SARS CoV-2 from biological samples based on the principle of multiplex RT PCR technology or equivalent with appropriate equipment.

title: Reagent Test Kit for Molecular Identification and Detection of SARS CoV 2 from Biological Samples Based on the Principle of Multiplex RT PCR Technology or Equivalent with Appropriate Equipment

description: Reagent test kit for molecular identification and detection of SARS CoV 2 from biological samples based on the principle of multiplex RT PCR technology or equivalent with appropriate equipment.

street: 50 Divizija No.6


title: Spare parts for helicopters Mi-8MT/17

title: Spare parts for helicopters Bell 206B3

title: Spare parts for aircraft ZLIN 242L and ZLIN 143L

title: Spare parts for simulators for Mi-17 and Mi-24V

title: Spare parts for aircrafts (Mi-8MT/17, Bell 206B3, Zlin 242L and Zlin 143L) and simulators

Ministry of Defence

street: Orce Nikolov 116


name: Ministrry of Interior affairs of R.Macedonia

description: Call centre services.

title: Call Centre Services

description: Call centre services.


description: Procurement of electricity for the second time for the needs of PCE Derven Veles

title: Purchase of electricity for the second time

PCE Derven

street: Vardarska 18


name: PHI City General Hospital 8-mi Septemvri Skopje

description: Laboratory reagents.

title: Laboratory Reagents

description: Laboratory reagents.

street: Pariska bb


description: Provided in the tender documentation.

title: Sophisticated measuring equipment for quality parameters measurement in public mobile networks

description: Provided in the tender documentation.

Agencyfor Electronic Communications

street: Kej Dimitar Vlahov No.21


title: Supply of Electricity for the Needs of the Municipality of Radovis for Street Lighting

currency: EUR

description: Supply of electricity for the needs of the Municipality of Radovis for Street Lighting.

title: Supply of Electricity for the Needs of the Municipality of Radovis, Primary School ‘Kosta Susinov’ Radovish, ‘Nikola Karev’ Radovish, ‘Krste Petkov Misirkov’ Radovish, Orce Nikolv v.Injevo and JOUDG

description: Supply of electricity for the needs of the Municipality of Radovis, Primary School ‘Kosta Susinov’ Radovish, ‘Nikola Karev’ Radovish, ‘Krste Petkov Misirkov’ Radovish, Orce Nikolv v.Injevo and JOUDG Aco Karamanov.

title: North Macedonia-Radovish: Electricity

description: Supply of electricity for the needs of the Municipality of Radovis.

Radovish Municipality

street: Boulevard "Aleksandar Makedonski"7


name: Ministry of Interior Affairs of R. Macedonia

title: Representative Material for EBR

description: Representative material for EBR.

title: Flags

description: Flags.

title: Plackets

description: Plackets.

title: Indicators

description: Indicators.

title: Flowers Arrangements

description: Flowers arrangements.

title: Visit cards

title: Polo shirts

description: Polo shirts.

title: Paper bags

description: Paper bags.

title: Pens, Lighters, Aromatic Bags

description: Pens, lighters, aromatic bags.

title: Pendants and Other Representative Material with Logo of MOI

description: Pendants and other representative material with logo of MOI.

title: Representative Material

description: Representative material.


description: Set for Femoral artery access set consisting of a hydrophilic sheath 10 cm long and 4 to 9 in diameter in diameter, with anti-fold layer, hemostatic valve, dilator, 45cm wire 0.038 "thick, and compatible 18G metal needle 7cm long for femoral puncture

title: Set for radial artery access set consisting of a hydrophilic sheath with a diameter of 4 to 6 cm with anti-bending layer and a length of 10 to 25 cm, dilator, with hemostatic valve, plastic wire with

description: Set for radial artery access set consisting of a hydrophilic sheath with a diameter of 4 to 6 cm with anti-bending layer and a length of 10 to 25 cm, dilator, with hemostatic valve, plastic wire with a length of 45 to 80 cm and thickness of 0.025 ", 20G puncture cannula, length 32 mm, syringe 2.5ml

description: Hydrophilic sheath for peripheral interventions 45 to 90 cm long, straight or curved at the tip, 5 to 8 mm outer diameter (inner diameter at least 2.2 mm by 6F), with support steel spiral anti-wrinkle and x-ray marker on tip, inner PTFE layer and nylon outer layer, coated with 5 and 35cm hydrophilic layer on distal end, with Tuohy-Borst Sidearm or Cross-cut valve, 0.038 ", compatible wire.

title: Nitinol guidewire, hydrophilic, for placement of catheters through peripheral arteries with standard or rigid support, (stiff type) conical thinned tip with powder, angled or curved 1.5 mm and 2.0 mm

description: Nitinol guidewire, hydrophilic, for placement of catheters through peripheral arteries with standard or rigid support, (stiff type) conical thinned tip with powder, angled or curved 1.5 mm and 2.0 mm J style. Coated with fluoroscopically visible polyurethane with tungsten and hydrophilic layer along its entire length. Diameter from 0.018 to 0.035 inches and length 180 to 300 cm

title: Guidewire, hydrophilic, with hybrid design for peripheral interventions. Made of nitinol core with a conical tip with 25 cm polyurethane coating, x-ray visible marker, and the rest of the PTFE spiral

description: Guidewire, hydrophilic, with hybrid design for peripheral interventions. Made of nitinol core with a conical tip with 25 cm polyurethane coating, x-ray visible marker, and the rest of the PTFE spiral coating with a hydrophilic layer. Tip design with 1cm flexibility for 0.014 and 0.018 inch diameters and 5cm for 0.035 inch diameter. Lengths from 180 to 300 cm.

description: Diagnostic catheters for access to cervical arteries and cerebral access with different SIMMONS curves (1,2,3) diameter from 4 to 6 French, Internal diameter for 5 French from 0.047 inches made of hydrophilic nylon, span length from 100 cm to 125 cm with double mesh wire support. Compatible with 0.038 "guide wires.

description: Coronary microcatheter with hydrophilic coating, from 130cm to 150cm in length, distal outer diameter 1.8 Fr.proximal outer diameter max 2.6 Fr., compatible with 0.014 inches of wire. With conical "tapered" lumen and gold X-ray visible marker on tip

description: Peripheral support catheter with dual internal mesh and 3 RTG visible markers located 4 and 6 cm from the tip for accurate lesion measurement and stent or balloon positioning. Diameter 2.6 Fr and 4 Fr, compatible with wire 0.018`` 0.035``, lengths 60, 90, 135 and 150 cm. With straight or 30 degree curved tip.

description: Radial band for hemostasis after radial approach, with visibility at the puncture site and selective compression of the radial artery, on the principle of inflation with a double balloon in a set with a 25 ml syringe, adaptable size, and choice of dimensions.

title: Vascular closure system with collagen and resorbable suture, in set with an introducer and a wire guide of 70 cm, size 6 to 8 French for Transfemoral haemostasis

description: Temporary transvenous electro-stimulation electrode, bipolar with a pole distance of 10 mm, with a balloon, and with gilded pins. Diameter compatible with 5 Fr introducer and 110 cm long, with deep markers every 10 cm. The kit needs to contain a suitable syringe.

title: Hydrophilic sheath and flexible for peripheral interventions, up to 110 cm long choice of straight or curved tip, 6F outer diameter with at least 2.2 mm inner diameter with anti-bending steel spiral

description: Hydrophilic sheath  and flexible for peripheral interventions, up to 110 cm long choice of straight or curved tip, 6F outer diameter with at least 2.2 mm inner diameter with anti-bending steel spiral and x-ray visible marker on tip, PTFE inner layer, with Tuohy-Borst Sidearm valve, with optional carotid set with carotid catheter. Compatible with 0.035 inch wire

description: Lunderquist support guideweire, Extra-stiff design of extremely strong support for catheter placement and replacement through highly extremely tortuous or calcified arteries, support stronger than amplitude ultra stiff design, made of steel and with flopy atraumatic tip with spiral design curved or straight. tip 4-7 cm fluoroscopically visible, 0.035 inches in diameter and at least 260 cm long

description: Thermodilution catheter set for jugular and femoral application with a diameter of 7 francs. The set should contain: proximal lumen port 29cm, catheter with a length of 110cm thermodilution catheter with a balloon Swan-ganz catheter maximum 1.5cm and inflation diameter 12mm (with allowed deviation of +/- 5%), control syringe, extension lines, 4 channels, long distal lumen, clear color code and labeled extension lines, with sterile full-length protection and integrated flush / balloon test chamber.

description: Central venous pressure set 7F double-layer polyurethane catheter with soft atraumatic tip, lumen size 16G / 16G, outer diameter 2.5 mm and length 20 cm, transparent additional extension with side clamp, flexible guide wire with J-tip steel with dimensions 0.9mm x 50cm, bending resistant, Soldinger 18G introducer needle with length 70mm, radio lens in length, integrated movable fixation

description: Hydrophobic sheath for peripheral interventions 45 to 90 cm long, straight or curved at the tip, 4 to 6 mm outer diameter (inner diameter at least 1.61 mm for 4F, 1.91 mm for 5F and 2.25 mm for 6F), with anti-bending steel spiral, and outer layer coated with 30cm hydrophobic material at distal end Compatible with 0.035 inch wire

description: Coronary Balloon Catheter, coated with Paclitaxel drug on BTHC carrier, for dilatation of stent restenoses, compatible with 5 French guiding catheter and 0.014 inch wire, Rx. Range of diameters from 2.0 to 4 mm, and balloon lengths from 10 to 30 mm.

description: Balloon catheters for Peripheral arterial dilatation, arteries below the knee, compatible with 0.014 "wire, diameter range 1.5 to 4.0 mm lengths 20 to 220 mm. Shaft length 90 to 150 cm. Hydrophilic lining of the balloon, hydrophobic on the shaft , Compatible with 4 Frenci guiding introducer

title: Balloon catheter for peripheral arteries compatible with 0.035 "guidewire, OTW system, balloon diameter from 3.0mm to 10.0mm balloon length of 20-200 mm, compatible with 5 and 6 F guiding introducer,

description: Balloon catheter for peripheral arteries compatible with 0.035 "guidewire, OTW system, balloon diameter from 3.0mm to 10.0mm balloon length of 20-200 mm, compatible with 5 and 6 F guiding introducer, catheter length 80 to 135 cm , Hydrophilic coating

description: Balloon catheter for peripheral arteries OTW coated with cytostatics coated with Paclitaxel on BTHC carrier, compatible with 0.018 "guidewire with balloon diameters from 2.0 mm to 7.0 mm and 40-120 mm balloon length with catheter length 90,130 and 150 cm ,, compatible with 4 and 5 F intoducer

title: Balloon catheter for peripheral arteries OTW, coated with cytostatic drug Paclitaxel (concentration 3 /g / mm2) in 1: 1 mixture with "Shellac", hydrophilically coated "distal shaft", compatible with 0

description: Balloon catheter for peripheral arteries OTW, coated with cytostatic drug Paclitaxel (concentration 3 /g / mm2) in 1: 1 mixture with "Shellac", hydrophilically coated "distal shaft", compatible with 0.035 "guidewire with balloon diameters of 4.0 mm to 8.0 mm and with lengths of 20mm to 150mm

title: Femoral artery access set consisting of a 30 to 35 cm long sheath with a diameter of 12 to 20 frets with a hemostatic valve, a dilator with a gradually conical diameter, for a large diameter femoral p

description: Femoral artery access set consisting of a 30 to 35 cm long sheath with a diameter of 12 to 20 frets with a hemostatic valve, a dilator with a gradually conical diameter, for a large diameter femoral puncture (TAVI, TEV . EVAR)

title: Guidewire for peripheral interventions, below the knee, with hybrid design, steel core and nitinol tip, suitable for shaping, hydrophilic polymer coating that is at least 3 cm visible, the cord weighi

description: Guidewire for peripheral interventions, below the knee, with hybrid design, steel core and nitinol tip, suitable for shaping, hydrophilic polymer coating that is at least 3 cm visible, the cord weighing 2.8 to 3.5 g, diameter 0.014 inches and length from 190 to 300 cm.

title: Guidewire for coronary interventions made of nitinol alloy, intended for routine interventions, with a diameter of 0.014 inches, length from 190 to 300 cm, flexible tip from 0.6 to 0.8 g weight with a

description: Guidewire for coronary interventions made of nitinol alloy, intended for routine interventions, with a diameter of 0.014 inches, length from 190 to 300 cm, flexible tip from 0.6 to 0.8 g weight with a spiral part of 3 cm RTG visible, shaping strip tip, a coil at the tip, and a predefined straight or curved shape at the tip. The wire may have a hydrophilic or hydrophobic coating on the tip,

title: Guidewire for coronary interventions made of steel alloy, Hydrophilic, with polymer sleeve 20-30 cm from the tip,Guiding for tortuous arteries, dimension 0.014 ", from 190 cm to 300 cm in length with

description: Guidewire for coronary interventions made of steel alloy, Hydrophilic, with polymer sleeve 20-30 cm from the tip,Guiding  for tortuous arteries, dimension 0.014 ", from 190 cm to 300 cm in length with different degree of support at the tip (LS, MS, , Extra support) different tip weights from 0.8 to 1.5 g.Core-to tip style, in a whole for better control, RTG visibility on the tip of 3 cm.

title: Guidewire for coronary interventions made of nitinol alloy, for reinforced support during interventions, with a diameter of 0.014 inches, length from 190 to 300 cm, flexible tip of 0.8 g weight with

description: Guidewire  for coronary interventions made of nitinol alloy, for reinforced support during interventions, with a diameter of 0.014 inches, length from 190 to 300 cm, flexible tip of 0.8 g weight with a spiral part of 4.5 cm RTG visible, with a tip style and a shaping strip. with hydrophilic coating, predefined straight or curved shaped tip.

title: Wire guide for coronary interventions made of steel alloy, with flexible and delicate -floppy - tip for interventions, diameter 0.014 inches, length 190 to 300 cm, flexible tip 0.4 to 0.6 g weight wit

description: Wire guide for coronary interventions made of steel alloy, with flexible and delicate -floppy - tip for interventions, diameter 0.014 inches, length 190 to 300 cm, flexible tip 0.4 to 0.6 g weight with spiral part of 2 cm RTG visible, with tip style in the form of coils and shaping strip. by hydrophilic or hydrophobic coating.

title: Guidewire for coronary interventions made of steel alloy, for extra support during interventions, 20 grams at 70 mm from the tip, with a diameter of 0.014 inches, length 190 to 300 cm, flexible tip of

description: Guidewire for coronary interventions made of steel alloy, for extra support during interventions, 20 grams at 70 mm from the tip, with a diameter of 0.014 inches, length 190 to 300 cm, flexible tip of 0.9 g weight, length 3 cm X-ray visible predefined straight or curved shaped tip. Core-to-tip style -, as a whole, for better control, wire with hydrophobic coating.

title: Guidewire for coronary interventions in chronic occlusions, made of steel alloy, 0.014 inches in diameter, length 190 to 300 cm, Hydrophilic with a polymer sleeve at least 20 cm and without polymer a

description: Guidewire for coronary interventions in chronic occlusions, made of steel alloy, 0.014 inches in diameter, length 190 to 300 cm, Hydrophilic with a  polymer sleeve at least 20 cm and without polymer at the top, penetrating tip from 4.7 to 14 g weight, 3 cm long RTG visible, predefined straight or curved shaped tip. Core-to-tip style - from, as a whole, for better control, certain models with conical narrowing at the tip.

title: Guidewire for coronary interventions in chronic occlusions, made of steel alloy, diameter 0.010 inches, length 190 to 300 cm, Hydrophilic, at the tip penetrating tip conical narrowed from 4.7 to 9 g w

description: Guidewire for coronary interventions in chronic occlusions, made of steel alloy, diameter 0.010 inches, length 190 to 300 cm, Hydrophilic, at the tip penetrating tip conical narrowed from 4.7 to 9 g weight, length 3 cm RTG visible , predefined straight or curved formal tip. Tip-style - core-to-tip - from core to tip, as a whole, for penetrating technique when passibg through occlusion.

title: Non-deformable coronary balloon catheters NC - (non compliant), intended for predilation of hard calcified coronary lesions and occlusions and for post-dilatation in the stent. Compatible with 0.014 i

description: Non-deformable coronary balloon catheters NC - (non compliant), intended for predilation of hard calcified coronary lesions and occlusions and for post-dilatation in the stent. Compatible with 0.014 inch wire. Diameter in the range from 1.5 to 5 mm, and lengths in the range from 6 to 25 mm. Rx wire placement, low pass profile of 0.026 inches and input profile of 0.019 inches. Compatible with kissing technique (KBT) with simultaneous passage of two balloons in 6 French catheters

title: Balloon catheters for Peripheral Arterial and Carotid Dilatation, RX (rapid exchange) compatible with 0.014 "wire, diameter range 4.0 to 7.0 mm lengths 15 to 40 mm. Shaft length 80 to 135 cm. Material

description: Balloon catheters for Peripheral Arterial and Carotid Dilatation, RX (rapid exchange) compatible with 0.014 "wire, diameter range 4.0 to 7.0 mm lengths 15 to 40 mm. Shaft length 80 to 135 cm. Material properties for fast inflation and deflation and the ability to quickly remove from the dilatation site.compatible with 6 French guiding catheters.

title: Percutaneous Closure System with punctured artery or vein suture, after completion of large diameter procedures up to 21 Fr. for arterial and 24 Fr. for venous vessel. Compatible and recommended for

description: Percutaneous Closure System with punctured artery or vein suture, after completion of large diameter procedures up to 21 Fr. for arterial and 24 Fr.  for venous vessel. Compatible and recommended for Transfemoral haemostasis after TAVI and related procedures.

title: Percutaneous system Closure of the punctured femoral artery, after the completion of procedures that use introducers from 12 Fr to 25 Fr outer diameter. The systemneeds to enable rapid hemostasis with

description: Percutaneous system Closure of the punctured femoral artery, after the completion of procedures that use introducers from 12 Fr to 25 Fr outer diameter. The systemneeds to enable rapid hemostasis with collagen material. Available as 14 and 18 Fr. Compatible and recommended for Transfemoral haemostasis after TAVI and related procedures.

title: Guidewire for peripheral interventions in chronic occlusions, single core design, for peripheral interventions with penetrating technique, diameter of 0.014 and 0.018 inches, and conical tip of 0.008

description: Guidewire for peripheral interventions in chronic occlusions, single core design, for peripheral interventions with penetrating technique, diameter of 0.014 and 0.018 inches, and conical tip of 0.008 to 0.013 inches, at least 15 cm RTG visibility at the tip, weight at the tip of 20, 30 and 40 gr.

title: Guide wire for complex coronary interventions for overcoming long complex lesions and total occlusion with microchannel technique, Built with one-piece Core wire and coating, with a "tapered tip" 0.01

description: Guide wire for complex coronary interventions for overcoming long complex lesions and total occlusion with microchannel technique, Built with one-piece Core wire and coating, with a "tapered tip" 0.014 /0.010"(0.36mm/0.23mm) hydrophilic coated tip with polymer 16 cm RTG visible and a "tip load" 0.6-1.0 g, from 190 cm to 300cm length range

title: Guidewire for coronary intervention for lateral branch access in highly tortuous and stenotic coronary arteries, constructed of single-alloy steel, for better control, 0.014 inches in diameter, 180 to

description: Guidewire for coronary intervention for lateral branch access in highly tortuous and stenotic coronary arteries, constructed of single-alloy steel, for better control, 0.014 inches in diameter, 180 to 300 cm long, flexible tip 0.5 to 0.5 1 g weight, Hydrophilic coating in length from 18 to 40 cm RTG visible in the first three cm,

title: Guidewire for complex coronary interventions for overcoming long complex lesions and total occlusion with penetration technique, with steel alloy core with one-piece core design, with a tapered tip 0.

description: Guidewire for complex coronary interventions for overcoming long complex lesions and total occlusion with penetration technique, with steel alloy core with one-piece core design, with a tapered tip 0.014 "/0.010- 0.012 "hydrophilic coating tip 40 cm, 15 cm RTG visible and a " tip load "1.7-4.5 g, length 190 cm, given specific shape of the tip, which is a penetrating uncoated cone

title: Guiding Catheters for coronary interventions (minimum diameter span 5F, 6F, 7F, 8F) left, right, different curves, RTG visibility and RTG marker (strip) just below the tip, atraumatic tip, large inter

description: Guiding Catheters for coronary interventions (minimum diameter span 5F, 6F, 7F, 8F) left, right, different curves, RTG visibility and RTG marker (strip) just below the tip, atraumatic tip, large internal lumen of at least 0.071 "for 6F, compatible with two-balloon kissing technique, minimum catheter length span 90-100 cm (with the possibility of making more curves as needed) Compatible with 0.038 "guidewires.

title: Hydrophilic catheter for thrombic aspiration in coronary arteries, on the principle of vacuum 6 i 7 F, length up to 140 cm, compatible with 0.014 "wire, hardening steel stylet, in a set with two vacuu

description: Hydrophilic catheter for thrombic aspiration in coronary arteries, on the principle of vacuum 6 i 7 F, length up to 140 cm, compatible with 0.014 "wire, hardening steel stylet, in a set with two vacuum syringes and filter baskets.

title: Balloon catheter for Valvuloplasty diameter from 18mm to 30mm and length 30 to 60 mm.

title: Coronary Balloon Dilatation Catheters, semicompliant, low profile tip 0.41 mm or 0.016 inches, for stenosis, hydrophilic lining, compatible with 5 French guiding catheter, Rx and OTW options, compatib

description: Coronary Balloon Dilatation Catheters, semicompliant, low profile tip 0.41 mm or 0.016 inches, for stenosis, hydrophilic lining, compatible with 5 French guiding catheter, Rx and OTW options, compatible with 0.014 inch wire. Minimum span of diameters from 1.5 to 4 mm, and minimum span of balloon lengths from 6 to 30 mm. Compatible with ballon kissing technique with simultaneous passage of two balloons in 6 French catheters

title: Single-volume infusion catheter for thrombolysis, with occlusive valve at the distal tip, and lateral openings for continuous or pulse infusion of thrombolytics, with distal and proximal marker for in

description: Single-volume infusion catheter for thrombolysis, with occlusive valve at the distal tip, and lateral openings for continuous or pulse infusion of thrombolytics, with distal and proximal marker for infusion section, length 40-135 cm, infusion part  5-50 cm, compatible with a guidewire of 0.035-0.038``

title: Intervention set consisting of Balloon Pump with ergonomic holder, working pressure 30 atm, mechanism for controlling the pressure during dilation Then epsilon (Y) connector, inlet needle, torque

title: Guiding Catheter Extension, "rapid exchange", compatible with 6 F guiding catheters, made of nylon with mesh as support, hydrophilic coating, total length of 145 cm, hardening steel stylet and 25 cm t

description: Guiding Catheter Extension, "rapid exchange", compatible with 6 F guiding catheters, made of nylon with mesh as support, hydrophilic coating, total length of 145 cm, hardening steel stylet and 25 cm tube with an inner diameter of 0.057 inches, with RTG type markers on the ends of the tube, soft atraumatic tip.

title: Coronary artery Thrombus Aspiration , vacuum principle, minimum compatible with 6 F introducer, length up to 140 cm, compatible with 0.014 "wire, with a distal hydrophilic sheath, with distal end prof

description: Coronary artery Thrombus Aspiration , vacuum principle, minimum compatible with 6 F introducer, length up to 140 cm, compatible with 0.014 "wire, with a distal hydrophilic sheath, with distal end profile less than 0.025``

title: Diagnostic angiographic catheters, multi-shaped and multi-purpose, diameter 5,6,7 Fr, lengths 100 cm and 125 cm, with atraumatic tip, resistant to irregular bending, made of Trillon or equivalent mate

description: Diagnostic angiographic catheters, multi-shaped and multi-purpose, diameter 5,6,7 Fr, lengths 100 cm and 125 cm, with atraumatic tip, resistant to irregular bending, made of Trillon or equivalent material, with full-length wire braiding

title: Steel wire guide for initial access and catheters standard support strength, atraumatic flexible tip with spiral wire design, J-shaped tip or Rosen and Benston-shaped tip, 0.032 to 0.038 thick and len

description: Steel wire guide for initial access and catheters standard support strength, atraumatic flexible tip with spiral wire design, J-shaped tip or Rosen and Benston-shaped tip, 0.032 to 0.038 thick and length range from 180cm to at least 260cm, with steel core and PTFE coating

title: Steel guide wire for TAVI procedure of 0.035 inches (with pre-formed loop shape in three sizes from 29 to 50 mm at the distal 15 cm at the tip) Wire with a total length of at least 260 cm, designed fo

description: Steel guide wire for TAVI procedure of 0.035 inches (with pre-formed loop shape in three sizes from 29 to 50 mm at the distal 15 cm at the tip) Wire with a total length of at least 260 cm, designed for stable atraumatic wire placement in the left ventricle during the TAVI procedure. Wire strength less than Amplatz ultra hard wire.

title: Wire guide for interventions in highly tortuous and stenotic coronary arteries, made of steel alloy, for extra support during interventions, diameter 0.014 inches, length 182 to 300 cm, flexible tip o

description: Wire guide for interventions in highly tortuous and stenotic coronary arteries, made of steel alloy, for extra support during interventions, diameter 0.014 inches, length 182 to 300 cm, flexible tip of 2.0 g weight, Hydrophilic with polymer sleeve in length of 35 cm RTG visible, Core-to-tip style -as a whole, for better control,

title: Peripheral hydrophilic wire, stainless steel core, 0,014-0,018``, with lengths from 190 to 300 cm, with a tip load of 12 to 30 grams

title: Peripheral hydrophilic wire, stainless steel core, 0.018``, with lengths from 110 to 300 cm, with a tip load of 6 to 8 grams

title: Peripheral hydrophilic wire, stainless steel core, 0.014``, with lengths from 185 to 300 cm, with a tip load of 3 to 6 grams

title: Coronary Balloon Catheters - NS- (Non Compliant) for predilatation of large coronary arteries with hard calcified lesions and occlusions and for post-dilatation in the LM stent. Compatible with 0.014

description: Coronary Balloon Catheters - NS- (Non Compliant) for predilatation of large coronary arteries with hard calcified lesions and occlusions and for post-dilatation in the LM stent. Compatible with 0.014 inch wire. Diameter in the range of 5.5 and 6 mm, and lengths in the range from 8 to 20 mm. Rx and OTW wire placement, low pass profile of 0.017 inches or 0.43mm.

title: Peripheral catheter, for support with hydrophilic coating of the distal 40 cm, with steel mesh in the wall, 4 x visible visible markers on tip of each of 5 cm. Support for arteries under the knee with

description: Peripheral catheter, for support with hydrophilic coating of the distal 40 cm, with steel mesh in the wall, 4 x visible visible markers on tip of each of 5 cm. Support for arteries under the knee with 4 French guiding introducers Conical tip straight or curved at an angle, catheter with a diameter 2.3 French  from 65 to 150 cm long, compatible with 0.014 inch wire

title: High pressure hose min 84 Bar (1200 psi) intended for angiography with injector with rotating m / f adapter, made of transparent polyurethane minimum length 100 cm.

title: Wire extension guide for PCI 0.014 "/ 0.36mm compatible 150cm to 165cm minimum length range

title: Balloon catheters for peripheral arteries compatible with 0.035 inch wire guide, OTW system, balloon diameter 3.0 to 14 mm, balloon length 20 to 250 mm, compatible with 5-7F guiding introducer, cathet

description: Balloon catheters for peripheral arteries compatible with 0.035 inch wire guide, OTW system, balloon diameter 3.0 to 14 mm, balloon length 20 to 250 mm, compatible with 5-7F guiding introducer, catheter length 80 to 135 cm , hydrophobic coating.

title: Medical consumables for intervention procedures - Introducers, wires, catheters and other medical consumables

description: Medical consumables for intervention procedures - Introducers, wires, catheters and other medical consumables

University Clinic of Cardiology

street: University Clinic of Cardiology, Vodnjanska 17, Meri Mandalova, 1000, Skopje, Republic of Macedonia; +389 2 3147181, +389 2 3164134,


title: Non-scheduled passenger transportation with buses for military personnel

description: Detailed description is given in the tender documentationhttps://e-nabavki.gov.mk/PublicAccess/home.aspx#/dossie/ce2a0cfb-b2ad-47a6-bd96-2e69e3efbbc5/1

title: Non-scheduled passenger transportation with passenger cars for military personnel

description: Detailed description is given in the tender documentation https://e-nabavki.gov.mk/PublicAccess/home.aspx#/dossie/ce2a0cfb-b2ad-47a6-bd96-2e69e3efbbc5/1

title: Non-scheduled passenger transportation with buses and passenger cars for military personnel

description: Detailed description for organized long distance transport for employees of MOD and ARM is given in the tender documentation https://e-nabavki.gov.mk/PublicAccess/home.aspx#/dossie/ce2a0cfb-b2ad-47a6-bd96-2e69e3efbbc5/1

Ministry of Defence

street: Orce Nikolov No 116


name: Ministrry of Interior affairs of R. Macedonia

description: Food services for ESZ.

title: Food Services for ESZ

description: Food services for ESZ.


description: genarators on disel power

title: Genarators on disel power

description: genarators on disel power


title: Glavni generatori serije 621,642/3 i 661

description: Opis nabave u tehnickoj specifikacii i dokumentacii za nadmetanje

title: Vučni elektromotori za seriju 621,412,441/461,642/3,661

description: Opis nabave u tehnickoj karakteristike i dokumentacii za nadmetanje

title: inicijalni generatori serija 642/643 i pomocni generatori serija 642/643 i 661

description: Opis nabave u tehnickе specifikacije i dokumentcii za nadmetanje

title: Utishnica

description: Opis nabave u tehncke specifikacije i dokumentacii za nadmetanje

title: Odrzavanje i popravak vucni elektromotora, generatora, drzaci za cetke i utisnica

description: Kratak opis u tehnickoj specifikacii i dokumentacii za nadmetanje


name: Company for audit, consulting and tax services Diloit DOO Skopje

name: Audit and Consulting Company DFK Consulting - Audit DOO Belgrade

name: Company for audit services PricewaterhouseCoopers audition DOO Skopje

name: Fund for Innovations and Technology Development

title: Project Management Services — Monitoring

description: Project management services — monitoring.

awardCriteria: ratedCriteria

street: bul. Goce Delcev b.b.

countryName: RS


description: Semi-trailers Variopress up to 30 m3

title: Semi-trailers Variopress up to 30 m3

description: Semi-trailers Variopress up to 30 m3


name: Ministry of interior of the Republic of North Macedonia

title: Maintenance of Equipment, Mechanical Cleaning

description: Maintenance of equipment, mechanical cleaning.

title: Building and Craft Services

description: Building and craft services.

title: Servicing and Maintenance of High-pressure Equipment, Therm-technical Installation, Chemical and Mechanical Cleaning of Boiler Units

description: Servicing and maintenance of high-pressure equipment, therm-technical installation, chemical and mechanical cleaning of of boiler units.


description: Procurement of heating oil EL-1Place of delivery of the goods or performance of the services, or location of the works: extra light burning oil 1 (EL-1) (for heating of asphalt bases and central heating) will be delivered to the following dispositions and that— Asphalt base in Lepenec (Skopje);— Asphalt base in Bitola Bitola - village Kukurecani - Bitola;— Asphalt base in Veles - street Skopski pat bb-Veles;— Asphalt base in Stip - highway A4 at 10 km, Skopje-st Quay 13-ti Noemvri No 76;— Branch Bitola-ul. Gjuro Gjakovic No 12-Bitola ;— Branch Veles-Veles, ul Skopski pat bb-Veles;— Branch Stip-Stip, ul. Partizanska br.29-Stip.

title: Procurement of Heating Oil EL-1

description: Procurement of heating oil EL-1.

Makedonijapat

street: Dame Gruev Str., No 14


name: JZU Univerzitetska Klinika za Hematologija — Skopje

title: Enoxoparin

description: Solution for injection 4 000 anti-Xa ie/0,4ml pre-filled syringe.

title: Nadroparin

description: Solution for injection 3800IE anti-Xa ie/0,4ml pre-filled syringe.

title: Sodium Chloride

description: Solution for infusion 0.9 % 100 ml ‘PE/LDPE bottle 0.9 % 100 ml’.

title: Sodium Chloride

description: Solution for infusion 0.9 % 1 000 ml bottle x 1 000 ml.

title: Trospium

description: Tablet 5 mg tablet.

title: Trospium

description: Solution for injection 0.2 mg /5 ml ampoule.

title: Imipenem+Cilastatin

description: Powder for solution for infusion/powder for solution for injection 500 mg/500 mg Vial.

title: Azithromycin

title: Azithromycin

description: Powder for solution for infusion 500 mg Vial.

title: Ciprofloxacin

description: Film-coated tablet 500 mg таблета.

title: Immunoglobulins, Normal Human (i.v.)

description: Solution for infusion 10 g/100 ml,10 g/200 ml, Vial.

title: Chlorambucil

description: Film-coated tablet 2 mg tablet.

title: Melphalan

description: Film-coated tablet 2 mg tablet.

title: Melphalan

description: Powder and vehicle for solution for injection or infusion 50 mg Vial.

title: Busulphan

description: Tablet 2 mg tablet.

title: Dacarbazine

description: Powder for solution for infusion or injection 100 mg Vial.

title: Methotrexate

description: Solution for injection or infusion 50 mg/5 ml.

title: Methotrexate

description: Tablet 2.5 mg tablet.

title: Hydroxycarbamide

description: Capsule, hard 500 мг capsule.

title: Tretinoin

description: Capsule, soft 10 mg capsule.

title: Lidocaine,Epinephrine

description: Solution for injection (40 mg+0,025 mg)/2 ml ampoule.

title: Erythropoetin

description: Solution for injection 2 000 IE pre-filled syringe.

title: Lenograstim

description: Powder and vehicle for solution for injection or infusion 263 mcg (33.6MIE/ml) Vial.

title: Nivolumab

description: Solution for injection or infusion 100 mg/10 ml Vial.

title: Lenalidomide

description: Capsule 5 mg capsule.

title: Sorafenib

description: Film-coated tablet 200 mg tablet.

title: Polatuzumab vedotin

description: Powder for concentrate for solution for infusion 140 mg/20 ml Vial.

title: Ropeginterferon alfa -2b

description: Solution for injection 250 mcg/0,5 ml pre-filled syringe.

title: Busulfan

description: Concentrate for solution for infusion 6 mg/ml(10 ml) Vial.

title: Voriconazole

description: Powder for solution for infusion 200 mg Vial.

title: Anagrelide

description: Capsule, hard 0.5 mg capsule.

title: Vinorelbine

description: Capsule 30 mg capsule.

title: Arsenic trioxide

description: Concentrate for solution for infusion 2 mg/ml(6 ml) Vial.

title: 02767/2021 Drugs

description: Tender documentation.

street: Majka Tereza No.17


title: List no.1 - Ductile water pipes

description: Ductile water pipes

title: List no.2 - PE hoses and PE pipes for minimum nominal pressure of PN16bar

description: Water pipes (ductile pipes and PE hoses) - according to specification listsII.

JP Vodovod i Kanalizacija Skopje

street: ul.Lazar Licenoski br.9


title: Public Procurement for Weather Modification with Cloud Seeding by Airplane

description: This procurement involves seeding of clouds against hail with reagent-silver iodide (AgJ) from airplane.The cloud seeding hypothesis is based on the concept of useful competition.Competition is a lack of natural ice cores in the cold cloud environment and injection of silver iodide creates a significant number of artificial crystallization cores.Arable and artificial ice crystals are ‘competing’ for the content of cool cloud water in the bearing cloud, and as a result of the method, the grains formed in the sown cloud volume will be smaller in size, causing less damage if they fall to the ground.Artificial cores seeding with silver iodide is carried out by specialized aircraft and seeding procedures depending on the meteorological conditions.

Hydrometeorological service

street: ul. Skupi 28

DVT Aeroklub Skopje DOO

street: Stenkovec br.151


name: Public Health Institution Healthcare Center of Skopje CO Skopje

title: Automatized Analyzer for Determining Erythroid Cells Sedimentation in Human Full Blood

description: Automatized analyzer for determining erythroid cells sedimentation in human full blood.

title: Automatized Integrated Immunochemical Analyzer

title: Procurement of New Medical Equipment

description: Automatized analyzer for determining erythroid cells sedimentation in human full blood, Automatized integrated immunochemical analyzer.

street: Krste Misirkov No 28


name: Public Entreprise for State Roads — Skopje

description: Awarding a Contract for Public Procurement of Electric Power for a 2-Year Period

street: Str."Dame Gruev", No. 14


name: Railways of Republic of North Macedonia Transport JSC Skopje

title: Kompozitne kočione papuče za ser. 712 L i R i 711/411 L i R

description: U dokumentaciji za nadmetanje i dodatnoj dokumentaciji.

title: Kompozitne kočione papuče za putničke i teretne vagone

title: Kočione kompozitne papuče

street: Bul.3-ta Makedonska brigada no.66


title: Medicinski plinovi 1

description: Medicinski plinovi.

title: Medicinski plinovi 2

description: Medicinski plinovi.

title: Medicinski plinovi

description: Medicinski plinovi.


title: Specijalizirana radna snaga za mlinsko postrojenje br.1

title: Specijalizirana radna snaga za mlinsko postrojenje br.2

title: Specijalizirana radna snaga za mlinsko postrojenje br.3

title: Specijalizirana radna snaga za mlinsko postrojenje br.4

title: Specijalizirana radna snaga za mlinsko postrojenje br.5

title: Specijalizirana radna snaga za remont rosta

title: Specijalizirana radna ruka za popravak usisnog ventilatora i ventilatora za gas

title: Specijalizirana radna snaga za remont sekundarnih zračnih kanala

title: Specijalizirana radna snaga za armaturu

title: Specijalizirani rad za mjerenje debljine cijevi na gr.263

title: Specijalizirana radna ruka za zamjenu oštećenih cijevi i koljena na gr.263 od uboda u dovodnom razvodniku uboda kroz sito cijevi do 1,5 m od prednje strane pregrijača prema crtežu.423.263

title: Specijalizirana radna snaga za zamjenu i rehabilitaciju štitnika u cijevnom sustavu kotla OB -380

title: Specijalizirana radna snaga za radiografsku kontrolu cijevi

title: Proizvodnja (nabava), demontaža i ugradnja cijevi iz pregrijača pare na kotlu br. 1 (poz. 7,8,9 i 10 (po pet cijevi) prema crtežu br.1-11893954c)

title: Specijalizirana radna snaga za ispitivanje penetrantima i magnetski tok za kotао i pomoćne objekte

title: Specijalizirani ručni rad za zamjenu zglobova ventila i lakta

title: Specijalizirani rad za stroj za spaljivanje

title: Specijalizirana radna snaga za turbinu, generator i budilicu, test prodiranja i magnetski tok za turbinu

title: Specijaliziranа radnа ruka za mjerenje vibracija nakon remonta turbine

title: Specijalizirana radna snaga za mjerenje vlastitih frekvencija

title: Specijalizirani rad za pjeskarenje dijelova turbine

title: Specijalizirana radna snaga za izolacijske vatrootporne radove

title: Specijalizirani rad za demontažu i ugradnju toplinske izolacije

title: Super kontrola

description: Remontne aktivnosti za te 2021.)usluge specijalizirane radne ruke)REK Oslomej.

A.D. „Elem", Podružnica REK „Oslomej”

street: Oslomej


description: Procurement of electricity

title: Procurement of electricity

description: Procurement of electricity

Makedonijapat

street: Dame Gruev Str., No.14


title: Ortopedski implanti

description: Ortopedski implanti.

title: Ortopedski implanti

description: Ortopedski implanti.

title: Ortopedski implanti

description: Ortopedski implanti.


title: Procurement of Integrated System for Biometric Voter Identification

description: According to the technical specification, an integral part of the tender documentation.

title: Procurement of Integrated system for Biometric Voter Identification

description: Procurement of integrated system for biometric voter identification.


title: Zavojni materijal.

title: Zavojni materijal

description: Zavojni materijal.

title: Zavojni materijal

description: Zavojni materijal.


description: More detailed description of the subject for the agreement for public procurement is given in the tender documentation attached to this announcement

title: Supply of extra light household oil -1(EL-1) for the needs of the ULG building ,local primary schools, local high schools and local kindergartens in Municipality Shtip

Municipality Shtip

street: Vasil Glavinov-bb


title: 1

description: 9198 DNA isolation kit of bacterial cultures for 50 isolations

title: 2

description: 6455 Brucella extraction, amplification, identification and differentiation reagent kit based on the principle of Multiplex Real Time PCR technique with internal and positive control included or equivalent

title: 3

description: 9533 Molecular genetic specificity set specific Multiplex Real Time PCR kit including extraction, amplification and differentiation of Plasmodium falciparum, P. vivax, P. ovale, P. malariae with internal and positive control included or equivalent

title: 4

description: 9553 Molecular test for Real Time PCR for amplification and detection of Treponema pallidum, complete with extraction kit suitable for the test offered and the number of tests.

title: 5

description: 9187 Real Time PCR molecular test for amplification and detection of Francisela tularensis, complete with extraction kit suitable for the test offered and the number of tests.

title: 6

description: 9178 Molecular test for Real Time PCR for amplification and detection of Legionella pneumophila and / or Legionella spp., Complete with extraction kit suitable for the test offered and the number of tests.

title: 7

description: 9499 Molecular test for Real Time PCR for amplification and detection of Corynebacterium diphteriae and diphtheria toxin, complete with extraction kit suitable for the test offered and the number of tests.

title: 8

description: 9176 Multiplex Real Time PCR molecular test for amplification and minimum detection of Bordetella pertussis / B.parapertussis, complete with extraction kit suitable for the test offered and the number of tests

title: 9

description: 9185 Multiplex Real Time PCR molecular test for amplification and detection of ESBL (extended spectrum beta-lactamase) resistance genes, complete with extraction kit suitable for the test offered and the number of tests.

title: 10

description: 7004 Multiplex Real Time PCR molecular test for amplification and detection of respiratory bacterial agents, minimum for Legionella pneumophila, Mycoplasma pneumoniae, Chlamydophila pneumonia complete with extraction kit suitable for the test offered and the number of tests.

title: 11

description: 9182 Multiplex Real Time PCR molecular test for amplification and detection of carbapenem resistance genes (KPC-like, VIM-like, IMP-like, NDM)

title: 12

description: 9873 PRIMER AND PROBES detection resistance genes to Colistin (mcr1 -mcr5) mcr1_F 5'-AGTCCGTTTGTTCTTGTGGC-3 'mcr1_R 5'-AGATCCTTGGTCTCGGCTTG-3' mcr2_F 5'-CAAGTGTGTTGGTCGCAGTT-3 'mcr2_R 5'-TCTAGCCCGACAAGCATACC-3' mcr3_F5 '-AAATAAAAATTGTTCCGCTTATG-3' mcr3_R 5'-AATGGAGATCCCCGTTTTT-3 'mcr4_F 5'-TCACTTTCATCACTGCGTTG-3' mcr4_R 5'-TTGGTCCATGACTACCAATG-3 'mcr5_F 5'-ATGCGGTTGTCTGCATTTATC-3' mcr5_R 5'-TCATTGTGGTTGTCCTTTTCTG-3 '

title: 13

description: 7393 Real Time PCR molecular test for amplification and detection of Neisseria gonorhoeae with pre-prepared paraffin or lyophilized (or equivalent) reagents individually prepared, complete with extraction kit suitable for the test offered and the number of tests.

title: 14

description: 8935 Real Time PCR molecular test for amplification and detection of Trichomonas vaginalis with pre-prepared paraffin or lyophilized (or equivalent) reagents individually prepared, complete with extraction kit suitable for the test offered and the number of tests.

title: 15

description: 7241 Molecular test for Real Time PCR for amplification and detection of Chlamydia trachomatis with pre-prepared reagents in the form of paraffin or lyophilized (or equivalent) individually prepared, complete with extraction kit suitable for the test offered and the number of tests

title: 16

description: 9174 Molecular test Multiplex Real Time PCR for amplification and detection of at least the following agents: Chlamydia trachomatis, Neisseria gonorrhoeae, Trichomonas vaginalis with already prepared paraffin reagents or lyophilized (or equivalent) prepared individually prepared for and the number of tests

title: 17

description: 9870 Automatic DNA and / or RNA swab extraction kit

title: Procurement of Molecular Tests from Bacterial Agents

street: 50 Divizija Br.6


description: Lijek – Ocrelizumab; Ocrelizumab 300mg/10ml - koncentrat za infuzijsku otopinu; 280 bočica.

title: Lijek – Ocrelizumab

description: Lijek – Ocrelizumab 300mg/10ml; lijek za liječenje multiple skleroze.

JZUU Klinika za Nevrologija – Skopje

street: Vodnjanska br. 17


description: According to the technical specification, an integral part of the tender documentation.

description: Procurement of integrated system for biometric voter identification.

State Election Comission

street: ul.Sv. Kiril i Metodij br.54, Skopje

DERMALOG Identification System GmbH

countryName: DE


title: TRANSFUSION SYSTEM

description: System for transfusion

title: Facial mask (anesthetic) made of silicone

title: Medical consumables

description: Medical consumables

street: Pariska bb


description: Place of delivery of goods or performance of services, or location of works: fuel for diesel engines will be delivered to the following locations:— Subsidiary Highway;— Petrovec;— Subsidiary SkopjeC Madzari, Tetovo, Gostivar, Debar, Kr. Palanka;— Subsidiary BitolaCMechanization Bitola, Prilep, Kicevo, Ohrid, Struga;— Branch Veles-Veles.

title: Procurement of Diesel Fule — Eurodisel BS(D-E 5)

description: Place of delivery of goods or performance of services, or location of works: fuel for diesel engines will be delivered to the following locations:— subsidiary highway;— Petrovec;— subsidiary SkopjeC Madzari, Tetovo, Gostivar, Debar, Kr. Palanka;— Subsidiary BitolaCMechanization Bitola, Prilep, Kicevo, Ohrid, Struga;— Branch Veles-Veles.


title: Frozen Meat, Meat Processing, Fish and Canned Meat and Fish Products

description: The object of this lot is procurement of frozen meat, meat processing, fish and canned meat and fish products.

title: Pasta and other Products, Flour and Bread

description: The object of this lot is procurement of pasta and other products, flour and bread.

title: Agricultural Products, Fresh Fruits and Vegetables

description: The object of this lot is procurement of agricultural products, fresh fruits and vegetables.

title: Spice and Sugar Related Products

description: The object of this lot is procurement of spice and sugar related products.

title: Eggs

description: The object of this lot is procurement of eggs.

title: Milk and Dairy Products

description: The object of this lot is procurement of milk and dairy products.

title: Canned Vegetables

description: The object of this lot is procurement of canned vegetables.

title: Food

description: The object of the public procurement is procurement of supplies — food for the needs of the contracting authority


name: Trade and Services Company CNG SYSTEMS LLC Skopje

measure: smeBids

name: Public utility company for energy related activities “Strumica-Gas” Strumica

title: Procurement of Natural Gas

description: Procurement of 900 000 nm3 of compressed natural gas (CNG) of the needs of JPED ‘Strumica-Gas’ Strumica of the period of 2021-2022

title: Procurement of Natural Gas

description: Procurement of compressed natural gas (CNG) of the needs of JPED ‘Strumica-Gas’ Strumica of the period of 2021-2022

street: 24Oktomvri No. 2


description: On the Tender Documentation.

title: Services from Private Agencies for Temporary Employment

description: On the Tender Documentation.

Municipality of Cair — Skopje

street: Hristijan Todorovski Karpos No. 5


title: Consumables Compatible with WALKAWAY Plus 96 Apparatus, or Equivalent Indivisible Lot

description: In the tender documentation.

title: Tests and reagents compatible with the Mispa i3 apparatus or equivalent INDIVISIBLE LOT

description: In the tender documentation

title: Infectious diseases diagnostics tests compatible with the VirClia automatized analyzer or equivalent INDIVISIBLE LOT

description: In the tender documentation

title: Reagents for isothermal amplification compatible with the ANSR apparatus or equivalent INDIVISIBLE LOT

description: In the tender documentation

title: Tests for isothermal amplification compatible with the Amplex easyplex system apparatus or equivalent INDIVISIBLE LOT

description: In the tender documentation

title: Microbiological tests for infections compatible with the ALTA M test reader or equivalent – INDIVISIBLE LOT

description: In the tender documentation

title: Reagents compatible with Real Time - PCR DTlite (software DTMaster), abGenix and InnuPure C16 apparatuses or equivalent – INDIVISIBLE LOT

description: In the tender documentation

title: Consumables for positive control for CMV compatible with the InGenius apparatus or equivalent INDIVISIBLE LOT

description: In the tender documentation

title: Tests, reagents and consumables compatible with Vitek 2 Compact apparatus or equivalent INDIVISIBLE LOT

description: In the tender documentation

title: Spare parts compatible with Elix UV 3 + Tank 30 L apparatus or equivalent INDIVISIBLE LOT

description: In the tender documentation

title: Spare parts compatible for the RiOs Essential 5 Water Purification System apparatus or equivalent INDIVISIBLE LOT

description: In the tender documentation

title: Reagents for Vidas 3 apparatus or equivalent INDIVISIBLE LOT

description: In the tender documentation

title: Means for Appliances for Microbiological Laboratory

description: More details in the tender documentation.

PHO Center for Public Health — Skopje

street: Nr.18, Third Macedonian Brigade str.


description: Procurement of service-excavation and cleaning of road canopies for a period of 2 years.

title: Procurement of Service-excavation and Cleaning of Road Canopies for a Period of 2 Years


description: Construction of Regional Road R2246, section Izvor (connection with A2) – Garski Most.

mainProcurementCategory: works


name: Ministry of Interior of the Republic of Macedonia

title: Skopje — Services for Health Examinations of Employees in Organisational Units for the Needs of the Ministry of Interior, and the Public Security Bureau in Skopje and Employees in the Sector for Inter

description: Skopje — services for health examinations of employees in organisational units for the needs of the Ministry of Interior, and the Public Security Bureau in Skopje and employees in the Sector for Internal Affairs — Skopje as employees of RCGR North.

title: Bitola — Services for Health Examinations of Employees in the Sector for Internal Affairs — Bitola, Based in Bitola

description: Bitola — services for health examinations of employees in the sector for internal affairs — Bitola, based in Bitola.

title: Veles — Services for Health Examinations of Employees in the Sector for Internal Affairs — Veles, based in Veles

description: Veles — services for health examinations of employees in the sector for internal affairs — Veles, based in Veles.

title: Kumanovo — Services for Health Examinations of Employees in the Sector for Internal Affairs — Kumanovo, Based in Kumanovo

description: Kumanovo — services for health examinations of employees in the sector for internal affairs — Kumanovo, based in Kumanovo.

title: Ohrid — Services for Health Examinations of Employees in the Sector for Internal Affairs — Ohrid, Based in Ohrid, as Employees of RCGR West

description: Ohrid — services for health examinations of employees in the Sector for Internal Affairs — Ohrid, based in Ohrid, as employees of RCGR West.

title: Strumica — Services for Health Examinations of Employees in the Sector for Internal Affairs — Strumica, Based in Strumica

description: Strumica – services for health examinations of employees in the sector for internal affairs — Strumica, based in Strumica.

title: Tetovo — Services for Health Examinations of Employees in the Sector for Internal Affairs — Tetovo, Based in Tetovo

description: Tetovo — services for health examinations of employees in the sector for internal affairs — Tetovo, based in Tetovo.

title: Stip — Services for Health Examinations of Employees in the Sector for Internal Affairs — Stip, with Headquarters in Stip

description: Stip — services for health examinations of employees in the sector for internal affairs — Stip, with headquarters in Stip.

title: RC Istok — Services for Health Examinations of Employees Based in Delchevo

description: RC Istok — services for health examinations of employees based in Delchevo.

title: RC South — Services for Health Examinations of Employees Based in Kavadarci

description: RC South — services for health examinations of employees based in Kavadarci.

title: Skopje — Services for Health Examinations of the Employees in the OSPO of the Ministry of Interior in Skopje

description: Skopje — services for health examinations of the employees in the OSPO of the Ministry of Interior in Skopje.

title: Systematic Health Examinations for Employees

description: Systematic health examinations for employees.

street: Dimce Mircev 9


title: Office chairs type 1

description: Detailed description of the procurement is provided in the tender documentation

title: Metal cabinets and drawers

title: Fireproof metal sefs

description: is provided in the tender documentation

title: Office desks

description: is provided in the tender documentation

title: Office chairs

description: is provided in the tender documentation

title: Office Furniture and Equipment

description: Office Furniture and Equipment


description: Reagent for detection of HbsAg applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD acoording tehnical specification-60 000tests

description: Reagent for detection of Anti HCV applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD  acoording tehnical specification -60.000tests

description: Reagent for detection of HIV Ag/Att Combo applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD  acoording tehnical specification -60.000tests

description: Reagent for detection of Syphilis (TP) applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD acoording tehnical specification -60.000tests

description: HbsAg calibrator applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD  acoording tehnical specification - 56 ml

description: HbsAg control applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD  acoording tehnical specification - 192 ml

description: ANTI HCV calibrator applicable to immune analyzer Architect i2000sr "or equivalent for CHEMYLUMIMSENCE TEST METHOD acoording tehnical specification - 28 ml

description: ANTI HCV control applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD acoording tehnical specification - 192 ml

description: HIV calibrator applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD acoording tehnical specification - 28ml

description: HIV control applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD acoording tehnical specification - 384ml

description: SYPHILIS calibrator applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD acoording tehnical specification - 28ml

description: SYPHILIS control applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD acoording tehnical specification -192ml

description: Rubber bottle caps for reagents applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD acoording tehnical specification -1800pieces

description: Replaceable plastic bottle caps for reagents applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD acoording tehnical specification -200pieces

description: Concentrated solution for washing needles internally and reactive cells (1.5 M phosphate puffer with tolerance of + / - 1%) applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD acoording tehnical specification -920l

description: Solution for chemiluminescent pre-activation in analysis sample (1.32% hydrogen peroxide with tolerance of + / - 1%) applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD acoording tehnical specification -120l

description: Solution for chemiluminescent activation in analysis sample (PEG oxyphenyl ether 1.98% and NaOH 1.38% with tolerance of + / - 1%) applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD acoording tehnical specification -220l

description: Plastic cups for sample application, calibrators and controls applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD acoording tehnical specification -2000pieces

description: Reactive cells applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD acoording tehnical specification -260000pieces

description: Solution for needle conditioning (recalcifying human plasma) applicable to immune analyzer Architect i2000sr "or equivalent" for CHEMYLUMIMSENCE TEST METHOD acoording tehnical specification -200ml

description: REAGENTS, MATERIALS AND CALIBRATORS FOR THE CONTROL OF BLOOD TRANSMITTED DISEASES FOR APPARATUS Architect i2000sr or equivalent  for CHEMYLUMIMSENCE TEST METHOD according tehnical specification

Institute of Transfusion Medicine

street: St Mother Tereza 17


description: Preventive and adaptive maintenance of the digital archive system and digitalisation of documentation.

title: Preventive and Adaptive Maintenance of the Digital Archive System and Digitalisation of Documentation

description: PREVENTIVE AND ADAPTIVE MAINTENANCE OF THE DIGITAL ARCHIVE SYSTEM AND DIGITALIZATION OF DOCUMENTATION


title: Purchase of 1 600 Pair of Shoes for the Needs of the Prison Police Members

description: Purchase of 1 600 pair of shoes for the needs of the prison police members, as given in the technical specification.

title: Purchase of 1 600 Uniforms for the Needs of the Prison Police Members

description: Purchase of 1 600 uniforms for the needs of the prison police members, as given in the technical specifications.

title: Purchase of Uniforms and Shoes for the Needs of Prison Police Members

description: Purchase of uniforms and shoes for the needs of prison police members.