title: duae.signed.pdf

title: data despre participant.signed.pdf

title: f3.3 formular informativ.semnat.pdf

description: Formular informativ despre participant

title: gb 21013214.semnat.pdf

description: Garantie Bancara

title: duae lp 21013214.semnat.pdf

description: DUAE

title: anexa licenta0001.semnat.pdf

description: Anexa la licenta

title: licenta import0001.semnat.pdf

description: Licenta

title: certificat de inregistrare.semnat.pdf

description: Certificat de inregistrare

title: declaratii lp.semnat.pdf

title: formular informativ.semnat.pdf

title: duae.semnat.pdf

title: raport financiar.semnat.pdf

title: extras registru.semnat.pdf

title: cerificat fisc.semnat.pdf

title: oferta teh. lot 3.signed.pdf

title: anexa teh. lot 3.signed.pdf

title: declarație new items.signed.pdf

title: formular informativ despre ofertant.signed.pdf

title: declarație notificare.signed.pdf

title: duae.signed.pdf

title: 2021-02-14-sfj-certificate-iso13485-(gb).signed.pdf

title: declaration of conformity tissue-tek prisma plus (gb).signed.pdf

title: declaratie p. 8.signed.pdf

title: garantie-bancara-scm1.signed.pdf

title: operating manual tissue-tek prisma plus (gb).signed.pdf

title: prisma.signed.pdf

title: 2022-05-07-sfa-iso-certificate-13485.signed.pdf

title: iso 13485_en.semnat.pdf

title: declaratie.semnat.pdf

title: imputernicire.pdf

title: garantie nr.1.docx

title: f3.1.semnat.pdf

title: duae (16) (4).semnat.pdf

title: ce certificate_en.semnat.pdf

title: oferta.semnat.pdf

title: lot 10 - ventilator pulmonar neonatal cu frecventa inalta.semnat.pdf

title: datasheet_aura-v-neo.semnat.pdf

title: f3.3.semnat.pdf

title: duae.signed.pdf

title: data despre participant.signed.pdf

title: garantie-bancara-scm1.signed.pdf

title: duae.signed.pdf

title: declaratie p. 8.signed.pdf

title: oferta teh. lot 1.signed.pdf

title: declarație notificare.signed.pdf

title: operating manual histo-tek vp1 (gb).signed.pdf

title: declaration of conformity histo-tek vp1 (gb).signed.pdf

title: declarație new items.signed.pdf

title: anexa la sp. tehnica lot 1.signed.pdf

title: brochure histo-tek vp1 (gb).signed.pdf

title: specificatie-item1-histoprocesor.signed.pdf

title: f4.2.semnat.pdf

title: duae lp 21013214.semnat.pdf

description: DUAE

title: f4.1.semnat.pdf

title: axcent_broschuere_patient-monitoring_2019_01_v01_web.pdf

description: Specificatii tehnice (brosura)

documentType: illustration

title: licenta import0001.semnat.pdf

description: Licenta

title: konformitätserklärung cetus x15 1.2.pdf

description: Declaratie de Conformitate CE

title: gb 21013214.semnat.pdf

description: Garantie Bancara

title: declaratie privind deservirea si timpul de reactie.semnat.pdf

description: Declaratie

title: certificat de inregistrare.semnat.pdf

description: Certificat de inregistrare

title: f3.3 formular informativ.semnat.pdf

description: Formular informativ despre participant

title: anexa nr. 11.semnat.pdf

description: Anexa nr.11.  Confirmarea parametrilor tehnici solicitati

title: anexa licenta0001.semnat.pdf

description: Anexa la licenta

title: documente.semnat.pdf

title: documente.semnat.pdf

title: declaratie.semnat.pdf

title: imputernicire.pdf

title: duae (16) (4).semnat.pdf

title: declaratie pe propria raspundere i.semnat.pdf

title: declaratie de participant 10.semnat.pdf

title: fda free sale certificate_exp aug 2021.semnat.pdf

title: omni3_043-150-0001_r1.semnat.pdf

title: infinium iso 13485 certificate - copy of original.semnat.pdf

title: extras amed.md.semnat.pdf

title: 2. formarea pretului f4.2.signed.pdf

title: 8. mgf_declaratie _transportare&amp;amp;training.signed.pdf

title: 4. ce_e.signed.pdf

title: 8. babyguard_u-1131_operator&amp;apos;s manual  phototerapy.signed.pdf

title: 8. 058-80manro02101 (btl-4000 smart&amp;amp;premium).signed.pdf

title: 8. instilar 1488_infusion pump_operation manual.signed.pdf

title: 11. garantia pentru oferta_scmnr1.signed.pdf

title: 8. babyguard_i-1120 - operator&amp;apos;s manual incubator nenatal.signed.pdf

title: 9. declaratie mgf conform cerintelor tehnice.signed.pdf

title: 8. instilar 1428 s - user manual seringe pump.signed.pdf

title: 3. duae_mgf.signed.pdf

title: 6. 1427.190312_iso_certificate_2019.signed.pdf

title: 6. iso 13485_e.signed.pdf

title: 2. formularul ofertei.signed.pdf

title: 2. formarea producator f4.1.signed.pdf

title: 1. date despre ofertant f3.3.signed.pdf

title: 7. dixion_eng_broshure.signed.pdf

title: 4. 058-85cecertificate_10414-2017_btl-4000_smart-premium_btl-vacii_104.signed.pdf

title: 10. mgf declaratie mentenanta.signed.pdf

title: 5. mgf declaratie_amed.signed.pdf

title: 8. storm_5500_5800_v1.4 pacient monitos.signed.pdf

title: licenta import0001.semnat.pdf

description: Licenta

title: duae lp 21013214.semnat.pdf

description: DUAE

title: certificat de inregistrare.semnat.pdf

description: Certificat de inregistrare

title: gb 21013214.semnat.pdf

description: Garantie Bancara

title: anexa licenta0001.semnat.pdf

description: Anexa la licenta

title: f3.3 formular informativ.semnat.pdf

description: Formular informativ despre participant

title: duaesemnat.pdf

description: DUAE

title: garantiabancarascm1semnat.pdf

description: Garantia ofertei

title: certificat de inregistrare.semnat.pdf

description: Certificat de inregistrare

title: slee_iso certificate 9001-2008_englisch_2015-2018.pdf

description: ISO 9001

title: slee_iso_13485_certificate.pdf

description: ISO 13485

title: anexa nr. 4.semnat.pdf

description: Anexa nr. 4. Confirmarea specificatiilor tehnice solicitate

title: f4.1.semnat.pdf

title: f3.3 formular informativ.semnat.pdf

description: Formular informativ despre participant

title: anexa licenta0001.semnat.pdf

description: Anexa la licenta

title: 20190304_ce_declaration_newest_2019_010.00.pdf

description: Declaratie de Conformitate CE

description: Declaratie

title: gb 21013214.semnat.pdf

description: Garantie Bancara

title: licenta import0001.semnat.pdf

description: Licenta

title: f4.2.semnat.pdf

title: cut4062_flyer_eng_2017_07_001.00.pdf

description: Specificatii tehnice (brosura)

title: duae lp 21013214.semnat.pdf

description: DUAE

title: certificat de inregistrare.semnat.pdf

description: Certificat de inregistrare

title: gb 21013214.semnat.pdf

description: Garantie Bancara

title: licenta import0001.semnat.pdf

description: Licenta

description: Specificatii tehnice (brosura)

title: duae lp 21013214.semnat.pdf

description: DUAE

title: anexa licenta0001.semnat.pdf

description: Anexa la licenta

title: f3.3 formular informativ.semnat.pdf

description: Formular informativ despre participant

title: f4.1.semnat.pdf

title: anexa nr.  18.semnat.pdf

description: Anexa nr. 18.  Confirmarea parametrilor tehnici solicitati

title: f4.2.semnat.pdf

description: Declaratie

title: lab8715a_product_information_anesthetic_gas_monitoring_or.pdf

description: Specificatii tehnice (brosura)

title: konformitätserklärung cetus x15 1.2.pdf

description: Declaratie de Conformitate CE

title: extras nou 2019.semnat.pdf

title: catalog_bistos_bt-500, 550, 400_nicu.semnat.pdf

title: f3.3 date despre participant.semnat.pdf

title: garantie banca.semnat.pdf

title: iso incubator.pdf

title: ce_1984-mdd-19-560 infinium ek ii.3 en.semnat.pdf

title: duae.semnat.pdf

title: f 4.1 nr.1.semnat.pdf

title: declaratie de participant 10.semnat.pdf

title: fișiere (12).zip

title: f 4.2 pret nr. 1.semnat.pdf

title: decl înreg amdm.pdf

title: date participant medist grup srl.pdf

title: decl inspecții planificate.pdf

title: op bancă semnat.pdf

title: f 4.1 semnat.pdf

title: duae medist srl românia.pdf

title: duae medist grup.pdf

title: decl echipamente neutilizate.pdf

title: oferta.semnat.pdf

title: incubator semnat.semnat.pdf

title: oferta.semnat.pdf

title: specificatia tehnica.semnat.pdf

title: anexa licenta0001.semnat.pdf

description: Anexa la licenta

title: gb 21013214.semnat.pdf

description: Garantie Bancara

title: f3.3 formular informativ.semnat.pdf

description: Formular informativ despre participant

title: certificat de inregistrare.semnat.pdf

description: Certificat de inregistrare

title: licenta import0001.semnat.pdf

description: Licenta

title: duae lp 21013214.semnat.pdf

description: DUAE

title: 5. mgf declaratie_amed.signed.pdf

title: 3. duae_mgf.signed.pdf

title: 2. formarea producator f4.1.signed.pdf

title: 6. iso 13485_e.signed.pdf

title: 1. date despre ofertant f3.3.signed.pdf

title: 4. ce_e.signed.pdf

title: 2. formularul ofertei.signed.pdf

title: 7. dixion_eng_broshure.signed.pdf

title: 2. formarea pretului f4.2.signed.pdf

title: babylog vn500_product_information_en.signed.pdf

title: duae.signed.pdf

title: babylog vn500_brochure_en.signed.pdf

title: duae medist grup.pdf

title: date participant medist grup srl.pdf

title: decl înreg amdm.pdf

title: decl echipamente neutilizate.pdf

title: decl inspecții planificate.pdf

title: op bancă semnat.pdf

title: duae medist srl românia.pdf

title: f 4.1 semnat.pdf

title: f4.1.semnat.pdf

title: f3.3 formular informativ.semnat.pdf

description: Formular informativ despre participant

title: anexa licenta0001.semnat.pdf

description: Anexa la licenta

title: duae lp 21013214.semnat.pdf

description: DUAE

description: Declaratie

title: gb 21013214.semnat.pdf

description: Garantie Bancara

title: 20_180_ba410e_emc_ce_1809_en.pdf

description: Declaratie de Conformitate CE

title: f4.2.semnat.pdf

title: licenta import0001.semnat.pdf

description: Licenta

title: anexa nr. 6.semnat.pdf

description: Anexa nr. 6. Confirmarea parametrilor tehnici solicitati

title: 20_180_ba410e_di_en.pdf

description: Specificatii tehnice (brosura)

title: certificat de inregistrare.semnat.pdf

description: Certificat de inregistrare

title: 20_180_ba410e_trino_50w_sp_en.pdf

description: Specificatii tehnice (brosura)

title: 0_00_iso13485_ce_2207_en.pdf

description: ISO 13485

title: 30_190_moticam_1080_sp_en.pdf

description: Specificatii tehnice (brosura)

title: 20_180_ba410e_lvd_ce_1809_en.pdf

description: Declaratie de Conformitate CE

title: accesorii adăugătoare incluse in oferta pentru motic ba410 e trinocular.semnat.pdf

description: Accesorii adaugatoare incluse

title: declaratie mentenanță.semnat.pdf

title: registru de stat a dm_ventilator sle.png

title: garantie scm1 11.2019.pdf

title: duae_completat_2019.semnat.pdf

title: f4.1.semnat.pdf

title: raf10 issue 11 - iso 13485 (issued 2019-01-16. expiry 2020-04-08 (bsi md689407).pdf

title: f3.1.semnat.pdf

title: anexa 1 ventilator.semnat.pdf

title: formular informativ f 3.7.semnat.pdf

title: sle6000 datasheet - v2.0.40 specifications - english.pdf

title: sle6000 brochure - english.pdf

title: declaratie neutilizat_corespundere.semnat.pdf

title: f4.2_nou.semnat.pdf

title: raf12 issue 12 - directive 93.42.eec ec certificate - issued 18 feb 2019. exp 08 april 2022 (bsi certificate ce688636).pdf

title: declaratie p. 8.signed.pdf

title: duae.signed.pdf

title: declarație notificare.signed.pdf

title: anexa tehnica lot 2.signed.pdf

title: garantie-bancara-scm1.signed.pdf

title: tek6.signed.pdf

title: declaration of conformity tissue-tek tec 6 (gb).signed.pdf

title: declarație new items.signed.pdf

title: brochure tissue-tek tec 6 (gb).signed.pdf

title: oferta teh. lot 2.signed.pdf

title: duae.semnat.pdf

title: garantie scm1 11.2019.pdf

title: registru de stat a dm_mindray_monitor t6.png

title: f4.1.semnat.pdf

title: iso 13485_mindray_2020.pdf

title: f3.1.semnat.pdf

title: formular informativ f 3.7.semnat.pdf

title: declaratie mentenanță.semnat.pdf

title: beneview_t5-t9_ce_20150804.pdf

title: ce_universal_(20150222-20200221).pdf

title: beneview_t9t8t6t5_datasheet_eng_20170307.pdf

title: duae_completat_2019.semnat.pdf

title: 01.t6_brochure_eng_20190516.pdf

title: f4.2_nou.semnat.pdf

title: anexa 4 monitor-gaze anestezice.semnat.pdf

title: garantia.semnat (1).pdf

title: oferta part 2.semnat.pdf

title: oferta semnat part 1.pdf

title: data despre participant.signed.pdf

title: duae.signed.pdf

title: licenta import0001.semnat.pdf

description: Licenta

title: duae lp 21013214.semnat.pdf

description: DUAE

title: anexa licenta0001.semnat.pdf

description: Anexa la licenta

title: certificat de inregistrare.semnat.pdf

description: Certificat de inregistrare

title: gb 21013214.semnat.pdf

description: Garantie Bancara

title: f3.3 formular informativ.semnat.pdf

description: Formular informativ despre participant

title: euromex iscope.semnat.pdf

title: extras.semnat.pdf

title: f 3.1.semnat.pdf

title: sz 4600759 01 get.eng cert iso 9001.semnat.pdf

title: declaratie.semnat.pdf

title: f 4.2.semnat.pdf

title: duae.semnat.pdf

title: declaratie1.semnat.pdf

title: f 3.3.semnat.pdf

title: f 4.1.semnat.pdf

title: garantia.semnat.pdf

title: decl echipamente neutilizate.pdf

title: op bancă semnat.pdf

title: f 4.1 semnat.pdf

title: decl inspecții planificate.pdf

title: decl înreg amdm.pdf

title: duae medist srl românia.pdf

title: date participant medist grup srl.pdf

title: duae medist grup.pdf

title: 2. formularul ofertei.signed.pdf

title: 7. dixion_eng_broshure.signed.pdf

title: 3. duae_mgf.signed.pdf

title: 4. ce_e.signed.pdf

title: 5. mgf declaratie_amed.signed.pdf

title: 6. iso 13485_e.signed.pdf

title: 2. formarea producator f4.1.signed.pdf

title: 1. date despre ofertant f3.3.signed.pdf

title: 2. formarea pretului f4.2.signed.pdf

title: oferta part 2.semnat.pdf

title: garantia.semnat (1).pdf

title: garantiaofertasemnat.pdf

description: GARANTIA OFERTA.semnat.PDF

title: duaesemnat.pdf

description: DUAE.semnat.PDF

title: licenta import0001.semnat.pdf

description: Licenta

title: f3.3 formular informativ.semnat.pdf

description: Formular informativ despre participant

title: gb 21013214.semnat.pdf

description: Garantie Bancara

title: duae lp 21013214.semnat.pdf

description: DUAE

title: anexa licenta0001.semnat.pdf

description: Anexa la licenta

title: certificat de inregistrare.semnat.pdf

description: Certificat de inregistrare

title: duae.signed.pdf

title: f4.2.signed.pdf

title: garantie pentru oferta.signed.pdf

title: nicu (bt-500 550 400)_rev.02.signed.pdf

title: bt-500 ce.signed.pdf

title: anexa nr. 3 la f4.1.signed.pdf

title: data despre participant.signed.pdf

title: declaratie.signed.pdf

title: anexa nr. 2 la f4.1.signed.pdf

title: f3.1.signed.pdf

title: f4.1.signed.pdf

title: iso13485.signed.pdf

title: bt-300, bt-350, bt-400 ce.signed.pdf

title: anexa nr. 1 la f4.1.signed.pdf

title: declaratie 2.semnat.pdf

title: compactplus-timetocare brochure.semnat.pdf

title: duae.semnat.pdf

title: iso.semnat.pdf

title: date despre participant.semnat.pdf

title: perfusor compact plus inst ru.semnat.pdf

title: f4,1.semnat.pdf

title: 38932201_infusomat compactplus_gb_ifu_0318.semnat.pdf

title: cp gen reg amed.semnat.pdf

title: f4,2.semnat.pdf

title: garantia bancara.semnat.pdf

title: declaratie.semnat.pdf

title: g2s 18 04 12974 457_bbm de_e.semnat.pdf

title: declaratie 2.semnat.pdf

title: iso.semnat.pdf

title: cp gen reg amed.semnat.pdf

title: g2s 18 04 12974 457_bbm de_e.semnat.pdf

title: f4,1.semnat.pdf

title: f4,2.semnat.pdf

title: duae.semnat.pdf

title: date despre participant.semnat.pdf

title: declaratie.semnat.pdf

title: garantia bancara.semnat.pdf

title: f4.2.signed.pdf

title: f4.1.signed.pdf

title: doc oly-ce anexa ii si v.signed.pdf

title: garantie pentru oferta.signed.pdf

title: anexa nr. 2 la f4.1.signed.pdf

title: iso 13485 medical syst corp oly-en-ro.signed.pdf

title: anexa nr. 1 la f4.1.signed.pdf

title: uhi-4-high-flow-insufflation-unit_brochure_de_5177.signed.pdf

title: f3.1.signed.pdf

title: duae.signed.pdf

title: data despre participant.signed.pdf

title: anexa nr. 3 la f4.1.signed.pdf

title: declaratie.signed.pdf

title: oferta financiara_tehnica_garantie.semnat.pdf

title: documente obligatorii solicitate.semnat.pdf

title: data despre participant.signed.pdf

title: duae.signed.pdf

title: data despre participant.signed.pdf

title: duae.signed.pdf

title: oferta semnat part 1.pdf

title: garantia.semnat (1).pdf

title: oferta part 2.semnat.pdf

title: f4.1 sp1.semnat.pdf

title: date participant n1 converted.semnat.pdf

title: duae spn1 converted.semnat.pdf

title: vulcan3.semnat.pdf

title: inregistrarea.semnat.pdf

title: inspectie.semnat.pdf

title: okn112.semnat.pdf

title: declaratie vulcan(3).semnat.pdf

title: inspectie.semnat.pdf

title: scolarizarea.semnat.pdf

title: duae spn1 converted.semnat.pdf

title: okn113.semnat.pdf

title: gar sp1.semnat.pdf

title: specifica+гii de pret f4.2.semnat.pdf

title: neutilizat.semnat.pdf

title: gar sp1.semnat.pdf

title: f4.1 sp1.semnat.pdf

title: specifica+гii de pret f4.2.semnat.pdf

title: inregistrarea.semnat.pdf

title: okn111.semnat.pdf

title: scolarizarea.semnat.pdf

title: neutilizat.semnat.pdf

title: ru 1 vulcan.semnat.pdf

title: ru vulcan3.semnat.pdf

title: okn11.semnat.pdf

title: oferta teh. lot 4.signed.pdf

title: anexa teh. lot 4.signed.pdf

title: declarație new items.signed.pdf

title: operating manual srm 200 (gb).signed.pdf

title: declaration of conformity accu-cut srm 200 (gb).signed.pdf

title: brochure accu-cut srm 200 (gb).signed.pdf

title: declaratie p. 8.signed.pdf

title: duae.signed.pdf

title: declarație notificare.signed.pdf

title: garantie-bancara-scm1.signed.pdf

title: 1. date despre ofertant f3.3.signed.pdf

title: 7. dixion_eng_broshure.signed.pdf

title: 2. formarea pretului f4.2.signed.pdf

title: 2. formarea producator f4.1.signed.pdf

title: 2. formularul ofertei.signed.pdf

title: 4. ce_e.signed.pdf

title: 3. duae_mgf.signed.pdf

title: 6. iso 13485_e.signed.pdf

title: 5. mgf declaratie_amed.signed.pdf

title: f 3.1.semnat (34).pdf

title: duae .semnat (12).pdf

title: prospect.semnat.pdf

title: f 4.1 si 4.2.semnat (8).pdf

title: iso q5 073283 0044 rev. 00.semnat.pdf

title: cert inreg, cont banc, extras, licenta.semnat.pdf

title: ce g2 073283 0046 rev. 00.semnat.pdf

title: transfer bancar.semnat (20).pdf

title: registru de stat.semnat.pdf

title: f. 3.3  3.4  3.5.semnat (1).pdf

title: inregistrarea.semnat.pdf

title: gar sp1.semnat.pdf

title: f4.1 sp1.semnat.pdf

title: certificat  iso 9001-2009.semnat.pdf

title: duae spn1 converted.semnat.pdf

title: gar sp1.semnat.pdf

title: specifica+гii de pret f4.2.semnat.pdf

title: neutilizat.semnat.pdf

title: scolarizarea.semnat.pdf

title: scolarizarea.semnat.pdf

title: radius01.semnat.pdf

title: ru vulcan3.semnat.pdf

title: inregistrarea.semnat.pdf

title: ru 1 vulcan.semnat.pdf

title: declaratie vulcan(3).semnat.pdf

title: f4.1 sp1.semnat.pdf

title: duae spn1 converted.semnat.pdf

title: neutilizat.semnat.pdf

title: inspectie.semnat.pdf

title: vulcan3.semnat.pdf

title: inspectie.semnat.pdf

title: specifica+гii de pret f4.2.semnat.pdf

title: cerinrbelar.semnat.pdf

title: 2. formarea pretului f4.2.signed.pdf

title: 5. mgf declaratie_amed.signed.pdf

title: 6. iso 13485_e.signed.pdf

title: 4. ce_e.signed.pdf

title: 3. duae_mgf.signed.pdf

title: 7. dixion_eng_broshure.signed.pdf

title: 2. formarea producator f4.1.signed.pdf

title: 1. date despre ofertant f3.3.signed.pdf

title: 2. formularul ofertei.signed.pdf

title: oferta part 2.semnat.pdf

title: garantia.semnat (1).pdf

title: data despre participant.signed.pdf

title: duae.signed.pdf

title: gb 21013214.semnat.pdf

description: Garantie Bancara

title: duae lp 21013214.semnat.pdf

description: DUAE

title: konformitätserklärung dermatome, mucotome.pdf

description: Declaratie de Conformitate CE

description: Declaratie

title: certificat de inregistrare.semnat.pdf

description: Certificat de inregistrare

title: licenta import0001.semnat.pdf

description: Licenta

title: f4.1.semnat.pdf

title: ce certificate with annex.pdf

description: CE certificat

title: f4.2.semnat.pdf

title: anexa licenta0001.semnat.pdf

description: Anexa la licenta

title: konformitätserklärung tcm 3000 bl.pdf

description: Declaratie de Conformitate CE

title: 67143-en_skin grafting_v3918_view.pdf

description: Specificatii tehnice (brosura)

title: zertifikate iso 13485_en_opt.pdf

description: ISO 13485

title: f3.3 formular informativ.semnat.pdf

description: Formular informativ despre participant

title: anexa  nr. 8.semnat.pdf

description: Anexa nr. 8.  Confirmarea parametrilor tehnici solicitati

title: certificat de inregistrare.semnat.pdf

description: Certificat de inregistrare

title: duae lp 21013214.semnat.pdf

description: DUAE

title: licenta import0001.semnat.pdf

description: Licenta

title: gb 21013214.semnat.pdf

description: Garantie Bancara

title: f3.3 formular informativ.semnat.pdf

description: Formular informativ despre participant

title: anexa licenta0001.semnat.pdf

description: Anexa la licenta

title: oferta part 2.semnat.pdf

title: garantia.semnat (1).pdf

title: data despre participant.signed.pdf

title: duae.signed.pdf

title: becor declaratie mentenanta.semnat.pdf

title: iso optica.semnat.pdf

title: optika_microscopios_serie_b-500_2017.semnat.pdf

title: ce optica.semnat.pdf

title: becor f 3.3p (2).semnat.pdf

title: becor f4.1.semnat.pdf

title: becor gb.semnat.pdf

title: becor f3.3 p1 (1).semnat.pdf

title: becor declaratie.semnat.pdf

title: duae.semnat.pdf

title: becor f3.1.semnat.pdf

title: doc confirmativ microtom.semnat.pdf

title: becor f4.1 p2.semnat.pdf

title: becor f 4.2.semnat.pdf

title: duae lp 21013214.semnat.pdf

description: DUAE

title: iso 9001 myr.pdf

description: ISO 9001

title: gb 21013214.semnat.pdf

description: Garantie Bancara

title: ec declaration of conformity - ec500.pdf

description: Declaratie de Conformitate CE

title: anexa nr. 2.semnat.pdf

description: Anexa nr. 2. Confirmarea specificatiilor tehnice solicitate

title: f3.3 formular informativ.semnat.pdf

description: Formular informativ despre participant

description: Declaratie

title: anexa licenta0001.semnat.pdf

description: Anexa la licenta

title: ec500_gb_redux(1).pdf

description: specificatii tehnice (brosura)

title: iso 13485 myr.pdf

description: ISO 13485

title: f4.2.semnat.pdf

title: licenta import0001.semnat.pdf

description: Licenta

title: f4.1.semnat.pdf

title: certificat de inregistrare.semnat.pdf

description: Certificat de inregistrare

title: gar sp1.semnat.pdf

title: magniter.semnat.pdf

title: f4.1 sp1.semnat.pdf

title: f4.1 sp1.semnat.pdf

title: inregistrarea.semnat.pdf

title: inregistrarea.semnat.pdf

title: decl amnp (1).semnat.pdf

title: specifica+гii de pret f4.2.semnat.pdf

title: scolarizarea.semnat.pdf

title: vulcan3.semnat.pdf

title: specifica+гii de pret f4.2.semnat.pdf

title: inspectie.semnat.pdf

title: duae spn1 converted.semnat.pdf

title: scolarizarea.semnat.pdf

title: neutilizat.semnat.pdf

title: gar sp1.semnat.pdf

title: inspectie.semnat.pdf

title: duae spn1 converted.semnat.pdf

title: neutilizat.semnat.pdf

title: ru 1 vulcan.semnat.pdf

title: declaratie vulcan(3).semnat.pdf

title: ru vulcan3.semnat.pdf

title: garantia.semnat (1).pdf

title: oferta part 2.semnat.pdf

title: garantia.semnat (1).pdf

title: oferta part 2.semnat.pdf

title: garantiabancarascm1semnat.pdf

description: Garantie

title: duaesemnat.pdf

title: specificatia tehnica.semnat.pdf

title: gb 21013214.semnat.pdf

description: Garantie Bancara

title: duae lp 21013214.semnat.pdf

description: DUAE

title: licenta import0001.semnat.pdf

description: Licenta

title: certificat de inregistrare.semnat.pdf

description: Certificat de inregistrare

title: anexa licenta0001.semnat.pdf

description: Anexa la licenta

title: f3.3 formular informativ.semnat.pdf

description: Formular informativ despre participant

title: declaratie.semnat.pdf

title: imputernicire.pdf

title: duae (16) (4).semnat.pdf

title: !documente_de_calificare.signed.pdf

title: !duae - biosistem.signed.pdf

title: !_oferta_biosistem_f3.1_f3.3_f3.4_f3.5_f4.1_f4.2_scm_nr1.signed.pdf

title: declaratie.semnat.pdf

title: duae (16) (4).semnat.pdf

title: imputernicire.pdf

title: 4. ce_e.signed.pdf

title: 5. mgf declaratie_amed.signed.pdf

title: 2. formarea producator f4.1.signed.pdf

title: 2. formarea pretului f4.2.signed.pdf

title: 3. duae_mgf.signed.pdf

title: 7. dixion_eng_broshure.signed.pdf

title: 2. formularul ofertei.signed.pdf

title: 1. date despre ofertant f3.3.signed.pdf

title: 6. iso 13485_e.signed.pdf

title: oferta part 2.semnat.pdf

title: oferta semnat part 1.pdf

title: garantia.semnat (1).pdf

title: garantiabancarascm1semnat.pdf

description: Garantie Bancara

title: duaesemnat.pdf

description: DUAE

title: garantia.semnat (1).pdf

title: oferta part 2.semnat.pdf

title: Dispozitive medicale conform necesităților IMSP Spitalul Clinic Municipal nr. 1 pentru anul 2019

description: Alte temeiuri de respingere a ofertei

description: Alte temeiuri de respingere a ofertei

description: Alte temeiuri de respingere a ofertei

description: Alte temeiuri de respingere a ofertei

description: Alte temeiuri de respingere a ofertei

description: Alte temeiuri de respingere a ofertei

title: Colorator de lamele

description:
Configurația
Amplasare pe masă
Tip suprafață
Anticorozivă
Mecanismul de translație a suportului cu lamele
Sistem cu transfer robotic
Sistemul de colorare
Numărul de containere
Minim 40 containere
Volumul containerului
Minim 310 ml
Numărul de lamele pe rack
Minim 20 de lamele
Capacitate totală de colorare pe oră
Minim 280 de lamele
Uscător de lamele integrat
Da
Managementul proceselor
Posibilitate de modificare a programelor de colorare
Da
Numărul de programe de colorare concomitent
Minimum 10 programe
Posibilitate de întrerupere manuală a procesului de colorare
Da
Funcție de eficientizare a proceselor
Da
Interfața cu utilizatorul
Ecran tactil încorporat
Funcționalități speciale
Sistem de filtrare a vaporilor toxici
Prin filtru de cărbune
Existența unui sistem automat de montare a lamelelor pe lamă compatibil
Da
Sursa de alimentare
Tensiune de alimentare
220 V, 50 Hz
Cablu de alimentare
Tip Schuko
Consumabile și accesorii
Rack pentru lame (26x76 mm ) de multiplă utilizare
10 bucăți
Chit de conectare la sursa de apă
Da
Chit de conectare la sursa de canalizare
Da
Cerințe față de producător și furnizor
Garanție
Minimum 3 ani
Certificate
CE certificat european
Standarte
ISO 9001, ISO 13485
Logistică
Manuale de utilizare
Da, în limba română sau rusă
Manual de service
Da, engleză
Transportare la locul necesar
Da
Instalare
Da
Testare
Da
Training pentru utilizatori
Da, pregătire de lucru, mod de utilizare a dispozitivului, întreținerea zilnică
Training pentru personal tehnic
Ajustarea, calibrarea, înlăturarea defecțiunilor minore, parole de acces
Deservire în perioada de garanție
Da

title: Unitate de fototerapie cu raze ultraviolete

description:
Iradiator UV portativ
Tensiunea rețelei electrice: 220V
Frecvența: 50 Hz
Puterea de consum: max 1000 VA
Lampa- sursa de radiație:  DRT – 240 sau echivalent (plus una în complet, de rezervă)
Dimensiuni exterioare nu mai mare 215x120x1700 mm
Greutatea: max 1,2 kg

title: Dispozitiv fizioterapeutic pentru amplipuls terapie

description:
Dispozitiv de fizioterapie cu curenți de joasă și medie frecvență
Forme de curenti:  galvanici, diadinamici, sinusoidali modulați
Ecran LED
Taimer încorporat
Control automat al curentului pacientului
Tensiunea rețelei electrice: 220V
Frecvența: 50 Hz
Puterea: max 40 VA
Clasa de electrosecuritate: II

title: Insuflator CO2

description:
Debit de insuflare- 20-35 l
Evacuare automată a fumului
Circuit de evacuare fum separat față de cel de insuflare
Afișarea parametrilor presetați (debit, presiune, gaz consumat)
Reglare debit de insuflare
Domeniu debit de insuflare de la 0,5 pînă la 35 l/min.
Reglare continuă
Tub de insuflare reutilizabil din cauciuc siliconic
Alarmă la lipsa gazului de insuflare
Taste de comandă cu protecție la umezeală
Furtun de conectare la butelia de înaltă presiune
Să permită alimentarea cu gaz atît de la tub de înaltă presiune direct, fără reglator, cît și de la sistemul central de gaz al spitalului, de joasă presiune

title: Electrodermatom

description:
Modul de alimentare:
1.	Tensiunea de alimentare de intrare 220V, 50 Hz
2.	Protecția la scurtcircuit, supratensiune
3.	Protecție la suprasarcină
4.	Buton de pornire/oprire
Caracteristicile tehnice dermatom:
1.	Autoclavabil
2.	Buton de ajustare viteză taiere, cu blocare de siguranță
3.	Posibilitatea de ajustare a lățimii
4.	Grosimea lamboului prelevant 0,1-1,0 mm
5.	Lățimea lamboului prelevat 40 mm-100  mm
6.	Mod de funcționare: oscilatoriu
Accesorii și consumabile:
1.	Lame de schimb, oțel inoxidabil, 10 buc
2.	Container specializat pentru sterilității, 1 buc
3.	Cos pentru plasarea dispozitivului pentru spălare și dezinfectare cu capac, 1 buc.

title: Microscoape pentru examinări în histopatologie

description:
Ocular
Tipul ocularului
Binocular
Amplificare ocular
Minimum 10x
Număr de cîmp al ocularului
Minimum 22
Tip ocular
Cu ajustare dioptrică
Compensare ametropie
De la +5 la -5
Distanța interpupilară
Ajustabilă
Distanța interpupilară
45 -75
Interval unghi de înclinare a tubului de observare
30- 45˚
Turelă rotativă și obiective
Număr de obiective minim incluse
5
Tip obiectiv
Plan Acromat
Obiectiv necesar
4x
Numărul aperturii
Minimum 0,1
Distanța de lucru
Minimum 18
Obiectiv necesar
10x
Numărul aperturii
Minimum 0,25
Distanța de lucru
Minimum 10,5
Obiectiv necesar
20x
Numărul aperturii
Minimum 0,4
Distanța de lucru
Minimum 0,92
Obiectiv necesar
40x
Numărul aperturii
Minimum 0,65
Distanța de lucru
Minimum 0,36
Obiectiv necesar
60x/100x draft
Numărul aperturii
Minimum 0,7
Distanța de lucru
Minimum 0,2
Număr de cîmp
Minimum 22
Butoane de reglaj
Axe de reglare a suportului
X, Y, Z
Mod de acționare a suportului
Rapidă și fină
Treaptă de reglare fină a focusării
Maximum 1 µm
Iluminator
Sursă de lumină
LED
Durata de viață a sursei de lumină
Minimum 50000 ore
Mod iluminare
Prin transmitere (din partea inferioară a suportului)
Ajustare luminozitate
Da
Platformă de lucru
Cu cleme de fixare
Da
O clemă mobilă
Da
Diafragmă și condensor
Tip condensor
Acromatic și aplanatic
Tip diafragmă
Cu diametru ajustabil
Interval de ajustare a diafragmei
Pentru intervalul 4x-100x
Accesorii
Husă de protecție
1 bucată
Sursă de alimentare
Tensiune de alimentare
220V, 50 Hz
Tip fir de alimentare
Shuko
Standarte și certificate
Certificat
CE ( Certificat european)
Standard
ISO 9001:2015 Standart de calitate a dispozitivelor
ISO 13485:2016 Standart de calitate a dispozitivelor medicale
Instruire
Training pentru utilizatori
Da
Module necesare:
Prezentare generală, pregătire de lucru, operarea cu dispozitivul, întreținere și dezinfectare
Training pentru personal tehnic
Da
Module necesare:
Modul de efectuare a mentenanței periodice, modul de testare a funcționalității
Documentație
Manualul de utilizare
română sau rusă
Manualul de utilizare
engleză
Manualul de service
engleză
Logistică
Transportare la adresa solicitată
Da
Instalare
Da
Testare
Da
Garanție
Perioada de garanție
Minim 4 ani
Efectuarea mentenanței în perioada de garanție de către furnizor
Conform recomandărilor producătorului

title: Monitor- analizator gaze anestezice

description:
 Monitor analizator gaze anestezice
Color minim 800x600
Diagonala minim 15"
Opțional funcționalitate touchscreen
 Parametri monitorizați continu:
ECG (minim 3 și 5 derivații), SPO2, IBP1, IBP2, RESP/CO2 (Freezable)
BIS,
 Parametri numerici
HR, SPO2, Puls
NIBP (SYS, DIA, MAP),
IBP1 (SYS, DIA, MAP),
IBP2 (SYS, DIA, MAP),
RR,
T1,
T2,
PPV, SVV
EtCo2, FiCo2,
RR,
EtO2, FiO2,
EtAA, FiAA, MAC
BIS
Aritmie
 Analiza automată a traseului EKG: VF, VT, RUN, AIVR, BGM, TGM, CPT, TACH, BRDY, PAUS, ASYS

Analiza ST

- Rezoluție 0.01 mV

- Interval măsurători de la -2mv la +2mv

-	Alarma de la -2mv la +2mv
 NIBP
Tensiunea arterială non-invaziv Metoda măsurare Oscillometric

Mod măsurare Manual/Automat
 SpO2
Interval minim de măsurare 40-100%

Acuratețe minim SpO2: 2%
Puls
Interval minim de măsurare 25-250 b/min
 Temperatura
Interval minim de măsurare 0-50ºC

Acuratețe minim: ± 0.2ºC
 Respirație
Interval minim de măsurare 0-150 R/min

Acuratețe minim: ± 0.2% sau 2 R/min
 IBP
2 canale
 Analizator gaze medicale
Metoda: absorbție infraroșu
Măsurători:
- CO2      0 ~ 10%
- O2      0 ~ 100%
- ISO   0 ~ 5%
- SEV      0 ~ 8%
Alarme sonore și vizuale
 mesaje eroare, limite parametri
Alimentare
 200-240 VAC, 50/60 Hz
Baterie internă
 utilizare minim 90 minute
Accesorii:
Cablu EKG
Proba Temp – adult
Manșetă adult tensiunea arterială non-invazivă
Cablu extesie SpO2
SPO2 Senzor deget
Analizator pentru Gaze: CO2, O2, agenți anestezici, AA ID
2 cabluri presiune invazivă
Cablu BIS
Stocare de date
Memorie interna mini pentru 72 ore
Comunicație:,
USB，DVI-D，RJ45
Manualul operatorului
Inclus
Limba de operare
Româna, engleză
CERTIFICARE
Certificat de marcaj european (CE Mark) sau declarație de conformitate a producătorului cu Directiva 93/42/EEC
ISO 9001 / ISO 13485 sau echivalent
GARANŢIE
Termen de garanție: minim 24 de luni de la data instalării și punerii în funcțiune a echipamentului
Timp maxim de intervenție post-garanție: 24 ore
SERVICII ASOCIATE INCLUSE:
Transportul până la sediul beneficiarului
Instalarea și punerea în funcțiune la sediul beneficiarului de către departamentul service al reprezentantului autorizat al producătorului.
Trainingul personalului medical și tehnic.

title: Incubator nou născuți

description:
Specificația tehnică:
Tipul pacientului: prematur, nou-născut
Tipul incubatorului: mobil pe roți cu frînă
Dulap pentru scutece și consumabile medicale integrat, min 3 buc.
Ajustarea electrică a înălțimii incubatorului pe vertical
Canopie rezistentă la raze UV, pentru utilizarea lămpilor de fototerapie
Pereții canopiei să fie dubli pentru circulația aerului (cu efect anti-aburire)
Posibilitate de deschidere a canopiei prin ridicare integrală, cu fixarea canopiei
Mînere de transportare pe amblele părți
Uși laterale (stinga, dreapta) de acces în interiorul canopiei min. 2 buc.
Uși de acces pentru mîini min. 4 buc., cîte 2 pe ambele părți a canopiei (față, spate)
Uși de acces pentru mîini cu garnitură din cauciuc detașabilă
Port de acces pentru tuburi respiratorii și cabluri min. 8 buc.
Manipularea poziției saltelei trendelenburg și antitrendelenburg prin comandă electronică
Unghiul de ajustare trendelenburg și antitrendelenburg  ≥ ± 12°
Tavă pentru amplasarea casetei roentgen
Pătucul nou-născutului să fie glisant pentru efectuarea investigațiilor x-ray
Saltea pentru nou-născut confecționată din materiale moi, rezistente la agenții chimici
Ecran intern color, LCD min. 5 inch
Ecran extern color, LCD min. 10 inch
Posibilitatea reglării unghiului de vizualizare a ecranului extern
Ecran extern afișare: temperatura aer, concentratia SpO2, temperatura copil, concentratie oxigen, umiditate, putere încălzitor
Modul de umidificare:
Umidificator controlat prin microprocesor
Modul de umidificare și rezervor de apă integrat
Modul de umidificare detașabil (la nivel de utilizator) pentru curățare
Posibilitate de ajustare a nivelului de umiditate 30 % - 90%
Acurateția umidității  ≤  ± 5% RH
Intervalul de măsurare a umidității 15 % - 99 %
Modul de control a temperaturii:
Regim de control „Aer”
Intervalul de temperatură în regim „Aer” 25 °C – 37 °C
Funcția de temperatură > 37°C
Regim de control „Temperatură Copil”
Intervalul de temperatură în regim „Temperatură Copil”, 34 °C – 37 °C
Min. 2 senzori de temperatură copil
Acurateția de măsurare a temperaturii copil  ≤ 0.2 °C
Modul de oxigenare:
Modul de O2 controlat prin microprocesor
Control concentrație oxigen 21%-60%
Măsurare concentrație oxigen 21%-99%
Acurateția de măsurare ≤ 3%
Min. 2 senzori de oxigen pentru control și monitorizare
Modul de SpO2:
Măsurare SpO2, 70-100%
Acurateția SpO2, ≤ 3%
Modul de supraveghere greutate:
Cîntar pentru nou-născut integrat
Rezoluția min.100g - 5000g
Grafice:
Vizualizarea graficilor umiditate, perioada 24 ore
Vizualizarea graficilor temperatură, perioada 24 ore
Vizualizarea graficilor oxigen, perioada 24 ore
Vizualizarea graficilor greutate, perioada 24 ore
Alarme:
Anulare alarmă fără atingere, pentru prevenirea infecțiilor
Buton dedicate anulare alarme
Alarme sonore și vizuale
Alarma temperatura aerului ridicată
Alarma temperatura aerului scăzută
Alarma temperatura pielii ridicată
Alarma temperatura pielii scăzută
Alarma lipsă apă în rezervor cu pictogramă pe ecran
Alarma plasare incorecta umidificator
Alarma defecțiune sistem
Alarma defecțiune ventilator
Alarmă flux aer
Alarma concentratie oxigen
Sursa de tensiune:
Tensiune de alimentare 220 V, 50 Hz
Consumabile și accesorii adiționale:
Filtru pentru aer 10 buc.
Saltea pentru nou-născuți 4 buc.
Senzor de temperatură pentru copil 4 buc.
Senzor SPO2 reutilizabil, nou-născut, 1 buc.
Cerințe adăugătoare:
Standarte și certificate Certificat CE ( Certificat european)
Standard ISO 13485
Manual de utilizare
Manual tehnic
Transportare la adresa solicitată
Instalare
Testare
Perioada de garanție minim 2 ani
Efectuarea mentenanței în perioada de garanție de către furnizor, conform recomandărilor
producătorului
Instruire pentru utilizatori
Instruire pentru personal tehnic

title: Sistem de impregnare (incluzionare) în parafină

description:
Amplasare
Pe masă
Mod de operare
Manual
Tip suprafață
Anticorozivă
Material de impregnare a probei
Parafină
Construcția dispozitivului
Modular
Sistemul de parafinare
Rezervor de parafină
Minimum 4 litri, cu ajustare a temperaturii

Interval de temperatură a rezervorului de parafină
50 – 70 °C
Compartiment de depozitare a matricilor
Da, cu ajustare a temperaturii
Interval de temperatură a compartimentului pentru matrici
50 – 70 °C
Compartiment de depozitare a casetelor cu țesut
Da, cu ajustare a temperaturii
Interval de temperatură a compartimentului pentru caste cu țesut
50 – 70 °C
Suprafața de lucru
Încălzită
Zonă de răcire a matricei
Da
Sursă de lumină
Da, integrată în dispensor
Mod de turnare
Cu pedală/buton funcțional comode
Ajustarea fluxului de parafină
Da
Interfața cu utilizator
Ecran
Sistemul de răcire a blocurilor de parafină
Modalitate de răcire
Automat sau ajustat de utilizator
Suprafața efectivă
Pentru minimum 60 blocuri
Temperatura minimă de răcire
Maximum -6 °C
Funcționalități speciale
Setarea timpului de lucru
Da
Setarea temperaturii
Da, pentru rezervorul de parafină, zona caldă a dispensorului, zona de răcire a blocurilor,  compartimentul de depozitare a casetelor cu țesut, compartimentul de depozitare a matricilor
Posibilitate de conectare a pencetelor încălzite electric
Da
Sursa de alimentare
Tensiunea de alimentare
220 V, 50 Hz
Cablu de alimentare
Tip Schuko
UPS
Minimum o oră de funcționare
Accesorii și consumabile
Parafină în granule
15 kg
Pencetă încălzită electric
1 buc
Pencete din metal
3 buc
Matrice din oțel inoxidabil para free, adîncime de 11-14 mm, dimensiuni laterale 22 x 22 mm ± 2 mm
20 buc
Matrice din oțel inoxidabil para free, adîncime de 11-14 mm, dimensiuni laterale 32 x 25 mm ± 2 mm
10 buc
Matrice din oțel inoxidabil para free, adîncime de 11-14 mm, dimensiuni laterale 38 x 25 mm ± 2 mm
20 buc
Pedală de comandă
1 buc
Rezervor pentru evacuarea reziduurilor de parafină din sistem
1 buc
Cerințe față de producător și furnizor
Garanție
Minimum 5 ani
Certificate
CE certificat european
Standarte
ISO 9001, ISO 13485
Logistică
Manuale de utilizare
Da, în limba română sau rusă
Manual de service
Da, engleză
Transportare la locul necesar
Da
Instalare
Da
Testare
Da
Training pentru utilizatori
Da, pregătire de lucru, mod de utilizare a dispozitivului, întreținerea zilnică
Training pentru personal tehnic
Ajustarea, calibrarea, înlăturarea defecțiunilor minore, parole de acces

title: Procesor automat de țesuturi

description:
Configurația
Amplasare pe masă/podea

Tip suprafață
Anticorozivă
Material de impregnare a probei
Parafină, pentru microscopia optică
Mod de procesare
Cu cameră staționară
Mod de umplere a camerei
Prin vacuum
Camera de prelucrare
Volumul containerelor cu soluții și ceară
Minimum 2,5 L

Cantitatea probelor procesate concomitent
Minimum 150-200
Etape de prelucrare
Minim 16
Managementul proceselor
Interfața cu utilizatorul
Ecran, touchscreen
Posibilitate de control al timpului de prelucrare
Da
Intervalul timpului de prelucrare
0-59 ore
Posibilitate de control al temperaturii soluțiilor
Da, Solvenți și parafină
Interval de temperatură a parafinei
45-70 °C
Managementul deșeurilor
Automat
Sistem de evacuare a vaporilor chimici
Mod de evacuare a vaporilor chimici
Fără conexiune la sistem de ventilare, cu filtru
Tip filtru
Din cărbune
Sursa de alimentare
Tensiune de alimentare
220 V, 50 Hz
Cablu de alimentare
Tip Schuko
UPS sau sursă de alimentare integrată
Minimum 1 oră de funcționare
Consumabile și accesorii
Filtre de cărbune
6 buc.
Coș pentru încarcarea probelor de țesut
2 buc
Soluție Formalin 4%
7,5 litri
Xilen
45 litri
Etanol 96,6 – 100 %
37 litri
Parafină solidă
15 kg
Cerințe față de producător și furnizor
Garanție
Minimum 3 ani
Certificate
CE certificat european
Standarte
ISO 9001, ISO 13485
Logistică
Manuale de utilizare
Da, în limba română sau rusă
Manual de service
Da, engleză
Transportare la locul necesar
Da
Instalare
Da
Testare
Da
Training pentru utilizatori
Da, pregătire de lucru, mod de utilizare a dispozitivului, întreținerea zilnică
Training pentru personal tehnic
Ajustarea, calibrarea, înlăturarea defecțiunilor minore, parole de acces
Deservire în perioada de garanție
Da

title: Pompă de infuzie (perfuzor), volum mare

description:
Pompă de infuzie (perfuzor), volum mare - 5 unități
Descriere	Acest grup de produse înclude pompe de infuzie pentru volume mari cu cerințe avansate; pot avea 2 sau mai multe canale. Fluxul este calibrat în ml/oră.  Au posibilitatea de a efectua calcule a raportului medicament/doză, ce permite programarea fluxului reeşind direct din indicaţiile medicului.
Parametrul	Specificația
Display	Date afișate	Alarme
		Rata
		Volum infuzat
		Timpul infuzat
		Rata KVO
		Nivelul de încărcare a beteriei
		Nielul de ocluzie
		Evenimente ≥ 1500
		Istoria
		Data, ora
Capacitatea pompei	Diapazonul volumului infuzat	0.1 - 9999 ml
	Rata fluxului	0.1-1,200 ml/h
		 Setarea prin incrementare incepind cu 0.1 ml
	Regim de lucru KVO - mentinerea venei deschise	0.1 - 3 ml/h
	Acuratetia infuziei	≤5 %
	Functia bolus	Da
	Funcția de reglare a vitezei de infuzie în bolus	1 - 1000 ml/h
	Selectarea volumului infuzat	Da
	Selectarea ratei de infuzie	Da
	Calcularea dozei/medicament	Da
	Regim de lucru in picaturi (setarea picaturilor)	Da
	Regim de lucru in flux (setarea fluxului)	Da
	Trecerea automată în regim de așteptare 	da
	Încărcarea seringii	da
	Carcasa ermetizata pentru a preveni scurgerea de lichide in interiorul dispozitivului	Da
	Senzor de picaturi	Da
	Nivelul presiunei de ocluzie	Cel putin 8 niveluri
	Funcția de evacuare a bulei de aer din sistema IV	Da
	Funcție de chemare asistentă	Da
	Blocarea panoului de control	Da
	Alarma sonora	Da
	Alarma vizuala	Da
	Fixarea perfuzorului pe rampă verticală 	Da
Setări IV	Protecție la flux liber	da
	Capacitatea de captare aer	da
	Conectare IV a acului	da
Alarmă acustică și vizuală	Ocluzie	da
	Nivelul de ocluzie ajustabil în dependență de pacient	 ≥ 1 - 15 psi
	Afișarea ocluziei în timp real	da
	Alarmă de flux	da
	Bula de aer	da
	Eroare de sistem	da
	Set decuplat	da
	Rezervor gol	da
	Ușă deschisă	da
	Senzor de picaturi	Da
	Infuzia completă	da
	Baterie descărcată	da
	Posibilitatea de setare a volumului	da
	Mod silence a alarmei	da
Sistem de reducere a erorii (tehnologie inteligentă)	Setarea parametrilor impliciți dați de  sistemul de reducere a erorii de dozare la pornirea pompei	da
	Programe de analiză presetate	da
Port I/O	da
Alimentarea electrică	Rețea electrică 220 V, 50 Hz	da
	Baterie incorporata	da
	Durata de operare autonoma	≥5 h la fluxul de 25ml/h
Acesorii
Suport	sistem de fixare a pompei de stativul de infuzie	da
Circuite 	 	≥ 200 buc

title: Dispozitiv fizioterapeutic de inhalare

description:
Prezența unui cronometru, care va oferi: oprirea automată a inhalatorului  după un timp stabilit cu semnal sonor intermitent (taimer de timp încorporat)
Mod de operare: continuu
Dimensiunea particulelor până la 5 (80%), restul până la 20 microni
Volumul camerei pulverizate:  min  40 ml
Temperatura aerosolului 27 ... 29˚˚
Minimum  2000 ore de funcționare
Timp de funcționare continuu: min 6 ore (urmat de o pauză de 2 ore)
Frecvența oscilațiilor electrice ale generatorului 2,64 ± 3% MHz
Tensiunea rețelei electrice: 220V, 50 Hz
Puterea : max 130 VA
Clasa de electrosecuritate : Clasa I Tip B

title: Ventilator pulmonar neonatal cu frecvență înaltă

description:
Tip pacient: nou-născut, prematur
Mobil, pe suport cu rotile
Brat pentru tuburi, ajustabil
Mîner de transport
Umidificator activ, cu temperatură ajustabilă și măsurarea temperaturii în tubul respirator
Ecran min. 15 inch, color, touch screen
Parametrii de ventilare:
Mod ventilare: HFO, SIMV, A/C, CPAP, PC
Forma de unda flux: dreptunghiulara, descendentă
Presiunea de insipartie: ≥ 80 cmH2O
Timpul de inspirate: 0,1-3 s
Volumul tidal: ≥ 2-300 ml
Fluxul de inspiratie ≥ 30 lpm
PEEP/CPAP ≥ 0-30 cmH2O
Frecventa respiratorie in regimuri de ventilare convenționale ≥ 1-150 rpm
Frecventa respiratorie in regim de ventilare HFO ≥ 5-16 Hz
Amplitudinea HFO ≥ 0-150 cmH2O
Trigger presiune
Trigger volum
Concentratia de oxigen 21-100 %
Flush O2, ajustabil 21-100 %
Alarme:
Alarme vizuale si sonore
Alarme cu prioritate mică, medie și înaltă
Buton anulare alarmă
Lipsă alimentare gaz aer, oxigen
Lipsă alimentare electrică
Eroare de sistem
Alarme ajustabile, parametrii ventilatorii
Parametrii măsurați:
Presiunea inspiratorie de vîrf
Presiunea medie
PEEP
Rata IE
Frecventa respiratorie
Volumul tidal
Minut volum
Scurgerile
Complianta
Concentratia O2
Vizualizare grafice:
Presiune
Flux
Volum
Presiune-volum
Presiune-flux
Flux-volume
Conditiile de functionare:
Tensiunea 220V, 50 Hz
Presiuena gaz 3-4 bar
Furctun cu conectori O2, Aer, tip DIN
Accesorii și consumabile:
Circuite respiratorii reutilizabile nou-născut, 3 buc.
Cameră de umidificare reutilizabilă 2 buc.
Filtre antibacteriale 50 buc.
Senzor de debit 2 buc.
Plamăn de test 2 buc.
Cerințe generale:
Termen de garanție minim 24 de luni
Timp maxim de intervenție postgaranție 24 ore
Certificat de la producător CE
Certificat de la producător ISO 13485
Trainingul personalului medical și tehnic
Instalarea și punerea în funcțiune la sediul beneficiarului de către reprezentantul autorizat al producătorului .

title: Dispozitiv fizioterapeutic de frecvență ultraînaltă (unde scurte)

description:
Dispozitiv fizioterapeutic cu unde scurte
Tensiunea rețelei electrice: 220V, 50 Hz
Frecvența: 50 Hz
Frecvența de funcționare: max  27 MHz
Puterea : max 60 VA
Ecran LED cu indicarea puterii și timpului
Timpul de funcționare  max 20 min, pauză max 10 min
Electrozii înlocuiți: 3 perechi de diferite diametre
Clasa de securitate:  I, BF

title: Dispozitiv fizioterapeutic portativ pentru magnitoterapie

description:
Amplituda  inducției magnetice pe suprafața de lucru a aparatului cu o formă de curent sinusoidal:
- în poziția „I” a comutatorului valorii inducției magnetice - (10 ± 2,5) mT
- în poziția „II” a comutatorului valorii inducției magnetice - (30 ± 7,5) mT
Tensiunea rețelei electrice: 220V
Frecvența: , 50 Hz
Puterea de consum: max 30 VA
Greutatea : max 0,9 kg
Dimensiuni exterioare nu mai mari 148x114x45 mm
Amplituda  inducției magnetice pe suprafața de lucru a aparatului cu o formă de curent sinusoidal:
- în poziția „I” a comutatorului valorii inducției magnetice - (10 ± 2,5) mT
- în poziția „II” a comutatorului valorii inducției magnetice - (30 ± 7,5) mT
Tensiunea rețelei electrice: 220V
Frecvența: , 50 Hz
Puterea de consum: max 30 VA
Greutatea : max 0,9 kg
Dimensiuni exterioare nu mai mari 148x114x45 mm
Amplituda  inducției magnetice pe suprafața de lucru a aparatului cu o formă de curent sinusoidal:
- în poziția „I” a comutatorului valorii inducției magnetice - (10 ± 2,5) mT
- în poziția „II” a comutatorului valorii inducției magnetice - (30 ± 7,5) mT
Tensiunea rețelei electrice: 220V
Frecvența: , 50 Hz
Puterea de consum: max 30 VA
Greutatea : max 0,9 kg
Dimensiuni exterioare nu mai mari 148x114x45 mm

title: Microscop trinocular integrat/încorporat cu camera digitală rezoluție înaltă

description:
Locul amplasării
Secție de morfopatologie
Parametrii tehnici
Ocular
Tipul ocularului
Binocular sau trinocular în caz de montare a camerei în top
Amplificare ocular
Minimum 10x
Număr de cîmp a ocularului
Minimum 25
Tip ocular
Cu ajustare dioptrică
Compensare ametropie
De la +5 la -5
Distanța interpupilară
Ajustabilă
Distanța interpupilară
45 -75
Interval unghi de înclinare a tubului de observare
30- 45˚
Turelă rotativă și obiective
Număr de obiective minim incluse
5
Tip obiectiv
Plan Fluorit
Obiectiv necesar
4x
Numărul aperturii a obiectivului 4x
Minimum 0,13
Distanța de lucru a obiectivului 4x
Minimum 17
Obiectiv necesar
10x
Numărul aperturii a obiectivului 10x
Minimum 0,3
Distanța de lucru a obiectivului 10x
Minimum 11
Obiectiv necesar
20x
Numărul aperturii a obiectivului 20x
Minimum 0,3
Distanța de lucru a obiectivului 20x
Minimum 2,1
Obiectiv necesar
40x
Numărul aperturii a obiectivului 40 x
Minimum 0,75
Distanța de lucru a obiectivului 40x
Minimum 0,2
Obiectiv necesar
60x
Numărul aperturii a obiectivului 60x/100 draft
Minimum 0,85
Distanța de lucru a obiectivului 60x
Minimum 0,2
Număr de cîmp a obiectivelor
Minimum 25
Butoane de reglaj
Axe de reglare a suportului
Mod de acționare a suportului
Rapidă și fină
Treaptă de reglare fină a focusării
Maximum 1 µm
Iluminator
Sursă de lumină
LED
Mod iluminare
Prin transmitere (din partea inferioară a suportului)
Ajustare luminozitate
Da
Durata de viață a sursei de lumină
Minimum 50000 ore
Platformă de lucru
Cu cleme de fixare
Da
O clemă mobilă
Da
Diafragmă și condensor
Tip condensor
Acromatic și aplanatic
Tip diafragmă
Cu diametru ajustabil
Interval de ajustare a diafragmei
Pentru intervalul 4x- 100x
Sistem de achiziționare a imaginilor
Cameră video
Digitală
Modul de montare
Integrată în carcasă sau montată top
Tehnologia camerei
CCD
Calitatea imaginei furnizate
Minimum Full HD
Frecvența maximă de citire a imaginilor
Minimum 30 fps
Dimensiunea pixelului
Minimum 3,4 µm
Lungimea cuvîntului de codificare a imaginei
Minimum 12 bits
Modul de conectare la calculator
USB 3.0 sau Cablu FireWire 800
Sursă de alimentare
Tensiune de alimentare
220V, 50 Hz
Tip fir de alimentare
Shuko
Accesorii
Chit de conectare a camerei pe microscop
1 bucată
Chit de conectare a camerei video la calculator
1 bucată
Husă de protecție
1 bucată
Standarte și certificate
Certificat
CE ( Certificat european)
Standard
ISO 9001:2015 Standart de calitate a dispozitivelor
ISO 13485:2016 Standart de calitate a dispozitivelor medicale
Instruire
Training pentru utilizatori
Da
Module necesare:
Prezentare generală, pregătire de lucru, operarea cu dispozitivul, întreținere și dezinfectare
Training pentru personal tehnic
Da
Module necesare:
Modul de efectuare a mentenanței periodice, modul de testare a funcționalității
Documentație
Manualul de utilizare
română sau rusă
Manualul de utilizare
engleză
Manualul de service
engleză
Logistică
Transportare la adresa solicitată
Da
Instalare
Da
Testare
Da
Garanție
Perioada de garanție
Minim 4 ani
Efectuarea mentenanței în perioada de garanție de către furnizor
Conform recomandărilor producătorului
Instruire
Instruire pentru utilizatori
Da
Module necesare în instruirea pentru utilizatori
Prezentare generală, pregătire de lucru, operarea cu dispozitivul, întreținere și dezinfectare
Training pentru personal tehnic
Da
Module necesare în instruirea pentru personal tehnic
Modul de efectuare a mentenanței periodice, modul de testare a funcționalității

title: Monitor neonatal pentru monitorizarea funcțiilor vitale

description:
Monitor pentru monitorizarea funcțiilor vitale
Parametrul	Specificația
Tip pacient	Neonatal	da
Display, tip	Diagonala	≥ 15 inch
	Rezolutia 	≥ 1366 x 768
	LCD, Touchscreen	da
	Numărul de curbe afișate
concomitent	≥10 curbe
	Parametrii afișați	ECG
		Pulsul
		SpO2
		Fotopletismograma
		Presiunea sanguină neinvaziv
		Capnografia
		Temperatura
Analiza automată a undei ST	Trendingul ST	da
	Media în analiza segmetului ST	da
Modul ECG	Culegerea semnalului ECG prin
cablu 5 electrozi	da
	I, II, III, aVR, aVL, aVF, V	da
	Sensivitatea	2.5 mm/mV, 5 mm/mV, 10 mm/mV, 20 mm/mV
	Viteza de afisare	6.25mm/s, 12.5 mm/s, 25 mm/s, 50 mm/s
	Gama de rejecție a modului comun la 50 Hz	≥ 100 dB
	Frecventa	0,05-100 Hz
	Impedanta de intrare	>5MΩ
	Rata cardiacă	15 - 350 bpm
	Detectarea aritmiei	da
	Detectarea pacemaker	da
	Monitorizarea respiratiei	da
	Protecție înportiva șocurilor de
defibrilare	da
Modul SpO2	Diapazonul	1 - 100%
	Tehnologia	Masimo
	Acuratețea la 70 - 100%	≤ 3 %
	Monitorizarea rata pulsului	30 - 250 bpm
	Monitorizarea pletismogramei	da
Modul NIBP	Diapazonul	10 - 135 mmHg
	Regim de măsurare	manual, automat
	Regim automat	1,5, 10, 15, 30, 60, 120,240,480 min.
	Buton de activare/dezactivare
manual a NIBP	da
	Metoda de măsurare	oscilometrică
	Protecție de suprapresiune la
regimului de pacient neonatal	150 mmHg
	Eroarea medie de măsurare	≤ ±5mmHg
Modul Capnografie (CO2)	Modul Capnografie (CO2)	da
	Tehnologia	Sidestream

	Diapazonul	0 - 150 mmHg
	Rata respiratorie	2 - 150 rpm
	Monitorizarea unei CO2	da
Modul Temperatura	Diapazonul	0 - 50.0 °C
	Rezoluția	0.1 °C
	Număr de senzori	2 unit.
Protocoale de lucru preprogamate	da
Protocoale de lucru setate de utilizator	da
Memorie internă	da
Trendingul evenimentelor	da
Arhivarea datelor	da
Imprimantă	Încorporată	da
Forme de raportare	Viteza de înscriere	12.5mm/s, 25mm/, 50mm/s
	Înscriere continuă	da
	Curbe imprimate	≥ 3
	Preprogramate	da
	Personalizate	da
Alimentarea	Rețea electrică 220 V, 50 Hz	da
	Baterie internă	da
	Timp de lucru autonom a bateriei	minim 5 h
Alarma	Vizuala, sonora	da
	Manjetă deconectată	da
	Manjetă defect	da
	Senzor SpO2 deconectat	da
	Senzor SpO2 defect	da
	Alarmă concentrația SpO2 scăzuta/mărită	da
	Nivelul baterie scăzut	da
	Alarme rata cardiacă scăzuta/mărită	da
	Lipsă sursă externă de alimentare	da
	Buton de dezactivare/anulare alarmei sonore	da
	Ajustarea nivelului de alarmă	da
Accesorii și consumabile:
Cablu ECG	cu 5 electrozi  	1 buc.
Electrozi ECG	neonatal, unică utilizare	300 buc.
Senzor SpO2	neonatal, reutilizabil	2 buc.
Senzor Temperatură cu
fixare pe piele	neonatal, reutilizabil	1 buc.
Senzor Temperatură  rectal 	neonatal, reutilizabil	1 buc.
Manjete NIBP	neonatal Nr.1, 2, 3, 4, 5,
reutilizabil (1set/5buc)	5 set.
Hîrtie termică	10 buc.
Circuite pentru Capnografie (CO2)	neonatal, unică utilizare	10 set.
Suport pe rotile
(să se indice modelul suportului)	suport	da
	min. 5 rotile	da
	min. 2 rotile cu frînă	da
	coș pentru accesorii	da
	mîner pentru transportare	da
	sistem de fixare a dispozitivului
de suport	da

title: Analizator biochimic automat

description:
Cerinţele electrice ale sistemului:
Toate componentele echipamentului electric se vor conecta la priză prin fişă de reţea de tip F  “Schuko”
Electricitate: 220  Volţi monofazic
Frecvenţa 50 Hertz
Condiţiile fizice şi de mediu:
Temperatura mediului min -20 max +40oC
Umiditatea 30-100%
Variaţia tensiunii de reţea:
Echipamentul trebuie să funcţioneze satisfăcător de la  12.5% la +8% din tensiunea nominală a reţelei de  220 Volţi.
Echipamentul nu trebuie să fie distrus la variaţii de   21% pînă la +12.5% din tensiunea nominală a reţelei de  220 Volţi.
Interferenţa  electromagnetică (EMI)
Perormanța dispozitivului nu trebuie să fie afectată de EMI radiată sau indusă de liniile de alimentare ale altor dispozitive
Calitatea construcţiei
Echipamentul nu trebuie să aibă margini ascuțite
Toate componentele echipamentului trebuie bine fixate
Echipamentul trebuie să fie sigur și să asigure o protecție adecvată contra părților în mișcare și a părților aflate  sub tensiune electrică
Toate componentele echipamentului trebuie să fie bine fixate.
Echipamentul trebuie să fie sigur şi să asigure o protecţie adecvată contra părţilor în mişcare şi a părţilor aflate sub tensiune electrică
Echipamentul trebuie să fie  construit din materiale durabile şi să reziste la utilizare şi curăţire tipică.
Întrerupătoarele, butoanele și alte componente de comandă trebuie să fie proiectate pentru o utilizare intensivă
Terminalele mecanice, electrice și pneumatice precum și conectoarele, jacurile, articulațiile trebuie proiectate astfel  încît să prevină pătrunderea lichidelor, conectarea incorectă și folosirea inadecvată a conectoarelor
Conexiunile trebuie să fie sigure și să reziste la deconectări accidentale și după caz să mențină sterilitatea
Etichetarea:
Tot echipamentul furnizat va purta marcajul CE.
Etichetele şi marcajele trebuie să fie clare şi lizibile.
Etichetele şi marcajele trebuie să fi suficient de durabile încît să reziste la curăţarea de rutină şi uzura normală
Legenda adecvată de avertizări trebuie să fie plasată pe dispozitiv.
Cerințe Specifice:

Certificat  CE,ISO 9001
Tip „bench top”
Greutate
Compartiment reactive/probe
Posibilitatea de procesare nu mai putin de 70 seruri concomitent
Sistem de racire al reactivelor
Nu mai sus de 15°C
Teste/oră
Nu mai puţin de 220 teste /oră
Tip probe analizate
Plazma
Urina
Ser
Compartiment reactive
Posibilitatea de utilizare - nu mai puţin  de 50 de reactivi plasaţi concomitent .
Compartiment preparare/dozare
Dozare reactiv: 10÷440 mcL ,cu pas 0.126mcL
Dozare ser:  3÷40 mcL;cu pas 0.126 mcL
Viteza de dispensare : nu mai putin de 880 mcl/sec
Compartiment de reacţie/citire al probelor
Rotor  cu 100-120 de cuve de unică folosinţă de reacţie/citire,volum de reacţie 200-800mcL,lungimea optică 6mm ,37°C
Număr de probe se pot programa pe aparat  (la o listă de lucru)
Nu mai puţin  de 120  seruri plasate concomitent
Mecanizm de protectie
Detectie verticala a obstacolului, cu oprirea analizatorului (protectia de obstructie a acului dozator)
Consum de apa didtilată
Mai puțin de 0,5L pe ora
Analizorul nu va necesita conectare la sursa de apă externă
Diluţie automată a probelor
DA
Autoverificare
DA
Marcare valori anormale
DA
Afişare rezultate
Calculator extern
Cititor al codului de bare
OPTIONAL
Sistem
Analizator automat sistem inchis, cu procesarea testelor mono si bireagente. Ofertantul este obligat sa nu includa sistemul optional la nivel de soft in timpul exploatarii aparatului.
Complet automat, acces aleatoriu, discret
Exportul  datelor

Teste programate
Da
Interffaţa computer
Bidirecţional   ,
Teste difinite de utilizator
DA
Metode de testare
Punct final
Cinetic
Timp fix
Turbodimetrie
Diferenţial
Bicromatic
Diferenţial cu blank de probă
Teste de bază

Creatinina
Ureia
Proteina totală
Albumina
Acid uric
Colesterol
Trigliceride
Glucoza
Bilirubin total şi direct
Kaliu
Natriu
Calciu
Fosfor anorganic
Clor
Calciu
Magneziu
Fier
ALT
AST
Fosfataza alcalină
Lipaza
Amilaza
Gama GTP
Creatin Kinaza
Lactat dehidrogenaza
Colesterol HDL
Microalbumina
s.a.
Tip reagent
Lichid
Rezultate stocate (memorate)
Minimum  100 000
Întrerupere temporară a lucrului
Posibilitate de a întrerupe lucru temporar datorită unor factori externi ( lipsa ser, lipsa reactiv, lipsa apa, bidon reziduri plin)
Sistem optic
Filtre cu gama între 340-670nm,  8 filtre,reversie optică
Citire bicromatica sustinuta de orice protocol “end point”, “diff mode”, “cinetic”, “fixe time”
Rezoluţie
0,0001 Abs
Interval de măsurare
0÷2.5 Abs ,conversat la10mm
Sursa lumină
Lampă cu halogen
Sistema operationala
Licenţiată
Windows 7 ori mai performantă
Alimentare electrică
220V (50/60Hz)
Calculator
USB2.0; COM port (RS-232)
PENTIUM 4/ 2000MHz/1Mb/800MHz
RAM 1024 MB
HDD 256GB 7200rpm
DVD ROM+CD-RW IDE/,  tastatură, mouse
LCD 19 "
Sistem  UPS
220v ,2000VA,  AVR
Printer
Laser A4, min 1200x600dpi,20ppm,USB Alb/negru
Putere  consumata
Mai putin de 300VA
REAGENTI:
În set se va oferi reagenţii pentru un ciclu de lucru (indicaţi lista şi cantitatea)
CERINȚELE PENTRU ASISTENȚĂ
Piese
Indicați perioada în care piesele de schimb pentru echipamentul oferit vor fi disponibile
Termen de garanție:nu mai puțin de 3 ani, indifferent de distribuitorul de reagenți
Service ingineri certificați de producător –Obligatoriu
Autorizația producătorului pentru deservirea tehnică a utilajului oferit- Obligatoriu
Întreținerea profilactică: o descriere a întreținerii profilactice necesară pentru echipamentul oferit inclusive frecvența
Documente și manual: Tot echipamentul trebuie să fie livrat cu copii ale instrucțiunilor de utilizare și întreținere în limba de stat sau rusă - Obligatoriu

title: Pompă de infuzie (perfuzor) cu seringă

description:
Parametrul 	Specificația
Display	Date afișate	Doza	da
		Concentrația	da
		Viteza de infuzie	da
		Volumul de infuzie	da
		Timpul de infuzie	da
		Timpul pînă la final de infuzie	da
		Mărimea seringei	da
		Nivelul de ocluzie	da
		Rata KVO	da
		Data, ora	da
		Nivelul de încărcare a baterii	da
		Evenimente	da
		Alarme	da
Proprietățile pompei	Rata fluxului	diapazonul	0.1 - 1,200 ml/h
		Pasul de incrementare	0.1 mL/h
	Dozarea bolusului	Incrementarea	0.1-25 ml
	Rata KVO	0.1 - 3 ml/h
	Eroarea	≤2%
	Compatibil cu seringi produse de diferiți producători	min. 8 producători
	Seringi acceptate	5, 10, 20, 30, 50, 60
	Detector de mărime a seringei	da
	Nivelul presiunei de ocluzie	≥ 8 nivele
	Protecție la pătrunderea lichidelor în dispozitiv	IP 22 - 24
	Chemare asistentă	da
Caracteristici de siguranță	Protecție la curgere liberă din sistem	da
	Intervalul de timp de blocare, min	≤ 5-100
	Limita dozei acumulate	da
	Limitarea de acces prin buton de blocare sau parolă	da
Alarme și indicatori	Ocluzie sus/jos	da
	Eroare de flux	da
	Eroare de sistem	da
	Seringă goală	da
	Perfuzia pe sfîrșite	da
	Baterie descărcată	da
Alarmă sonoră	Volum reglabil	Da
	Buton mod silențios	da
Jurnalul de evidență	Numărul de evenimente memorate	≥ 1000
	Evenimente stocate	Butoane apăsate	da
		Coduri de eroare	da
		Alarme	da
		Medicamente injectate	da
		Cantitatea infuzată	da
		Setări	da
		Bolus solicitat	da
Alimentarea	Rețea electrică 	220 V, 50 Hz	da
	Baterie internă	Reîncărcabilă	da
		Timp de operare	≥ 7 h la 5 mL/h
Acesorii
Suport	sistem de fixare a pompei de stativul de infuzie	da
Seringi	5 ml	50 buc.
	10 ml	50 buc.
	20 ml	50 buc.
	30 ml	50 buc.
	50 ml	50 buc.
	60 ml	50 buc.
Circuite	circuit de interconectare 	300 buc.

title: Microtom de precizie și ”rotație” inclusiv accesorii și consumabile

description:
Modul de operare
Manual

Tip suprafață
Anticorozivă
Material de impregnare a probei
Parafină
Greutatea
Maximum 30 kg
Sistemul de tăiere
Grosimea feliei tăiate
0,5 – 60 µm

Acuratețea de tăiere a feliei
0,5 – 2 µm / Maximum 0,5 µm;
2 – 10 µm / Maximum 1 µm;
10 – 20 µm / Maximum 2 µm;
20 – 60 µm / Maximum 5 µm.
Excursia frontală maximă a sistemului de tăiere
Minimum 25 mm
Dimensiunile maxime ale probei
Minimum 45 x50 mm
Funcționalități speciale
Retracția probei în procesul de tăiere
Da
Sistem de oprire sau de alarmare la finalizarea tăierii probei
Da
Blocarea roții de acțiune
Da
Posibilitate de modificare a unghiului de tăiere a cuțitului
Da
Accesorii și consumabile
Lame de unică utilizare de profil îngust din oțel inoxidabil cu unghiul lamei de 35°  (pentru țesut moale)
1000 buc
Lame de unică utilizare de profil lat din oțel inoxidabil cu unghiul lamei de 35°  (pentru țesut dur)
400 buc
Suport pentru cuțit de profil îngust
1 buc
Suport pentru cuțit de profil lat
1 buc
Husă de protecție
1 buc
Ulei pentru lubrifierea angrenajelor
200 ml
Cerințe față de producător și furnizor
Garanție
Minimum 5 ani
Certificate
CE certificat european
Standarte
ISO 9001, ISO 13485
Logistică
Manuale de utilizare
Da, în limba română sau rusă
Manual de service
Da, engleză
Transportare la locul necesar
Da
Instalare
Da
Testare
Da
Training pentru utilizatori
Da, pregătire de lucru, mod de utilizare a dispozitivului, întreținerea zilnică
Training pentru personal tehnic
Ajustarea, calibrarea, înlăturarea defecțiunilor minore, parole de acces

description: Incubator nou născuți

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Sistem de impregnare (incluzionare) în parafină

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Electrodermatom

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Microtom de precizie și ”rotație” inclusiv accesorii și consumabile

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Dispozitiv fizioterapeutic de inhalare

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Unitate de fototerapie cu raze ultraviolete

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Monitor- analizator gaze anestezice

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Monitor neonatal pentru monitorizarea funcțiilor vitale

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Colorator de lamele

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Pompă de infuzie (perfuzor) cu seringă

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Insuflator CO2

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Ventilator pulmonar neonatal cu frecvență înaltă

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Procesor automat de țesuturi

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Dispozitiv fizioterapeutic de frecvență ultraînaltă (unde scurte)

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Analizator biochimic automat

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Pompă de infuzie (perfuzor), volum mare

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Microscoape pentru examinări în histopatologie

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Dispozitiv fizioterapeutic portativ pentru magnitoterapie

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: Dispozitiv fizioterapeutic pentru amplipuls terapie

description: Echipamente medicale

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

title: Planning Notice

title: duae (16) (4).doc

title: anunț de participare corectat.docx

title: documentația standart (9).docx

title: clarificări pentru operatorii economici.pdf

title: demers caiet de sarcini  23.07.2019 (5).docx

title: anunț de participare modificat.docx

title: anunțul de participare ajustat.docx

title: anunt de participare (2) (10).docx

title: documentația standart (9) (1) modificat.docx

title: anunț de participare corectat.docx

title: Buna ziua, se consideră ca eroare tehnică încărcare la specificarea tehnică : capacitatea pompei încărcarea cu seringă, astfel ”încărcarea cu seringă” este echivalent cu ”încărcarea dispozitivului - Răspuns din 23 Oct 2019, 13:57

answer: Buna ziua, am corectat eroarea tehnică inclusiv și în Anunțul de participare (Anunțul de participare corectat din data de 23.10.2019), multumesc

description: Vă rugăm mult să includeți corectarea aceasta în documentația de atribuție și să reînoiți fișierul în sistem, incluzînd modificările necesare. Cu respect.

title: Criterii de evaluare a ofertelor

answer: Buna ziua, criteriul de evaluare a ofertelor va fi : prețul cel mai mic cu coespunderea deplină a bunurilor, multumesc pentru întrebare

description: Bună seara! Referitor la criterii de evaluare a ofertelor: în documentația de atribuire indicați: criteriul de evaluare - cel mai mic preț. Iar mai jos indicați: factorii de evaluare - Prețul cel mai scăzut pe lot și calitatea produsului. Deci, indicați lucruri principial diferite. Vă rugăm să indicați exact, care va fi criteriul de evaluare: fie cel mai preț, fie cel mai mic preț cu corespunderea deplină a bunurilor, fie prețul cel mai scăzut pe lot și calitatea produsului. Dacă veți indica ultima opțiune, solicităm să indicați care este pontajul / ponderea fiecărui criteriu în evaluarea bunului. Astfel, operatorii economici vor înțelege corect după care principiu să ofere bunurile solicitate în licitație.

title: Procesor automat de țesuturi

answer:
Principiul tehnicii histopatologice impune transformarea unui prelevat dintr-un organ sau ţesut
dintr-o masa tisulară opacă într -un preparat fin, translucid care permite vizualizarea microscopica
a detaliilor structurale tisulare.

Pentru a realiza aceasta prelevatul suferă transformări succesive:

1. Fixarea - previne alterarea si autoliza ţesutului

2. prelucrarea histopatologică (deshidratare, clarificare şi impregnare cu parafină)
aceste două etape se efectuază cu ajutorul histoprocesorului...

Mai sunt alte etape ...
la care se folosește alt tip de aparate medicale de laborator
3. Includerea in bloc de parafina
4. Secţionarea la microtom a prelevatului inclus in parafina
5. Etalarea secţiunilor pe lame (obţinerea preparatului histopatologic necolorat)
6. Colorarea lamelor
7. Montarea lamelor - permite protecţia si conservarea preparatului histopatologic colorat
Produs final - preparatul histopatologic permanent care permite interpretarea corecta in vederea
stabilirii diagnosticului anatomo-patologic.

Protocoalele de lucru histopatologie sunt orientative; pot fi folosiţi reactivi diferiţi, timpul de
acţiune şi concentraţiile menţionate în reţete sunt orientative şi se adaptează în funcţie de
condiţiile locale (temperatură ambientală, duritatea şi pH-ul apei de robinet etc). Tehnica folosită
este considerată corectă dacă se obţin rezultatele preconizate prin coloraţia respectivă...

description:
Buna ziua.
La Procesor automat de țesuturi se solicita etape de procesare: minim 16.

Avand in vedere ca etapele esentiale ale unei procesari intr-un laborator de anatomie patologica sunt fixarea, deshidratarea, clarificarea si impregnarea cu parafina, va rugam sa acceptati echipamente care indeplinesc cel putin aceste 4 etape principale, ce asigura procesarea cu success, la un inalt nivel calitativ al probelor.
In cazul in carenu se accepta 4 etape. Va rugam sa enumerati toate 16 etape pe care le doriti. Va multumim anticipat.

title: Acest grup de produse include pompe de infuzie pentru volume mari cu cerințe avansate; pot avea 2 sau mai multe canale. Fluxul este calibrat în ml/oră. Au posibilitatea de a efectua calcule a raportului medicament/doză, ce permite programarea fluxului reieșind direct din indicațiile medicului. Astfel se are în vedere cel puțin două (2 sau mai multe canale) canale, respective instituția are nevoie de 5 pompe și nu de 10 - Răspuns din 23 Oct 2019, 13:57

answer: Deci se solicită pompă (5 pompe) de infuzie pentru volume mari cu cerințe avansate și respectiv 2 sau mai multe canale conform parametrilor tehnici solicitați de către instituție, mulțumim pentru întelegere

description: Este clar că aveți nevoie de cca 10 canale pentru administrarea substanțelor medicamentoase (5 pompe x2 canale). Care atunci este temeiul pentru care respingeți operatorii economici, ce pot satisface necesitățile insituției, oferind la un preț avantajos 10 pompe, dar nu 5, cu respectarea tuturor cerințelor tehnice față de dispozitiv? Nu este o decizie întemeiată din punctul de vedere a eficientizării resurselor economice de stat. Toți parametrii tehnici enumerați în caietul de sarcini sunt standarde pentru majoritatea pompelor calitative și ieftine. Dar din cauza că solicitați pompa să fie cu 2 canale, contribuiți la o irosire neîntemeiată a bugetului de stat! De aceea, solicităm să indicați în documentația de atribuire că acceptați 5 opmpe cu 2 canale fie 10 pompe cu 1 canal cu corespunderea tuturor cerințelor tehnice stipulate față de dispozitiv.

title: Incubator nou nascuti

answer: Buna ziua, stimate operator economic, cerințele și parametrii  tehnici ale dipozitivelor medicale solicitate  inclusiv și pentru incubator nou născuți ,  au fost elaborate reieșind din specificul de activitate al instituției, astfel instituția solicită incubator nou născuți cu  parametrii avansați (pentru copii născuți prematuri) , multumim

title: Procesor automat de țesuturi

answer:
Buna ziua, referitor la întrebarea Dvs, Întradevăr 5 grade nu este semnificativ și decesiv pentru impregnarea țesuturilor cu parafină (45-50-70 C;
NOTA temperatura de topire a parafinei histilogice este de cca 55-56 C) ...
în aceasta cerință se are in vedere intervalul  cu posibilitatea de reglaj automat/semiautomat și  manual - ca ajustare în caz de necesitate...(Interval de temperatură a parafinei). Totuși decizia definitivă o va lua grupul de lucru ținînd cont de parametri tehnici solicitați

description:
Parametrul: interval de temperatura a parafinei este solicitat 45-70 grade.
Va fi acceptat 50-70 grade ? Acet lucru nu influenteaza in nici un fel performanta systemului.
Multumesc

title: Incubator nou nascuti

answer:
1. Prin solicitarea de dulap s-a subînțeles oferirea spațiilor de stocare închise, care pot fi sub forma de dulap sau sertar, inclusiv, dar nu se acceptă interpretare de poliță deoarece nu oferă sterilitatea spațiului închis.

2. Conform specificației tehnice Incubator pentru noi născuți, în special, modul de control al temperaturii, funcția de temperatură > 37 o C, s-a subînțeles funcția, ce permite setarea unui regim de temperatură mai mare decât cel admis, doar în cazurile pacienților cu greutate extrem de mică la naștere. Este o funcție ce se conformează standardului EN 60601-2-19 și este obligatoriu pentru toate dispozitivele de acest gen.

3. Funcția: Anulare alarmă fără atingere - presupune mișcări libere a mâinii în fața părților specializate a incubatorului, fără atingeri tactile de suprafață. Așa dar, în timpul manipulărilor cu pacientul, contactul cu suprafețele exterioare ale incubatorului de exemplu: buton sau telecomandă impune un risc înalt al infecțiilor, deci solicitarea de a exclude acest parametru este nefondată.

4. Da, acceptăm, că fiecare incubator are loc special pentru umidificator, dar pentru acordarea unui proces clinic sigur, parametru: Alarma plasare incorectă a umidificatorului este necesar pentru a exclude factorul uman, care poate duce la scăderea bruscă a umidității în incubator și înrăutățirea stări pacientului.

5. Solicitarea de includere: control al oxigenului 21%-95% nu este întemeiată, deoarece diapazonul se încadrează în limitele stipulate în caietul de sarcini de 21%-60%, deci oferă o posibilitate largă de participare a operatorilor economici.

6. Diapazonul SPO2 indicat 70 -100%, este acel utilizat în clinica medicală, dar oferirea unui diapazon de 0 -100% este un avantaj.

7. Cerința cântar rezoluție 100g – 5 kg, nu are intenția de a limita concurența, deci posibilitatea de a oferi o rezoluție mai mare este un avantaj.

description:
Pentru cerinta:
* "dulap pentru scutece si consumabile medicale, minim 3 buc.": explicam ca un incubator nu poate avea 3 dulapuri. Corect ar fi polita si/sau sertar, minim 3 buc.
*"Functia de temperatura > 37°C, pina la ce temperatura? cit de mare? mai ales ca mai jos este stipulat intervalul de temperatura trebuie sa fie intre 34-37 °C.
*"anulare alarma fara atingere", explicati va rog.Totodata solicitam sa excludeti acest punct deoarece daca va referiti la telecomanda atunci este o optiune care costa si este nejustificata, ori butonul de anulare a oricarei alarme se afla in afara incubatorului iar motivul pentru prevenirea infectiilor nu este intemeiat.
*"alarma plasare incorecta umidificator", solicitam sa excludeti aceasta cerinta deoarece fiecare incubator are un loc fix determinat pentru plasarea umidificatorului, astfel aceasta cerinta fiind nejustificata.
*"acuratete concentratie de oxigen - 21-60%", corect este de la 21 pina la 95% cu reglare.
*"masurare SpO2-70-100%" daca vorbim despre incubator pentru nou-nascut atunci masurarea SpO2 trebuie sa fie 0-100%
*"cantar rezolutie 100 g-5000g" rugam sa analizati aceasta cerinta deoarece pot fi situatii cand copilul poate depasi limita normala de greutate.
Multumim.

title: intrebarea la lotul nr.19

answer: Buna ziua, se consideră  ca eroare tehnică încărcare la specificarea tehnică : capacitatea pompei încărcarea cu seringă, astfel ”încărcarea cu seringă” este echivalent cu ”încărcarea dispozitivului

description:
Bună ziua!
La lotul nr. 19 <<Pompă de infuzie (perfuzor), volum mare>> se cere să fie, după cum revine din specificații tehnice, dispozitiv pentru sisteme de infuzie cu pungă. Însă în specificații tehnice la locul dat este menționat: <<Încărcarea seringii - da>>. Însă dispozitivul de tipul dat nu este predestinat pentru lucru cu seringi!!! Pentru așa seringi este utilizat principial alt tip de pompă! Vă solicităm insistent să corectați erori comise în specificații tehnice.

title: Microscop trinocular  integrat/incorporat cu camera digitală rezoluție înaltă

answer: Buna ziua, stimate operator economic, cerințele și parametrii tehnici al dispozitivelor medicale solicitate au fost elaborate și aprobate  în strictă conformitate cu specificul activății a fiecărei secții și nemijlocit de necesitățile instituției, astfel o atare modificare la caietul de sarcini nu de acceptă, mulțumim pentru înțelegere

description:
Buna ziua. Va rugam sa ne comunicati daca se accepta distanta interpupilara ajustabila intre 48-75mm. Considerm ca e suficiena pentru o vizualizare confortabila. Totodata rugam sa cominicati daca acceptati si microscop cu unghi de înclinare a tubului de observare de 25˚/ - se solicita obiectiv 20x cu apertura numerica de minim 0,4 si distanta de lucru de minim 0,92. Avand in vedere in diagnosticul patologic se folosesc lame de microscop acoperite cu lamele ce au grosimea de 0,1-0,2mm va rugam sa acceptati si obiectiv de 20x cu distanta de lucru de 0,8mm intrucat nu afecteaza in nici un fel modelul de vizualizare al specimentului./

- se solicita obiectiv 60x/100x draft. Va rugam sa clarificati care este obiectivul preferat din cele doua, 60x sau 100x?  /

- se solicita treaptă de reglare fină a focusării maximum 1µm. Va rugam sa acceptati si treaptă de reglare fină a focusării de 2µm.

title: Incubator nou nascuti

answer:
Specificația tehnica a incubatorului pentru nou-născuți este in corespundere cu cerințele clinice ale  instituției noastre, care este una din cele mai mari instituții medicale din țară care se ocupa de îngrijirea și tratamentul  nou-născuților prematuri  cu greutate extrem de mica la naștere (600g și mai mari). Nou-născuții prematuri cu greutate extrem de mica la naștere sunt foarte vulnerabili la condițiile clinice de supraveghere, respectiv necesită o îngrijire specială: nu numai cu medicamente, dar și dispozitive medicale de performanță înaltă,  în primul rând incubatoare.
O mulțime de producători de incubatoare pentru nou-născuți produc incubatoare  cu min.3 dulăpioare/sertare/,  chiar si incubatoarele din dotarea instituției noastre, procurate anterior,  sunt dotate cu 4 dulăpioare/sertare/.  Ele sunt strict necesare pentru păstrarea  scutecelor, consumabilelor medicale, plapumă, echipamente de prim ajutor, lucruri personale ale mamei.
Parametrul  „Anularea alarmelor fără atingere” înseamnă, că atunci,  când medicul sau asistenta medicală  efectuează   anumite proceduri  nou-născutului, și  se   declanșează o  alarmă , cu ajutorul mâinii poți anula alarma, doar cu trecerea manii in fata senzorului de mișcare, astfel micșorând riscul de infecție al copilului. Considerăm, că aceasta este o funcție foarte importantă în timpul utilizării   incubatorului pentru nou-născuți și  nu o putem exclude. Totodată, menționăm, că sunt incubatoare pentru nou-născuți cu această funcție fabricate de mai mulți producători.

description: Totusi rugam respectuos sa faceti o analiza la interpelarile noastre…. ori vom fi nevoiti sa depunem o contestatie pina maine la ora 15.00. Chiar nu exista Incubator cu minim 3 dulapuri… chiar nu intelegem ce doriti cind spuneti "anulare alarma fara atingere"... fara atingerea cui? cu ce sa o anulezi? Va multumim respectuos pentru intelegere.

answer: Buna ziua, toate cerintele tehnice sint indicate în caietul de sarcini , mulțumim

description: Buna ziua. Rugam sa explicati cerinta fata de "60x/100 draft". Ce inseamna "draft"?

title: Incubator nou nascuti

answer: Specificația tehnica a incubatorului pentru nou-născuți este in corespundere cu cerințele clinice ale  instituției noastre, care este una din cele mai mari instituții medicale din țară care se ocupa de îngrijirea și tratamentul  nou-născuților prematuri  cu greutate extrem de mica la naștere (600g și mai mari). Nou-născuții prematuri cu greutate extrem de mica la naștere sunt foarte vulnerabili la condițiile clinice de supraveghere, respectiv necesită o îngrijire specială: nu numai cu medicamente, dar și dispozitive medicale de performanță înaltă,  în primul rând incubatoare.

description:
Raspuns la "Draga operator nu iti pare ca iti inpui propriile specificații tehnice???"
Buna ziua...daca  cititi atent mesajul meu...atunci avea sa vedeti ca nu impunem parametri, dar dorim sa generalizam o specificatie care seaman izbitor cu un produs anume.

title: Procesor automat de țesuturi

answer: Buna ziua, nu se acceptă specificațiile descrise mai sus, multumesc

description:
Va rugam sa ne spuneti daca se accepta urmatoarele devieri de la caietul de sarcini:
1. Tip filtr Din cărbune : in loc de filtru, sistem automat de purifocare

title: lotul 19 - solicitare pentru clarificare

answer: Acest grup de produse include pompe de infuzie pentru volume mari cu cerințe avansate; pot avea 2 sau mai multe canale. Fluxul este calibrat în ml/oră.  Au posibilitatea de a efectua calcule a raportului medicament/doză, ce permite programarea fluxului reieșind direct din indicațiile medicului. Astfel se are în vedere cel puțin două (2 sau mai multe canale) canale,  respective instituția are nevoie de 5 pompe și nu de 10 .

description: Lotul nr. 19 solicită 5 pompe de infuzie cu pungă cu <<2 sau mai multe canale>>. Indicînd acest parametru, limitați considerabil concureența sănătoasă, deaorece doar câțiva producători oferă tipul dat de dispozitive, iar după sinecost aceste sunt mult mai scumpe decât bugetul alocat. De aceea, solicităm insistent să excludeți acest parametru discriminatoriu și să indicați, cel puțin că fie aveți nevoie de 5 pompe cu două canale, fie 10 pompe cu 1 canal la același lot. astefl fiind, veți respecta dreptul participanților la o concurență sănătoasă și veți contribui la economisirea banilor publici, excluzînd obligațiunea ofertanților de a oferi produse prea scumpe (dar cu o funcționalitate de rând).

answer:
se solicita obiectiv 60x/100x draft. Va rugam sa clarificati care este obiectivul preferat din cele doua, 60x sau 100x?

RASPUNS: obiectivul preferat ESTE 60x sau 100x (PRECIZARE - DACĂ NE REFERIM LA CEL DE 100x OBIECTIV CU POSIBILITATEA DE A EXAMINA LAMELE CITOLOGICE ȘI/SAU HISTOLOGICE FĂRĂ A FOLOSI ULEI DE IMERSIE, ADICA OBIECTIV USCAT)

description: Va rugam sa nu dati raspunsuri evazive. Noi sim ca "toate cerintele tehnice sint indicate în caietul de sarcini" . Rugam sa explicati ce inseamna cerinta "draft". multumim

title: Colorator de lamele

answer: Buna ziua, nu se acceptă deoarece este nevoie de volumul indicat în caietul de sarcini, multumesc

description: Buna ziua. Va rugam sa ne comunicati daca se accepta ca sistemul de colorare sa fie pentru minim 35 containere  in loc de 40. Va multumim anticipat.

title: întrebarea referitor la lotul nr. 18 - Monitor-analizator gaze anestezice

answer: Buna ziua, caietul de sarcini pentru dispozitivele medicale pentru Secția Reanimare inclusiv și pentru lotul 18  a fost elaborat în dependență de necesitățile instituției, astfel considerăm că sînt niște parametrii generici, multumesc

description: În lotul nr. 18 - Monitor-analizator gaze anestezice - se solicită următoarele momente: 1) BIS; 2) PPV și SVV;3) Comunicație: USB, DVI-D, RJ45. 1) Tehnologia BIS presupune utilizarea consumabilelor costisitoare de tip închis, pe când există tehnologii alternative ce presupun utiizarea consumabilelor mult mai ieftine de tip deschis, iar rezultatul afișat oferind practic aceeași informație ca și indicele afișat de BIS-modul. Solicităm să indicați BIS sau metodă alternativă. 2) Parametrii PPV (pulse pressure variation) și SVV (stroke volume variation) nu sunt măsurate direct și sunt deduse matematic din valorile parametrilor calculate din contul altor parametri ce se măsoară la direct. PPV și SVV se regăsesc doar la un număr restrâns a producătorilor de monitoare exagerat de costisitoare. Solicităm să excludeți acești parametri sau să indicași că se acceptă și alți parametri deduși matematic în urma valorilor măsurate direct. 3) Indicând că se acceptă doar monitoare ce conțin porturile USB, DVI-D, RJ45, limitați concurența. Solicităm să indicați că se acceptă sau alte tipuri/modalități de primire/transmitere a datelor, care sunt recunsocute și considerate la fel drept standarte mondiale de transmitere a datelor.   Astfel fiind, în urma celor expuse, rugăm să fie incluse modificări solicitate anterior pentru a nu limita concurența la un singur producător / cerc îngust de producători foarte scumpi, ci să contribuiți la posibilitatea participării mai multor ofertanți cu dispozitive și prețuri avantajoase. De facto, aceste modificări vor permite să aveți moitoare cu aceleași funcționalități, dar să primiți un număr mai mar de oferte avantajoase economic, contribuind la o concurență sănătoasă.

title: Buna ziua, criteriul de evaluare a ofertelor va fi : prețul cel mai mic cu coespunderea deplină a bunurilor, multumesc pentru întrebare - Răspuns din 23 Oct 2019, 12:56

answer: Buna ziua, au fost operate modificări referitor la criteriul de evaluare și adjudecare a contractului, în Anunțul de participare (Anunț de participare corectat din 23.10.2019), multumesc

title: Procesor automat de țesuturi

answer:
Buna ziua, AVEȚI PERFECTĂ DREPTATE CĂ --- tinand cont de faptul ca xilenul este un reactiv neurotoxic deosebit de periculos pentru sanantatea cadrelor medicale, si, in consecinta, conform noilor directive europene, va trebui eliminat in viitorul apropiat, va rugam sa acceptati si un procesor automat de tesuturi, care poate functiona cu succes atat cu xilen cat si fara xilen,
POATE FI ACCEPTAT ACEST TIP DE HISTOPROCESOR DAR CU RESPECTAREA CRITERIULUI DE FUNCȚIONARE CU ȘI/SAU FĂRĂ XELEN... ȘI SĂ NU FIE DEPENDENȚĂ PROCESORULUI SAU ALTOR UTILAJE PENTRU HISTOPATOLOGIE DE UN ANUMIT TIP DE REACTIVI CHIMICI/CONSUMABILE, ETC ȘI MAI ALES DE UN ANUMIT TIP DE PRODUCĂTOR, LA FEL ȘI EXCLUDEREA UTILAJULUI MEDICAL CU CHIPARE(CHIP SET...) ȘI BLOCAJ CAZ CĂ NU ESTE UN ANUMIT TIP DE REACTIVI CHIMICI/CONSUMABILE FOLOSIȚI ÎN TEHNICA HISTOPATOLOGICĂ...
VĂ MULȚUMIM ANTICIPAT

description:
Avand in vedere raspunsul la clarificarea precendenta (Răspuns (22 oct 2019, 12:02)) pentru procesorul de tesuturi, cu privire la cele doua etape pe care trebuie sa le indeplineasca un processor de tesuturi - “1. Fixarea - previne alterarea si autoliza ţesutului 2. prelucrarea histopatologică (deshidratare, clarificare şi impregnare cu parafină) aceste două etape se efectuază cu ajutorul histoprocesorului” si tinand cont de faptul ca xilenul este un reactiv neurotoxic deosebit de periculos pentru sanantatea cadrelor medicale, si, in consecinta, conform noilor directive europene, va trebui eliminat in viitorul apropiat, va rugam sa acceptati si un procesor automat de tesuturi, care poate functiona cu succes atat cu xilen cat si fara xilen, cu indeplinirea celor doua etape mentionate in raspunsul la clarificarea anterioara, respectiv “1. Fixarea - previne alterarea si autoliza ţesutului 2. prelucrarea histopatologică (deshidratare, clarificare şi impregnare cu parafină) aceste două etape se efectuază cu ajutorul histoprocesorului”.
Va multumim anticipat

title: Cerinte obligatorii

answer: Buna ziua, solicitarea înregistrării dispozitivului medical este bazată pe prevederie  art. 7 din Legea nr. 102 din 09 iunie 2017 cu privire la dispozitivele medicale . Se acceptă extrasul (screenul) din Registrul de Stat al dispozitivelor medicale  sau Declarația operatorului medical privind  obligativitatea înregistrării dispozitivului medical pînă la prima livrare (semnată și ștampilată de către participant)" Va multumim

description: Rugam sa ne informati in baza la ce solicitati "Certificat/Extras de inregistrare in Registrul de Stat al Dispozitivelor Medicale", deoarece un astfel de certificat/extras nu se elibereaza de catre AMED. Unica informatie disponibila pentru toti este Registrul de Stat al DM care poate fi accesat cu usurinta si verificat daca produsul dat a fost inregistrat de catre operatorul economic sau nu. Solicitam sa revizuiti acest punct. Multumim.

title: Preț pentru lotul 18

answer: Buna ziua, mult stimate operator economic, autoritatea contractantă a estimat   de achiziție pentru lotul 18  în dependență de acoperirea  financiare în acest sens, multumes de întrebare

description: Stimată Autoritate Contractanctă! Preț indicat pentru lotul 18 - 100 000 lei fără TVA - este prea mic pentru complectația solicitată. Doar modulul muligas care permite detectarea CO2, O2, sevofluran și izofluran costă mai mult de la producător decât suma aloctă pentru set compus din monitor, modulul respectiv, modulul BIS, restul accesoriilor și consumabilelor. Vă rugăm să indicați valorile reale atunci când înaintați solicitarea bunurilor sofisticate. Prețul estimativ pentru lotul dat trebuie să fie cel puțin de două ori mai mare. Trebuie să Vă conduceți nu doar după posibilitățile proprii, dar și din realitatea actuală. Nu poți cumpăra un Mercedes nou cu preț de Zaparojeț doar deoarece nu ai bani suficienți. Sau soilicitați parametri mai simpli, cae într-adevăr se încadrează în preț, sau indicați bugetul mai mare care corespunde cu parametri sofisticați. Mulțumim anticipat pentru înțelegere.

rationale: General change of Contract Notice

rationale: General change of Contract Notice

rationale: General change of Contract Notice

rationale: General change of Contract Notice

rationale: General change of Contract Notice

rationale: General change of Contract Notice

rationale: General change of Contract Notice

description: Contracting process is planned

description: Echipamente medicale

IMSP SCM nr.1

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

GBG-MLD SRL

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

Romedcom

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

Medist Grup SRL

streetAddress: str. Mitropolit Gavriil Banulescu-Bodoni 25, of.33

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

Ancotec-Sistem

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

Medglobalfarm

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

SC"Imunotehnomed"SRL

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

DataControl SRL

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

SRL Endo-Chirurgie

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

IM Becor SRL

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

Restmed

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

Tetis International Co. SRL

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

Pharmony

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

M-INTER-FARMA

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

ÎM „Dutchmed-M” SRL

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

Health Medical Solutions

description: mun.Chişinău

description: MOLDOVA

description: test

ECHIPAMED-PLUS

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

Profilabdiagnostic SRL

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

INTERMED

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

IM Natusana SRL

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

SRL Biosistem mld

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

Medpunkt SRL

description: mun.Chişinău

description: MOLDOVA

description: test

Zeticon

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

ICS Eyecon Medical SRL

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

description: În atenția operatorilor economici!  pct. 5 privind criteriile de eligibilitate a fost modificat

description: Contestație retrasă – motiv necunoscut

status: Contestație retrasă - procedură anulată, contestație rămasă fără obiect


title: Echipamente IT și elemente periferice

title: Planning Notice

currency: MDL

status: planning

description: Contracting process is planned

electronicInvoicingPolicy: notAllowed

statusDetails: planning

scheme: CPV

description: Echipament şi accesorii pentru computer

procurementMethod: open

eligibilityCriteria: Regulile generale privind naționalitatea și originea, precum și alte criterii de eligibilitate sunt enumerate în Ghidul practic privind procedurile de contractare a acțiunilor externe ale UE (PRAG)

mainProcurementCategory: goods

submissionMethodDetails: Lista platformelor: achizitii, ebs, licitatie, yptender

procurementMethodDetails: microValue

I.M „Piața Centrala”

roles: funder procuringEntity buyer payer

streetAddress: str.Mitropolit Varlaam 63

scheme: CUATM

description: mun.Chişinău

description: MOLDOVA

description: mun.Chişinău

scheme: MD-IDNO

legalName: I.M „Piața Centrala”

project: Echipamente și elemente periferice

description: jhjh

initiationType: tender


title: Mobilier pentru Instituția Publică de Educație Timpurie (IPET) nr. 6 din oraș

title: Planning Notice

description: Contracting process is planned

description: Mobilier

procurementMethodDetails: smallValue

Primăria mun. Orhei

roles: payer funder procuringEntity buyer

streetAddress: str. V. Mahu 160

description: Orhei

description: MOLDOVA

description: mun.Orhei

legalName: Primăria mun. Orhei

project: Mobilier pentru Instituția Publică de Etducație Timpurie (IPET) nr. 6 din oraș


title: Reparatia strazii Sfinta Vineri din s.Hulboaca, com. Gratiesti, mun.Chisinau

title: The contract/lot is not awarded

status: unsuccessful

description: Other reasons (discontinuation of procedure)

statusDetails: empty

title: “Reparatia strazii Sfinta Vineri din s.Hulboaca, com. Gratiesti,mun.Chisinau”

status: cancelled

variantPolicy: notAllowed

description: Lucrări de construcţii complete sau parţiale şi lucrări publice

streetAddress: str.Stefan cel Mare 72

description: mun.Chişinău

description: MOLDOVA

description: s.Grătieşti

title: Planning Notice

title: documentatia standard str. sfinta vineri hulboaca.semnat.pdf

description: Documentatia standard

documentType: biddingDocuments

title: anunt_de_participare.la cop strada sf. vineri hulboaca.semnat.pdf

description: Anunt de participare la COP

documentType: tenderNotice

description: Anunt de participare la COP

title: duae.str. sfinta vineri.semnat.pdf

description: DUAE

title: duae.str. sfinta vineri.semnat.pdf

description: DUAE

title: anunt_de_participare.la cop strada sf. vineri hulboaca.docx

description: Anunt de participare la COP

description: Anunt de participare   la COP

description: Documentatia standard

description: Anunt de participare   la COP

rationale: General change of Contract Notice

rationale: General change of Contract Notice

rationale: S-au depistat erori la demararea achizitiei

description: Contracting process is planned

awardCriteria: priceOnly

mainProcurementCategory: works

Primaria comunei Gratiesti

description: mun.Chişinău

description: MOLDOVA

description: s.Grătieşti

legalName: Primaria comunei Gratiesti


title: Servicii de reparare şi de întreţinere a echipamentului medical şi de precizie

title: Planning Notice

description: Contracting process is planned

mainProcurementCategory: services

procurementMethodDetails: openTender

IMSP CRDM

streetAddress: str. C. Virnav 13

description: mun.Chişinău

description: MOLDOVA

scheme: other

description: locality

scheme: COFOG

legalName: IMSP CRDM