name: Ministerio de Salud Pública y Bienestar Social (MSPBS)

title: 1 - 1 - HOSPITAL REGIONAL DE ENCARNACION

currency: PYG

status: active

statusDetails: activo

openContractType: cantidad

title: 2 - 1 - HOSPITAL REGIONAL DE ENCARNACION

title: 3 - 1 - HOSPITAL REGIONAL DE ENCARNACION

title: 4 - 1 - HOSPITALES DISTRITALES - ITAPUA

title: 5 - 1 - LOTE 5

title: 6 - 1 - LOTE 6

title: 7 - 1 - LOTE 7

title: 8 - 1 - LOTE 8

title: 100 - 1 - Lote 100

title: 9 - 1 - LOTE 9

title: 101 - 1 - Lote 101

title: 102 - 1 - LOTE 102

title: 103 - 1 - LOTE 103

title: 104 - 1 - LOTE 104

title: 105 - 1 - LOTE 105

title: 106 - 1 - LOTE 106

title: 107 - 1 - LOTE 107

title: 108 - 1 - LOTE 108

title: 109 - 1 - LOTE 109

title: 110 - 1 - LOTE 110

title: 111 - 1 - LOTE 111

title: 112 - 1 - LOTE 112

title: 113 - 1 - LOTE 113

title: 10 - 1 - LOTE 10

title: 114 - 1 - Lote 114

title: 115 - 1 - Lote 115

title: 11 - 1 - LOTE 11

title: 12 - 1 - LOTE 12

title: 13 - 1 - LOTE 13

title: 116 - 1 - LOTE 116

title: 117 - 1 - Lote 117

title: 118 - 1 - Lote 118

title: 14 - 1 - LOTE 14

title: 119 - 1 - Lote 119

title: 120 - 1 - Lote 120

title: 121 - 1 - Lote 121

title: 122 - 1 - Lote 122

title: 123 - 1 - Lote 123

title: 124 - 1 - Lote 124

title: 125 - 1 - Lote 125

title: 126 - 1 - Lote 126

title: 127 - 1 - Lote 127

title: 128 - 1 - Lote 128

title: 129 - 1 - Lote 129

title: 130 - 1 - Lote 130

title: 131 - 1 - Lote 131

title: 132 - 1 - Lote 132

title: 133 - 1 - Lote 133

title: 134 - 1 - Lote 134

title: 135 - 1 - Lote 135

title: 136 - 1 - Lote 136

title: 137 - 1 - Lote 137

title: 138 - 1 - Lote 138

title: 139 - 1 - Lote 139

title: 140 - 1 - Lote 140

title: 141 - 1 - Lote 141

title: 142 - 1 - Lote 142

title: 143 - 1 - Lote 143

title: 144 - 1 - Lote 144

title: 145 - 1 - Lote 145

title: 146 - 1 - Lote 146

title: 147 - 1 - Lote 147

title: 148 - 1 - Lote 148

title: 149 - 1 - Lote 149

title: 150 - 1 - Lote 150

title: 151 - 1 - Lote 151

title: 152 - 1 - Lote 152

title: 153 - 1 - Lote 153

title: 154 - 1 - Lote 154

title: 155 - 1 - Lote 155

title: 156 - 1 - Lote 156

title: 157 - 1 - Lote 157

title: 158 - 1 - Lote 158

title: 159 - 1 - Lote 159

title: 160 - 1 - Lote 160

title: 161 - 1 - Lote 161

title: 162 - 1 - Lote 162

title: 80 - 1 - Lote 80

title: 81 - 1 - Lote 81

title: 82 - 1 - Lote 82

title: 83 - 1 - Lote 83

title: 84 - 1 - Lote 84

title: 85 - 1 - Lote 85

title: 86 - 1 - Lote 86

title: 87 - 1 - Lote 87

title: 88 - 1 - Lote 88

title: 89 - 1 - Lote 89

title: 90 - 1 - Lote 90

title: 91 - 1 - Lote 91

title: 92 - 1 - Lote 92

title: 93 - 1 - Lote 93

title: 94 - 1 - Lote 94

title: 95 - 1 - Lote 95

title: 96 - 1 - Lote 96

title: 97 - 1 - Lote 97

title: 98 - 1 - Lote 98

title: 99 - 1 - Lote 99

title: 16 - 1 - LOTE 16

title: 15 - 1 - LOTE 15

title: 17 - 1 - LOTE 17

title: 18 - 1 - LOTE 18

title: 19 - 1 - LOTE 19

title: 20 - 1 - LOTE 20

title: 21 - 1 - LOTE 21

title: 22 - 1 - LOTE 22

title: 23 - 1 - LOTE 23

title: 24 - 1 - LOTE 24

title: 25 - 1 - LOTE 25

title: 26 - 1 - LOTE 26

title: 27 - 1 - LOTE 27

title: 28 - 1 - LOTE 28

title: 29 - 1 - LOTE 29

title: 30 - 1 - LOTE 30

title: 31 - 1 - LOTE 31

title: 50 - 1 - LOTE 50

title: 32 - 1 - LOTE 32

title: 51 - 1 - LOTE 51

title: 33 - 1 - LOTE 33

title: 52 - 1 - LOTE 52

title: 53 - 1 - LOTE 53

title: 54 - 1 - LOTE 54

title: 34 - 1 - LOTE 34

title: 35 - 1 - LOTE 35

title: 55 - 1 - LOTE 55

title: 56 - 1 - LOTE 56

title: 57 - 1 - LOTE 57

title: 58 - 1 - LOTE 58

title: 59 - 1 - LOTE 59

title: 60 - 1 - LOTE 60

title: 61 - 1 - LOTE 61

title: 62 - 1 - LOTE 62

title: 63 - 1 - LOTE 63

title: 64 - 1 - LOTE 64

title: 65 - 1 - LOTE 65

title: 66 - 1 - LOTE 66

title: 67 - 1 - LOTE 67

title: 68 - 1 - LOTE 68

title: 69 - 1 - LOTE 69

title: 70 - 1 - LOTE 70

title: 71 - 1 - LOTE 71

title: 72 - 1 - LOTE 72

title: 73 - 1 - LOTE 73

title: 74 - 1 - LOTE 74

title: 75 - 1 - LOTE 75

title: 76 - 1 - LOTE 76

title: 77 - 1 - LOTE 77

title: 36 - 1 - LOTE 36

title: 78 - 1 - LOTE 78

title: 79 - 1 - LOTE 79

title: 37 - 1 - LOTE 37

title: 38 - 1 - LOTE 38

title: 39 - 1 - LOTE 39

title: 40 - 1 - LOTE 40

title: 41 - 1 - LOTE 41

title: 42 - 1 - LOTE 42

title: 43 - 1 - LOTE 43

title: 44 - 1 - LOTE 44

title: 45 - 1 - LOTE 45

title: 46 - 1 - LOTE 46

title: 47 - 1 - LOTE 47

title: 48 - 1 - LOTE 48

title: 49 - 1 - LOTE 49

description: KIT SOLUCION CONTADOR HEMATOLÓGICO

description: Kits Solucion Contador Hematologico

description: Kits de ensayo hematologicos o suministros

description: GLICEMIA

description: Glicemia

description: Reactivos o soluciones quimicos

description: REACTIVO PARA UREA

description: Reactivo para urea

description: Reactivos o soluciones quimicos

description: REACTIVO PARA CREATININA

description: Reactivo para creatinina

description: Reactivos o soluciones quimicos

description: REACTIVO PARA ÁCIDO ÚRICO

description: Reactivo para acido urico

description: Reactivos o soluciones quimicos

description: HDL COLESTEROL (Directo, sin precipitación)

description: HDL colesterol

description: Reactivos o soluciones quimicos

description: REACTIVO PARA COLESTEROL

description: Reactivo para colesterol

description: Reactivos o soluciones quimicos

description: REACTIVO PARA TRIGLICÉRIDOS

description: Reactivo para trigliceridos

description: Reactivos o soluciones quimicos

description: GOT

description: GOT

description: Reactivos o soluciones quimicos

description: GPT

description: GPT

description: Reactivos o soluciones quimicos

description: FOSFATASA ALCALINA

description: Fosfatasa alcalina

description: Reactivos o soluciones quimicos

description: BILIRRUBINA (TOTAL)

description: Bilirrubina

description: Reactivos o soluciones quimicos

description: BILIRRUBINA (DIRECTA)

description: Bilirrubina

description: Reactivos o soluciones quimicos

description: GAMMA GT

description: Gamma GT

description: Reactivos, soluciones o tinturas microbiologicos o bacteriologicos

description: Lipasa reactivo

description: Lipasa reactivo

description: Reactivos o soluciones quimicos

description: ALFA AMILASA

description: Alfa amilasa

description: Reactivos, soluciones o tinturas citologicos

description: LDH

description: LDH

description: Reactivos o soluciones quimicos

description: PROTEINAS TOTALES

description: Proteinas totales

description: Reactivos de analizadores quimicos

description: ALBÚMINA

description: Albumina

description: Reactivos o soluciones quimicos

description: CALCIO

description: Calcio

description: Analizadores de calcio

description: FÓSFORO

description: Fosforo

description: Reactivos, soluciones o tinturas citologicos

description: MAGNESIO

description: Magnesio

description: Reactivos o soluciones quimicos

description: CK MB

description: CK MB

description: Reactivos o soluciones quimicos

description: CK TOTAL

description: CK-TOTAL

description: Analizadores de calcio

description: PROTEINAS (Proteinuria)

description: Proteinas

description: Reactivos de analizadores quimicos

description: PROTEINAS LCR

description: Proteina LCR

description: Reactivos de analizadores de proteinas

description: HEMOGLOBINA GLICOSILADA (A1c)

description: Hemoglobina Glicosilada

description: Reactivos o soluciones de coagulacion

description: Reactivos o soluciones quimicos

description: Reactivos o soluciones quimicos

description: Anti Hav-IgM (Hepatitis A)

description: Anti Hav-IgM (Hepatitis A)

description: Analizadores inmunologicos

description: Reactivo antigeno de superficie para hepatitis B.

description: Reactivo antigeno de superficie para hepatitis B.

description: Reactivos de analizadores de banco de sangre

description: Reactivo para anticuerpo Core IgG  IgM-HB

description: Reactivo para anticuerpo Core IgG + IgM-HB

description: Reactivos de analizadores inmunologicos

description: Hepatitis C

description: Hepatitis C

description: Reactivos de analizadores de banco de sangre

description: REACTIVO PARA TOXOPLASMOSIS IgG

description: Reactivo para Toxomplasmosis IgG

description: Reactivos, soluciones o tinturas citologicos

description: TOXO IgM reactivo

description: Toxo IgM reactivo

description: Analizadores inmunologicos

description: RUBEOLA IgG

description: Rubeola IgG

description: Analizadores inmunologicos

description: RUBEOLA IgM

description: Rubeola IgM

description: Analizadores inmunologicos

description: Kits-Deteccion Citomegalovirus IgG

description: Kits-Deteccion Citomegalovirus IgG

description: Kits de ensayo inmunologicos o serologicos, o suministros

description: Kits-Deteccion Citomegalovirus IgM

description: Kits-Deteccion Citomegalovirus IgM

description: Kits de ensayo inmunologicos o serologicos, o suministros

description: IgA

description: IgA

description: Reactivos o soluciones quimicos

description: TROPONINA (I) cuantitativo

description: Troponina

description: Reactivos o soluciones quimicos

description: DIMERO D

description: Dimero D

description: Reactivos o soluciones quimicos

description: T3 REACTIVO

description: T3 Reactivo

description: Calibradores de deformacion

description: KITS - DETERMINACIÓN DE T4

description: Kits- Determinacion de T4

description: Kits de ensayo inmunologicos o serologicos, o suministros

description: FT3 REACTIVO

description: FT3 reactivo

description: Calibradores de deformacion

description: FT4 REACTIVO

description: FT4 reactivo

description: Calibradores de deformacion

description: TSH Reactivo

description: TSH Reactivo

description: Reactivos o soluciones quimicos

description: VITAMINA B12

description: Vitamina B12

description: Analizadores inmunologicos

description: REACTIVO ACIDO FOLICO

description: Reactivo Acido Folico

description: Reactivos o soluciones quimicos

description: HIERRO SERICO

description: Hierro serico

description: Reactivos o soluciones quimicos

description: TRANSFERRINA

description: Transferrina

description: Reactivos o soluciones quimicos

description: FERRITINA

description: Ferritina

description: Reactivos o soluciones quimicos

description: ANTIGENO PROSTATICO ESPECIFICO (PAS) - Total

description: Antigeno Prostatico Especifico (PAS)

description: Reactivos o soluciones quimicos

description: ANTIGENO PROSTATICO ESPECIFICO (PAS) - Libre

description: Antigeno Prostatico Especifico (PAS)

description: Reactivos o soluciones quimicos

description: CA 125

description: CA 125

description: Analizadores de calcio

description: CA 15-3

description: CA 15 - 3

description: Analizadores de calcio

description: CA 19-9

description: CA 19 - 9

description: Analizadores de calcio

description: CEA

description: CEA

description: Reactivos de analizadores quimicos

description: ALFA FETO PROTEINA

description: Alfa feto proteina

description: Reactores, kits o sustratos cromogenicos de deteccion de proteinas

description: AC. ANTI TIROGLOBULINA

description: Ac. Anti Tiroglobulina

description: Reactivos de analizadores inmunologicos

description: TPO

description: TPO

description: Reactivos o soluciones quimicos

description: REACTIVO PARA B HCG

description: Reactivo para BHCG

description: Reactivos o soluciones quimicos

description: CORTISOL

description: Cortisol

description: Reactivos o soluciones quimicos

description: ESTRADIOL

description: Estradiol

description: Reactivos o soluciones quimicos

description: FSH hormona folículo estimulante

description: FSH hormona foliculo estimulante

description: Reactivos de analizadores inmunologicos

description: LH hormona luteinizante

description: LH hormona luteinizante

description: Reactivos de analizadores inmunologicos

description: PROGESTERONA

description: Progesterona

description: Reactivos de analizadores quimicos

description: KITS - DETERMINACION  DE PROLACTINA

description: Kits- Determinacion de Prolactina

description: Kits de ensayo inmunologicos o serologicos, o suministros

description: TESTOSTERONA REACTIVO

description: Testosterona Reactivo

description: Reactivos y standares para unidad de control de calidad

description: DHEA-SO4 dehidroepiandrosterona sulfato

description: DHEA-SO4 dehidroepiandrosterona sulfato

description: Reactivos, soluciones o tinturas inmunologicos o serologicos

description: INSULINA

description: Insulina

description: Reactivos o soluciones quimicos

description: PROCALCITONINA

description: Procalcitonina

description: Reactivos o soluciones quimicos

description: VITAMINA D

description: Vitamina D Reactivo

description: Reactivos de analizadores quimicos

description: PCR TURBIDIMETRIA

description: Proteina C. Reactivo (P.C.R.)

description: Reactivos de analizadores quimicos

description: Electrolitos (Na, K, Cl)

description: Electrolitos (Na, K, Cl)

description: Reactivos de analizadores de gases sanguineos

description: KIT PARA EQUIPO DE ELECTROLITOS Y GASÓMETRO

description: Kit para equipo de Electrolitos y Gasometro

description: Analizadores de gases sanguineos

description: REACTIVO TP

description: Reactivo TP

description: Reactivos de analizadores de coagulacion

description: REACTIVO TTPA

description: Reactivo TTPA

description: Reactivos de analizadores de coagulacion

description: REACTIVO DE FIBRINÓGENO

description: Reactivo de Fibrinogeno

description: Reactivos, soluciones o tinturas citologicos

description: Antigeno Dengue

description: Antigeno Dengue

description: Reactivos, soluciones o tinturas microbiologicos o bacteriologicos

description: Test  Rapido SARS Cov 2

description: Test  Rapido SARS Cov 2

description: Reactivos o soluciones quimicos

description: KIT SOLUCION CONTADOR HEMATOLÓGICO

description: Kits Solucion Contador Hematologico

description: Kits de ensayo hematologicos o suministros

description: GLICEMIA

description: Glicemia

description: Reactivos o soluciones quimicos

description: REACTIVO PARA UREA

description: Reactivo para urea

description: Reactivos o soluciones quimicos

description: REACTIVO PARA CREATININA

description: Reactivo para creatinina

description: Reactivos o soluciones quimicos

description: REACTIVO PARA ÁCIDO ÚRICO

description: Reactivo para acido urico

description: Reactivos o soluciones quimicos

description: HDL COLESTEROL (Directo, sin precipitación)

description: HDL colesterol

description: Reactivos o soluciones quimicos

description: REACTIVO PARA COLESTEROL

description: Reactivo para colesterol

description: Reactivos o soluciones quimicos

description: REACTIVO PARA TRIGLICÉRIDOS

description: Reactivo para trigliceridos

description: Reactivos o soluciones quimicos

description: GOT

description: GOT

description: Reactivos o soluciones quimicos

description: GPT

description: GPT

description: Reactivos o soluciones quimicos

description: FOSFATASA ALCALINA

description: Fosfatasa alcalina

description: Reactivos o soluciones quimicos

description: BILIRRUBINA (TOTAL)

description: Bilirrubina

description: Reactivos o soluciones quimicos

description: BILIRRUBINA (DIRECTA)

description: Bilirrubina

description: Reactivos o soluciones quimicos

description: GAMMA GT

description: Gamma GT

description: Reactivos, soluciones o tinturas microbiologicos o bacteriologicos

description: ALFA AMILASA

description: Alfa amilasa

description: Reactivos, soluciones o tinturas citologicos

description: LDH

description: LDH

description: Reactivos o soluciones quimicos

description: PROTEINAS TOTALES

description: Proteinas totales

description: Reactivos de analizadores quimicos

description: ALBÚMINA

description: Albumina

description: Reactivos o soluciones quimicos

description: CALCIO

description: Calcio

description: Analizadores de calcio

description: FÓSFORO

description: Fosforo

description: Reactivos, soluciones o tinturas citologicos

description: MAGNESIO

description: Magnesio

description: Reactivos o soluciones quimicos

description: CK MB

description: CK MB

description: Reactivos o soluciones quimicos

description: CK TOTAL

description: CK-TOTAL

description: Analizadores de calcio

description: PROTEINAS (Proteinuria)

description: Proteinas

description: Reactivos de analizadores quimicos

description: Electrolitos (Na, K, Cl)

description: Electrolitos (Na, K, Cl)

description: Reactivos de analizadores de gases sanguineos

description: HEMOGLOBINA GLICOSILADA

description: Hemoglobina Glicosilada

description: Reactivos o soluciones de coagulacion

description: DIMERO D

description: Dimero D

description: Reactivos o soluciones quimicos

description: FERRITINA

description: Ferritina

description: Reactivos o soluciones quimicos

description: Antígeno Prostático Específico (PAS) - Total

description: Antigeno Prostatico Especifico (PAS)

description: Reactivos o soluciones quimicos

description: Antígeno Prostático Específico (PAS) - Libre

description: Antigeno Prostatico Especifico (PAS)

description: Reactivos o soluciones quimicos

description: TROPONINA CUANTITATIVA

description: Troponina

description: Reactivos o soluciones quimicos

description: Reactivo para Toxoplasmosis IgG

description: Reactivo para Toxomplasmosis IgG

description: Reactivos, soluciones o tinturas citologicos

description: Toxo IgM reactivo

description: Toxo IgM reactivo

description: Analizadores inmunologicos

description: Rubeola IgG

description: Rubeola IgG

description: Analizadores inmunologicos

description: Rubeola IgM

description: Rubeola IgM

description: Analizadores inmunologicos

description: Kits-Detección Citomegalovirus IgG

description: Kits-Deteccion Citomegalovirus IgG

description: Kits de ensayo inmunologicos o serologicos, o suministros

description: Kits-Detección Citomegalovirus IgM

description: Kits-Deteccion Citomegalovirus IgM

description: Kits de ensayo inmunologicos o serologicos, o suministros

description: Kits - Determinación de T4

description: Kits- Determinacion de T4

description: Kits de ensayo inmunologicos o serologicos, o suministros

description: FT3 reactivo

description: FT3 reactivo

description: Calibradores de deformacion

description: FT4 reactivo

description: FT4 reactivo

description: Calibradores de deformacion

description: TSH Reactivo

description: TSH Reactivo

description: Reactivos o soluciones quimicos

description: CA 125

description: CA 125

description: Analizadores de calcio

description: CA 15-3

description: CA 15 - 3

description: Analizadores de calcio

description: CA 19-9

description: CA 19 - 9

description: Analizadores de calcio

description: CEA

description: CEA

description: Reactivos de analizadores quimicos

description: Chagas IgM

description: Chagas

description: Reactivos de analizadores de banco de sangre

description: CHAGAS IgG

description: Chagas

description: Reactivos de analizadores de banco de sangre

description: Kit para equipo de electrolitos y gasómetro

description: Kit para equipo de Electrolitos y Gasometro

description: Analizadores de gases sanguineos

description: REACTIVO TP

description: Reactivo TP

description: Reactivos de analizadores de coagulacion

description: REACTIVO TTPA

description: Reactivo TTPA

description: Reactivos de analizadores de coagulacion

description: REACTIVO DE FIBRINÓGENO

description: Reactivo de Fibrinogeno

description: Reactivos, soluciones o tinturas citologicos

description: GLICEMIA

description: Glicemia

description: Reactivos o soluciones quimicos

description: REACTIVO PARA UREA

description: Reactivo para urea

description: Reactivos o soluciones quimicos

description: REACTIVO PARA CREATININA

description: Reactivo para creatinina

description: Reactivos o soluciones quimicos

description: REACTIVO PARA ÁCIDO ÚRICO

description: Reactivo para acido urico

description: Reactivos o soluciones quimicos

description: HDL COLESTEROL

description: HDL colesterol

description: Reactivos o soluciones quimicos

description: REACTIVO PARA COLESTEROL

description: Reactivo para colesterol

description: Reactivos o soluciones quimicos

description: REACTIVO PARA TRIGLICÉRIDOS

description: Reactivo para trigliceridos

description: Reactivos o soluciones quimicos

description: GOT

description: GOT

description: Reactivos o soluciones quimicos

description: GPT

description: GPT

description: Reactivos o soluciones quimicos

description: FOSFATASA ALCALINA

description: Fosfatasa alcalina

description: Reactivos o soluciones quimicos

description: BILIRRUBINA

description: Bilirrubina

description: Reactivos o soluciones quimicos

description: GAMMA GT

description: Gamma GT

description: Reactivos, soluciones o tinturas microbiologicos o bacteriologicos

description: PROTEINAS TOTALES

description: Proteinas totales

description: Reactivos de analizadores quimicos

description: ALBÚMINA

description: Albumina

description: Reactivos o soluciones quimicos

description: CALCIO

description: Calcio

description: Analizadores de calcio

description: FÓSFORO

description: Fosforo

description: Reactivos, soluciones o tinturas citologicos

description: MAGNESIO

description: Magnesio

description: Reactivos o soluciones quimicos

description: PROTEINAS (Proteinuria)

description: Proteinas

description: Reactivos de analizadores quimicos

description: Hemoglobina Glicosilada

description: Hemoglobina Glicosilada

description: Reactivos o soluciones de coagulacion

description: Anti A Monoclonal

description: Anti A Monoclonal

description: Reactivos o soluciones quimicos

description: Anti B Monoclonal

description: Anti B Monoclonal

description: Reactivos o soluciones quimicos

description: Anti D Monoclonal

description: Anti D Monoclonal

description: Reactivos o soluciones quimicos

description: Suero de coombs

description: Suero de coombs

description: Reactivos de analizadores de banco de sangre

description: Gradilla para tubos

description: Gradilla para tubos

description: Gradilla o porta tubos

description: Formol (formaldehido) Solucion

description: Formol (formaldehido) Solucion

description: Antisepticos de formaldehido

description: Jeringa desechable con aguja

description: Jeringa desechable con aguja

description: Jeringas medicas con aguja

description: Jeringa desechable con aguja

description: Jeringa desechable con aguja

description: Jeringas medicas con aguja

description: Jeringa desechable con aguja

description: Jeringa desechable con aguja

description: Jeringas medicas con aguja

description: Jeringa desechable con aguja

description: Jeringa desechable con aguja

description: Jeringas medicas con aguja

description: Jeringa desechable con aguja

description: Jeringa desechable con aguja

description: Jeringas medicas con aguja

description: Lamina cubreobjeto

description: Lamina cubreobjeto

description: Cubreobjetos

description: Lamina portaobjeto

description: Lamina portaobjeto

description: Porta objetos para laboratorio

description: Laminas Esmeriladas

description: Laminas Esmeriladas

description: Probetas graduadas de laboratorio y Erlenmeyer

description: Lanceta desechable

description: Lanceta desechable

description: Lancetas

description: Transportador para Muestra de Laboratorio

description: Transportador para Muestra de Laboratorio

description: Tubos o contenedores de almacenamiento

description: Film para laboratorio

description: Film para laboratorio

description: Pelicula selladora de laboratorio o papel film M

description: Test Rapido para HIV

description: Test Rapido para HIV

description: Kits de ensayos rapidos

description: Papel indicador universal pH

description: Papel indicador universal pH

description: Tiras para ensayos de pH

description: Soporte para Tincion de Lamina

description: Soporte para Tincion de Lamina

description: Soportes para portaobjetos

description: Sifilis

description: Sifilis - Test Rapido

description: Kits de ensayos rapidos

description: Pipeta automatica con volumen variable

description: Pipeta automatica con volumen variable

description: Pipetas serologicas

description: Pipeta automatica con volumen variable

description: Pipeta automatica con volumen variable

description: Pipetas serologicas

description: Pipeta automatica con volumen variable

description: Pipeta automatica con volumen variable

description: Pipetas serologicas

description: Pipeta automatica con volumen fijo

description: Pipeta automatica con volumen fijo

description: Pipetas serologicas

description: Pipeta automatica con volumen fijo

description: Pipeta automatica con volumen fijo

description: Pipetas serologicas

description: Pipeta automatica con volumen fijo

description: Pipeta automatica con volumen fijo

description: Pipetas serologicas

description: Pipeta automatica con volumen fijo

description: Pipeta automatica con volumen fijo

description: Pipetas serologicas

description: Pipeta automatica con volumen fijo

description: Pipeta automatica con volumen fijo

description: Pipetas serologicas

description: Pipeta automatica con volumen fijo

description: Pipeta automatica con volumen fijo

description: Pipetas serologicas

description: Puntas desechables para pipetas automaticas

description: Puntas desechables para pipetas automaticas

description: Puntas de pipeta universales

description: Chagas Test Rapido

description: Chagas Test Rapido

description: Kits de ensayos rapidos

description: Puntas desechables para pipetas automaticas

description: Puntas desechables para pipetas automaticas

description: Puntas de pipeta universales

description: Puntas desechables para pipetas automaticas

description: Puntas desechables para pipetas automaticas

description: Puntas de pipeta universales

description: Puntas desechables para pipetas automaticas

description: Puntas desechables para pipetas automaticas

description: Puntas de pipeta universales

description: Test Rapido para Hepatitits A

description: Test Rapido para Hepatitits A

description: Kits de ensayos rapidos

description: Puntas desechables para pipetas automaticas

description: Puntas desechables para pipetas automaticas

description: Puntas de pipeta universales

description: Test Rapido para Hepatitis B

description: Test Rapido para Hepatitis B

description: Kits de ensayos rapidos

description: Pipeta para eritrosidimentacion de vidrio

description: Pipeta para eritrosidimentacion de vidrio

description: Pipetas serologicas

description: Pipeta para eritrosidimentacion de vidrio

description: Pipeta para eritrosidimentacion de vidrio

description: Pipetas serologicas

description: Pipeta Pasteur

description: Pipeta Pasteur

description: Pipetas Pasteur o de transferencia

description: Reactivo antigeno V.D.R.L.

description: Reactivo antigeno V.D.R.L.

description: Reactivos o soluciones quimicos

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Pipetas serologicas

description: Placa para tipificacion

description: Placa para tipificacion

description: Placas o cajas Petri

description: Placa para VDRL

description: Placa para VDRL

description: Microplacas

description: Pinza para laboratorio

description: Pinza para laboratorio

description: Pinzas de laboratorio

description: Probeta

description: Probeta

description: Probetas graduadas de laboratorio y Erlenmeyer

description: Probeta

description: Probeta

description: Probetas graduadas de laboratorio y Erlenmeyer

description: Probeta

description: Probeta

description: Probetas graduadas de laboratorio y Erlenmeyer

description: Probeta

description: Probeta

description: Probetas graduadas de laboratorio y Erlenmeyer

description: Probeta

description: Probeta

description: Probetas graduadas de laboratorio y Erlenmeyer

description: Probeta

description: Probeta

description: Probetas graduadas de laboratorio y Erlenmeyer

description: Probeta

description: Probeta

description: Probetas graduadas de laboratorio y Erlenmeyer

description: Propipeta

description: Propipeta

description: Pipetas serologicas

description: Testigos de esterilización para autoclave

description: Testigos de esterilización para autoclave

description: Cintas indicadores de esterilizacion

description: Testigos de esterilización para estufa

description: Testigos de esterilización para estufa

description: Cintas indicadores de esterilizacion

description: Termometro

description: Termometro

description: Termometros para incubadoras de laboratorio

description: Tubo de centrifuga conicos de plastico

description: Tubo de centrifuga conicos de plastico

description: Tubos centrifugos y microcentrifugos

description: Tubo de centrifuga conicos de plastico

description: Tubo de centrifuga conicos de plastico

description: Tubos centrifugos y microcentrifugos

description: Tubo de centrifuga conicos de plastico

description: Tubo de centrifuga conicos de plastico

description: Tubos centrifugos y microcentrifugos

description: Tubos de ensayo de vidrio

description: Tubos de ensayo de vidrio

description: Tubos de ensayo multiproposito o generales

description: Tubos de ensayo de vidrio

description: Tubos de ensayo de vidrio

description: Tubos de ensayo multiproposito o generales

description: Tubos de ensayo de vidrio

description: Tubos de ensayo de vidrio

description: Tubos de ensayo multiproposito o generales

description: Tubos de ensayo de vidrio

description: Tubos de ensayo de vidrio

description: Tubos de ensayo multiproposito o generales

description: Tubos de ensayo de vidrio

description: Tubos de ensayo de vidrio

description: Tubos de ensayo multiproposito o generales

description: Tubos de Khan

description: Tubos de Khan

description: Tubos de ensayo multiproposito o generales

description: Vaso precipitado

description: Vaso precipitado

description: Vasos de precipitados de laboratorio

description: Vaso precipitado

description: Vaso precipitado

description: Vasos de precipitados de laboratorio

description: Vaso precipitado

description: Vaso precipitado

description: Vasos de precipitados de laboratorio

description: Vaso precipitado

description: Vaso precipitado

description: Vasos de precipitados de laboratorio

description: Tubo de Microcentrifuga

description: Tubo de Microcentrifuga

description: Tubos de ensayo multiproposito o generales

description: Tubo de Microcentrifuga

description: Tubo de Microcentrifuga

description: Tubos de ensayo multiproposito o generales

description: Fucsina Pro analisis

description: Fucsina Pro analisis

description: Reactivos y standares para unidad de control de calidad

description: Fenol Cristalizado

description: Fenol Cristalizado

description: Reactivos, soluciones o tinturas microbiologicos o bacteriologicos

description: Palitos de madera

description: Palitos de madera

description: Equipos de primer auxilio de servicios medicos de urgencia

description: Solucion Buffer de PH

description: Solucion Buffer de PH

description: Reactivos y standares para unidad de control de calidad

description: Pipeta Pasteur

description: Pipeta Pasteur

description: Pipetas Pasteur o de transferencia

description: Embudo de vidrio

description: Embudo de vidrio

description: Embudos de laboratorio

description: Hisopo con medio de transporte

description: Hisopo con medio de transporte

description: Tampones quirurgicos

description: Erlemmeyer de vidrio

description: Erlemmeyer de vidrio

description: Frascos o vasos de laboratorio

description: Erlemmeyer de vidrio

description: Erlemmeyer de vidrio

description: Frascos o vasos de laboratorio

description: Erlemmeyer de vidrio

description: Erlemmeyer de vidrio

description: Frascos o vasos de laboratorio

description: Bolsita colectora de orina para analisis

description: Bolsita colectora de orina para analisis

description: Contenedores de recogida de orina

description: Frascos Plasticos de polipropileno autoclavables c/ tapa

description: Frascos Plasticos de polipropileno autoclavables c/ tapa

description: Frascos o vasos de laboratorio

description: Frascos Plasticos de polipropileno autoclavables c/ tapa

description: Frascos Plasticos de polipropileno autoclavables c/ tapa

description: Frascos o vasos de laboratorio

description: Frascos Plasticos de polipropileno autoclavables c/ tapa

description: Frascos Plasticos de polipropileno autoclavables c/ tapa

description: Frascos o vasos de laboratorio

description: Goma de ligar

description: Goma de ligar

description: Abrazaderas de laboratorio

description: Gradillas de esponja para tubos

description: Gradillas de esponja para tubos

description: Placas, cajas o soportes con revestimiento para cultivo de tejidos

description: Gradilla para tubos

description: Gradilla para tubos

description: Gradilla o porta tubos

description: Gradilla para tubos

description: Gradilla para tubos

description: Gradilla o porta tubos

description: Gradilla para tubos

description: Gradilla para tubos

description: Gradilla o porta tubos

description: Gradilla para tubos

description: Gradilla para tubos

description: Gradilla o porta tubos

description: Gradilla para tubos

description: Gradilla para tubos

description: Gradilla o porta tubos

description: Gradilla para tubos

description: Gradilla para tubos

description: Gradilla o porta tubos

description: Soporte para pipetas de eritrosedimentacion

description: Soporte para pipetas de eritrosedimentacion

description: Soportes o estantes para pipetas

description: Gradilla para tubos

description: Gradilla para tubos

description: Gradilla o porta tubos

description: Gradilla para tubos

description: Gradilla para tubos

description: Gradilla o porta tubos

description: Asto

description: Asto

description: Reactivos, soluciones o tinturas microbiologicos o bacteriologicos

description: Factor Reumatoide

description: Factor Reumatoide

description: Reactivos, soluciones o tinturas microbiologicos o bacteriologicos

description: Proteina C. Reactivo (P.C.R.)

description: Proteina C. Reactivo (P.C.R.)

description: Reactivos de analizadores quimicos

description: Reactivo antigeno V.D.R.L.

description: Reactivo antigeno V.D.R.L.

description: Reactivos o soluciones quimicos

description: Reactivo para Diagnostico de Embarazo

description: Reactivo para Diagnostico de Embarazo

description: Reactivos, soluciones o tinturas citologicos

description: Tiras reactivas para Leishmaniasis

description: Tiras reactivas para Leishmaniasis

description: Tiras o papeles para ensayos quimicos

description: Test Rapido

description: Test Rapido - Virus Respiratorio

description: Kits de ensayos rapidos

description: IgM Dengue

description: IgM Dengue

description: Reactivos o soluciones quimicos

description: IgG Dengue

description: IgG Dengue

description: Reactivos o soluciones quimicos

description: Test multidrogas para orina

description: Test multidrogas para orina

description: Equipo de pruebas de narcoticos

description: Test multidrogas para orina

description: Test multidrogas para orina

description: Equipo de pruebas de narcoticos

description: Kit Antigenos Febriles

description: Kit Antigenos Febriles

description: Reactivos, soluciones o tinturas microbiologicos o bacteriologicos

description: Test para mononucleosis infecciosa con controles + y

description: Test para mononucleosis infecciosa con controles + y -

description: Reactivos o soluciones quimicos

description: Test Rapido para Rotavirus

description: Test Rapido para Rotavirus

description: Kits de ensayos rapidos

description: Test para sangre oculta

description: Test para sangre oculta

description: Reactivos, soluciones o tinturas citologicos

description: Test Inmunocromatografico para Malaria

description: Test Inmunocromatografico para Malaria

description: Reactivos de analizadores inmunologicos

description: Tiras Reactivas de Orina

description: Tiras Reactivas de Orina

description: Tiras o papeles para ensayos quimicos

description: Acetona pro analisis

description: Acetona pro analisis

description: Reactivos y standares para unidad de control de calidad

description: Acido acetico p.a.

description: Acido acetico p.a.

description: acidos inorganicos

description: Acido clorhidrico pro analisis

description: Acido clorhidrico pro analisis

description: Reactivos y standares para unidad de control de calidad

description: Tubo con anticoagulante

description: Tubo con anticoagulante

description: Tubos de ensayo multiproposito o generales

description: Acido Tricloroacetico p.a

description: Acido Tricloroacetico p.a

description: Reactivos y standares para unidad de control de calidad

description: Tubo con anticoagulante

description: Tubo con anticoagulante

description: Tubos de ensayo multiproposito o generales

description: Azul brillante de cresil

description: Azul brillante de cresil

description: Reactivos o soluciones quimicos

description: Tubos con Citrato de Sodio para Coagulación

description: Tubos con Citrato de Sodio para Coagulación

description: Tubos de ensayo multiproposito o generales

description: Tubos con Citrato de Sodio para Coagulación

description: Tubos con Citrato de Sodio para Coagulación

description: Tubos de ensayo multiproposito o generales

description: Tubo de plastico

description: Tubo de plastico

description: Tubos de ensayo multiproposito o generales

description: Alcohol de quemar

description: Alcohol de quemar

description: Alcoholes o sus sustitutos

description: Alcohol metanol

description: Alcohol metanol

description: Reactivos o soluciones quimicos

description: Tubo con anticoagulante

description: Tubo con anticoagulante

description: Tubos de ensayo multiproposito o generales

description: Tubo con anticoagulante

description: Tubo con anticoagulante

description: Tubos de ensayo multiproposito o generales

description: Iodopovidona Jabon Liquido

description: Iodopovidona Jabon Liquido

description: Geles o soluciones topicas de yodo

description: Anticoagulante gotas

description: Anticoagulante gotas

description: Reactivos de analizadores de banco de sangre

description: Rubeola Test Rapido

description: Rubeola Test Rapido

description: Kits de ensayos rapidos

description: Rubeola Test Rapido

description: Rubeola Test Rapido

description: Kits de ensayos rapidos

description: Aceite de inmersion

description: Aceite de inmersion

description: Aceite de inmersion para microscopio

description: Alcohol gel

description: Alcohol gel

description: Alcoholes o sus sustitutos

description: Agua destilada o deionizada

description: Agua destilada o deionizada

description: Agua destilada o deionizada

description: Agua destilada o deionizada

description: Agua destilada o deionizada

description: Agua destilada o deionizada

description: Propipeta

description: Propipeta

description: Pipetas serologicas

description: Gradillas soporte

description: Gradillas soporte

description: Placas, cajas o soportes con revestimiento para cultivo de tejidos

description: Agujas para extraccion de sangre

description: Agujas para extraccion de sangre

description: Agujas para extraccion de sangre

description: Banda adhesiva

description: Banda adhesiva

description: Cintas adhesivas medicas y quirurgicas para el uso general

description: Porta Lamina

description: Porta Lamina

description: Porta objetos para laboratorio

description: Porta Lamina

description: Porta Lamina

description: Porta objetos para laboratorio

description: Camara de Neubauer

description: Camara de Neubauer

description: Porta objetos para laboratorio

description: Lamina para cubre camara de newbauer

description: Lamina para cubre camara de newbauer

description: Cubreobjetos

description: Capilares c/heparina para hematocrito

description: Capilares c/heparina para hematocrito

description: Tubos capilares o de hematocrito

description: Contenedor / frascos de esputo

description: Contenedor / frascos de esputo

description: Aparatos o contenedores de recogida de esputos

description: Cepillo para tubo de ensayo

description: Cepillo para tubo de ensayo

description: Cepillos centrifugos

description: Cepillo para tubo de ensayo

description: Cepillo para tubo de ensayo

description: Cepillos centrifugos

description: Cepillo para tubo de ensayo

description: Cepillo para tubo de ensayo

description: Cepillos centrifugos

description: Crioviales de polipropileno, tapa rosca

description: Crioviales de polipropileno, tapa rosca

description: Tubos criogenicos

description: Contenedor para objetos corto punzantes

description: Contenedor para objetos corto punzantes

description: Contenedores, carros o accesorios para el desecho de agujas, cuchillas y otros objetos punzantes

description: Colorante Wrigh

description: Colorante Wrigh

description: Reactivos de analizadores de banco de sangre

description: Contenedor para objetos corto punzantes

description: Contenedor para objetos corto punzantes

description: Contenedores, carros o accesorios para el desecho de agujas, cuchillas y otros objetos punzantes

description: Colorante Wrigh

description: Colorante Wrigh

description: Reactivos de analizadores de banco de sangre

description: Colorante Giemsa

description: Colorante Giemsa

description: Reactivos o soluciones quimicos

description: Colorante May Grunwald

description: Colorante May Grunwald

description: Reactivos o soluciones quimicos

description: Cristal violeta

description: Cristal violeta

description: Reactivos o soluciones quimicos

description: Anticoagulante EDTA frasco gotero

description: Anticoagulante EDTA frasco gotero

description: Reactivos de analizadores de banco de sangre

description: Citrato de sodio anticoagulante gota

description: Citrato de sodio anticoagulante gota

description: Reactivos o soluciones quimicos

description: Anticoagulante Fluoruro

description: Anticoagulante Fluoruro

description: Reactivos o soluciones quimicos

description: Diluyente para globulos blancos

description: Diluyente para globulos blancos

description: Reactivos, soluciones o tinturas hematologicos

description: Glucosa

description: Glucosa

description: Reactivos y standares para unidad de control de calidad

description: Reactivo Benedict

description: Reactivo Benedict

description: Reactivos o soluciones quimicos

description: Ractivo pandy

description: Ractivo pandy

description: Reactivos o soluciones quimicos

description: Xilol

description: Xilol

description: Reactivos de analizadores de banco de sangre

description: Glicerina liquida

description: Glicerina liquida

description: Glicerina

description: Tapa Boca

description: Tapa Boca

description: Mascarillas de aislamiento o quirofano para el personal sanitario

description: Temporizador para Laboratorio

description: Temporizador para Laboratorio

description: Cronometros o relojes de laboratorio

description: Pipeta automatica con volumen variable

description: Pipeta automatica con volumen variable

description: Pipetas serologicas

title: LPN N° 155/2024 "ADQUISICIÓN DE REACTIVOS E INSUMOS PARA SÉPTIMA REGIÓN SANITARIA ITAPÚA"

title: Garantía de Mantenimiento de Ofertas

description: Los oferentes, proveedores o contratistas deberán garantizar la seriedad de sus ofertas, mediante la garantía de mantenimiento de la oferta

description: La garantía debe ser de un cierto porcentaje, periodo y tipo

title: Porcentaje de la garantia

title: Validez de la garantia en días

title: nota-de-respuesta-03-02-25-67a0fe430cac0.pdf

title: Pliego para la Adquisición de Bienes y/o Servicios - CONVENCIONAL - Ley Nº 7021/22. 455249 Version 5

documentTypeDetails: Pliego Electrónico de bases y Condiciones

title: nota-respuesta-24-01-25-6793d629973f0.pdf

title: dictamen-de-eett-66fd9a2ddae72.pdf

title: reporte-plurianual-2024-66fd9aeb0d0a8.pdf

title: estudio-de-precio-referencial-66fd7015eaa1d.pdf

title: nota-dncp-justificacion-contrato-abierto-66fd7052cd548.pdf

title: Pliego para la Adquisición de Bienes y/o Servicios - CONVENCIONAL - Ley Nº 7021/22. 455249 Version 5

title: vii-rs-reporte-de-incidencia-66fd702be9187.pdf

title: expte-sime-43-899-2024-ampliacion-presupuestaria-66fd6fdec5fa7.pdf

title: resolucion-dgaf-n-2828-24-aprobacion-del-pbc-66fd7001de1e6.pdf

title: nota-dncp-no-anticipo-66fd706609e32.pdf

title: nota-respuesta-678ac31d2c4ab.pdf

title: Lote 115 – ítem 2  Pipeta Autom. variable

answer: Ajustarse a la última versión del PBC-

description:
¿Podemos ofertas Pipeta automática con volumen variable de 5 – 50 ul?
 El mismo cumple con el rango solicitado.

title: Especificaciones Tecnicas

answer: Ajustarse al PBC. Las EETT fueron suficientemente estudiadas en base a la necesidad para los parámetros de sensibilidad y especificidad de los reactivos solicitados

description: En el apartado de especificaciones técnicas, en el lote 11, ítem 1: (Chagas Test Rápido), donde dice: “Test rápido inmunocromatográfico IgG/IgM para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.” Se solicita a la convocante admitir una sensibilidad de 94% como mínimo ya que esta sensibilidad sigue siendo altamente efectiva para pruebas de tamizaje, y es la sensibilidad más común para estos tipos de prueba. Y de esta manera también permitir una mayor participación de oferentes. Por lo expuesto, solicitamos que dicho requerimiento quede redactado de la siguiente manera: “Test rápido inmunocromatográfico IgG/IgM para tamizaje. Los reactivos deben tener una sensibilidad mínima del 94%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.”

title: LOTE 4 ITEM 48 ESPECIFICACIONES TECNICAS

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description: Se podría separar en un lote distinto el analizador de gases en sangre, debido a las especificaciones del equipo limitan la libre competencia por el lote completo. Se solicita una aclaración al respecto, favor contestar sin colocar ajustarse al pliego como respuesta.

title: LOTE 4 ITEM 27

answer: Ajustarse al PBC. Los equipamientos y sus características fueron ajustados dentro de las especificaciones técnicas elaboradas, conforme a los requerimientos técnicos y a las necesidades específicas del servicio. Utilizar equipos independientes para un solo test, retrasaría la capacidad técnica del servicio.

description: Se podria admitir un equipo de soporte que realice las determinaciones de electrolitos

title: Lote 4 – Item 1 – Especificaciones técnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: Se solicita amablemente a la convocante modificar el requisito de “25 (veinticinco) equipos automatizados de 3 partes para hemograma con gradillas en rack”, ya que los equipos con recuento diferencial de 3 partes no admiten el uso de gradillas en rack, sino que cuentan con una sonda que aspira cada muestra de manera individual. Por ello, se sugiere reformular el requisito como “25 (veinticinco) equipos automatizados de 3 partes para hemograma”. De lo contrario, se considera que se estaría beneficiando a los proveedores que poseen tales características, limitando así la libre competencia.

title: Periodo de validez de la Garantía de los bienes Pag. 8 del PBC

answer: Ajustarse a la última versión del PBC.

description:
VENCIMIENTO
Bajo el punto C dice: Reactivos e Insumos en General (que no requieren equipos en comodato): El vencimiento mínimo de los reactivos e INSUMOS es de 12 (doce) meses al momento de la entrega en los lugares indicados para la entrega. En caso de no cumplir con el requerimiento, mencionan que debemos presentar Autorización de entrega, acompañados de otros documentos.
Solicitamos a la convocante EXCLUIR el pedido de estos documentos (Nota de Autorización de Entrega, Carta de Compromiso de Canje y Póliza de Seguro) para los INSUMOS INERTES de laboratorio como ser gradillas de plástico, propipetas de plástico y de goma, cámara de Neubauer de vidrio, frascos para esputo u orina, pipetas automáticas, puntas para pipetas, cronómetros, láminas cubre-objetos, morteros de porcelana, placas para tipificación, placa para VDRL,  sellador de tubos, etc., etc., (es decir: productos de metal, de vidrio, de plástico, de porcelana y algodón) que por su naturaleza no cuentan con vencimiento alguno, por ende NO DEBE SER APLICADO el pedido de estos documentos para los INSUMOS INERTES mencionados más arriba.

title: Lote 1, Item 77 solicitud de separar el lote de analizador gasómetro.

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Se solicita a la convocante poder separar el lote de analizador gasómetro, en un lote distinto, debido a las especificaciones del equipo que se solicita, es una limitante en la competencia.

title: LOTE 3 ITEM 1

answer: Ajustarse al PBC. Las EETT fueron suficientemente estudiadas y planificadas en el contexto de la necesidad del servicio.

description: SE PODRIA ADMITIR UN KIT QUE APARTE DE SARS COV 2 TAMBIEN MIDA INFLUENZA

title: LOTE 1 ITEM 76 ESPECIFICACIONES TECNICAS

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description: Se podría separar en un lote distinto el analizador de gases en sangre, debido a las especificaciones del equipo limitan la libre competencia por el lote completo. Se solicita una aclaración al respecto, favor contestar sin colocar ajustarse al pliego como respuesta.

title: Capacidad técnica – Para Reactivos

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description: Consultamos amablemente a la convocante aclarar si con "Constancia de Trámite" se refiere a la mesa de entrada del trámite de la Solicitud de Registro Sanitario. Favor aclarar esta duda para que los oferentes puedan prepararse correctamente, ya que es un requisito excluyente para la oferta, y si los mismos no cumplen con los requisitos desistirán de la oferta.

title: Especificaciones Tecnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

title: Lote 115 – ítem 4  Pipeta Autom. variable

answer: Ajustarse a la última versión del PBC.

description:
¿Podemos ofertas Pipeta automática con volumen variable de 100 – 1.000 ul?
 El mismo cumple con el rango solicitado.

title: Especificaciones Tecnicas

answer: Ajustarse a la última versión del PBC

description: En el apartado de especificaciones técnicas, en el lote 60, ítem 1: (Rubeola Test Rápido), donde dice: “Rubeola IgM. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.”  Se solicita amablemente a la convocante admitir una presentación que permita la identificación de ambos anticuerpos, tanto IgG como IgM en el mismo test, ya que esto no afectaría la finalidad de uso del test ni compromete su efectividad. Además, se respetaría el número total de pruebas solicitadas, garantizando el cumplimiento de los requerimientos sin alterar el objetivo del proceso. Esta modificación también favorecería la participación de un mayor número de oferentes, promoviendo un proceso más justo y transparente para todos. Por lo expuesto, solicitamos que dicho requerimiento quede redactado de la siguiente manera: “Rubeola IgM o IgG/IgM. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.”

title: Lote 115. Item 1: PIPETA AUTOMATICA CON VOLUMEN VARIABLE

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Solicitamos amablemente a la convocante aclarar el tipo de pipeta que precisa, de volumen fijo o  de volumen variable, ya que hay discrepancia de lo solicitado en la planilla de precios con la descripción de las EE.TT del PBC.

title: Capacidad técnica – Para Reactivos

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description: Se reitera amablemente a la convocante, si con “Constancia o Certificado de Trámite emitido por DINAVISA” se refiere al expediente de Mesa de Entrada de la Solicitud de Registro Sanitario. Se recuerda que a la fecha no se expide por parte de DINAVISA ninguna constancia de trámite ni certificado de trámite.

title: Muestras

answer: Ajustarse a lo requerido en el PBC. Se aclara que el Comité Evaluador solicitará las muestras "si lo considera necesario". En el Pliego de Bases, página 8

description: Solicitamos a la convocante considerar aceptar Catálogos en lugar de muestras físicas debido a que no todos los proveedores contamos en stock de los insumos de laboratorio requeridos.

title: Plan de entrega de los bienes

answer: Ajustarse al PBC. Se solicita respetar los tiempos de entrega de los productos teniendo en cuenta el stock critico de reactivos en los servicios de salud.

description: En el apartado de PLAN DE ENTREGA DE LOS BIENES, en el punto REACTIVOS E INSUMOS EN GENERAL (no requieren equipos en comodato), donde menciona: “1ra ENTREGA 20% DE LA CANTIDAD MÍNIMA: El oferente tendrá un plazo de hasta 20 (veinte) días calendario para la entrega de los productos adjudicados a partir de la recepción de la Orden de Compra, la cuál será emitida dentro de los 20 (VEINTE) días de haber firmado el contrato.” Se solicita amablemente a la convocante extender el plazo de entrega a 45 días calendario. Esta ampliación es necesaria debido a los tiempos requeridos para la fabricación, la logística de envío y los trámites aduaneros, los cuales podrían superar fácilmente un mes. Ampliar el plazo permitiría que más oferentes, que tienen la capacidad de suministrar los productos solicitados, puedan participar en el proceso, fomentando así una competencia más equitativa. De este modo, se evitaría favorecer únicamente a aquellos que disponen del producto en stock, asegurando una oportunidad justa para todos los participantes.

title: Lote 115 / Item 14

description: : Se podría admitir presentaciones de puntas de bolsas por 300 unidades.

title: abastecimiento simultaneo

answer:
No es viable la modalidad de abastecimiento simultáneo, para los Laboratorios de la DRNL, por la infraestructura, espacio físico, cantidad de RRHH, productividad, y otros, ya que se deben manejar más de un equipo, probablemente más de un software, más de una marca de reactivo, depósitos refrigerados, etc.,   además, no es posible, instalar equipos de gran porte y desinstalar de manera rápida al terminar uno de ellos y utilizar el otro, lleva días instalar y desinstalar dichos equipos de gran porte, y no se puede paralizar las prestaciones del servicio.

El artículo 43 de la Ley 7021/22 reglamenta este tipo de modalidad para aquellos casos en que ningún proveedor pueda proveer o ejecutar la totalidad de los bienes o servicios. Por experiencias previas con la modalidad de las LPN, para los lotes con equipos en COMODATO, no es viable la modalidad de abastecimiento simultáneo, para los Laboratorios de la DRNL.
Citamos inconvenientes por los cuales, no es viable esta modalidad, cuando lo que se solicita es trabajar en la modalidad de reactivos con equipos en comodato.

-	Si el llamado es Abastecimiento Simultáneo, serán adjudicados hasta 3 oferentes; implica que se tendrán mínimamente 3 equipos, de 3 diferentes proveedores.
-	No se cuenta con espacio físico suficiente para esa cantidad de equipos, ni RRHH que trabaje en simultáneo, con más cantidad de equipos.
-	No es viable ya que los reactivos de la marca de un proveedor no pueden ser utilizados en otro equipo de otra marca del otro proveedor, a fin de garantizar las Buenas prácticas de Laboratorio, la Garantía de calidad y buena procedencia de los reactivos, equipos deben ser de la misma marca.
-	Los controles, calibradores, consumibles, insumos, etc., deberían entregar 3 proveedores, acarrea la dificultad por no contar con espacio físico, ni personal necesario para el control y monitoreo.

description: Solicitamos amablemente a la convocante cambiar el sistema de adjudicación a  abastecimiento simultáneo para todos los lotes, a el fin de garantizar la adecuada provisión de los insumos requeridos.

title: LOTE 4

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: Lote 4: Se solicita amablemente a la convocante la modificación de las especificaciones técnicas del pliego para el gasómetro, reemplazando el punto “Mínimas características: electrodos libres de mantenimiento para determinaciones de electrolitos: Na, K, Cl y Ca. Para Gasometría pH, pCO2, pO2, Sat de O2, Hemoglobina total, Hematocrito. H+, CHCO3, CTCO2, BE, BEACT, BESS, BB, pHST, CHCO3ST, pAO2, NCa, QS/QT, p50, sO2 (C), AaDO2, a/AO2, eAVDO2, AGOER, Hct (C), PAO2t, H+t, AaDO2t. Temperatura de medición 37 + - 0,2ºC" por “Mínimas características: electrodos libres de mantenimiento para determinaciones de electrolitos: Na, K, Cl y Ca. Para gasometría: pH, cloruro, pCO2, pO2, Hct y parámetros calculados como Hb, HCO3, BE, BE-B, BE-ECF, TCO2, AG, AG (K), O2 Sat, O2 Ct, SBC, nCa, TCa, pO2%, A, AaDO2 y a / A", ya que estas estas especificaciones se ajustan a los estándares tecnológicos necesarios para el análisis de gases en sangre, además de permitir la mayor participación de oferentes sin limitar la libre competencia.

title: Especificaciones Tecnicas

answer: Ajustarse al PBC. Las EETT fueron suficientemente estudiadas teniendo en cuenta la necesidad del servicio, en cuanto a la demanda de las pruebas, capacidad de RRHH e infraestructura.

description: En el apartado de especificaciones técnicas, en el lote 12, ítem 1: (Test Rápido para Hepatitis A), donde dice: “Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Capacidad de realizar como mínimo 10 y máximo 20 determinaciones por presentación.” Se solicita amablemente a la convocante admitir la dilución de las muestras en un buffer de extracción antes de depositarlas en las tiras o casetes. Este procedimiento mejora la precisión de los resultados, ya que el uso de un buffer estandariza la concentración de anticuerpos en la muestra. Al diluirla, se reducen las interferencias potenciales y se garantiza que la cantidad de anticuerpos sea adecuada para su detección, lo que ayuda a prevenir resultados falsos positivos o negativos.

title: Especificaciones Tecnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: En el apartado de especificaciones técnicas, en el lote 108, ítem 1 (LÁMINAS PORTAOBJETO), donde dice: “De vidrio, 76 x 26 mm. Traslúcidas, libre de ralladuras y que no se peguen entre sí. La oferta es por unidad de lámina.” Se solicita amablemente a la convocante la admisión de un rango en las dimensiones de la lámina, ya que cada marca maneja dimensiones específicas con variaciones mínimas entre ellas. Esto ayudaría a evitar que se favorezca a un solo oferente con una marca que coincida exactamente con las dimensiones descritas. Por lo expuesto, solicitamos que dicho requerimiento quede redactado de la siguiente manera: “De vidrio, 76 x 26 mm ± 1 mm. Traslúcidas, libre de ralladuras y que no se peguen entre sí. La oferta es por unidad de lámina.”

title: Lote 4 – Items 2 al 26

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: La convocante solicita "capacidad de carga de 40 posiciones de muestra como mínimo". Considerando que la mayoría de los equipos con capacidad 200test/h, poseen 30 posiciones de muestra. Se solicita amablemente a la convocante, aceptar como mínimo una capacidad de carga de 30 muestras, ya que la capacidad del equipo sigue siendo de 200test/h, para fomentar una mayor participación de los oferentes.

title: LOTE 4 - Ítem 46

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: Lote 4: Se solicita amablemente a la convocante la exclusión del item 46 Chagas IgM del lote 4, ya que la detección de anticuerpos IgM para Chagas tiene una utilidad clínica limitada en el diagnóstico de esta enfermedad, siendo que la mayoría de los casos corresponden a infecciones crónicas, donde la detección de IgM no es relevante. Además, los métodos serológicos actuales se enfocan en la detección de IgG (ítem 47), que son más sensibles y específicos para el diagnóstico de la enfermedad de Chagas en sus etapas crónicas. Por lo tanto, consideramos que la inclusión de este ítem no aporta valor diagnóstico significativo y podría generar costos innecesarios, además de ser una determinación analizada en limitados analizadores del mercado, evitando una mayor oportunidad de competencia entre proveedores.

title: Lote Nº 115 Ítem 15 Puntas

answer: Ajustarse a la última versión del PBC.

description: ¿Podemos ofertar Puntas con volumen de 0,5-20 ul para pipeta automática variable y fijo con volúmenes de hasta 20 ul? Aclaramos que el mismo cumple con el rango solicitado en el PBC

title: Lote 1 - Item 76

answer: Ajustarse a la última versión del PBC.

description: : Se podría admitir un analizador de gases en sangre que no realice los cálculos de los siguientes parámetros: QS/QT, p50, AaDO2, a/AO2, eAVDO2, AGOER, RItPHtPCO2t, AaDO2t.

title: Especificaciones Tecnicas

answer: Ajustarse a la última versión del PBC.-

description: En el apartado de ESPECIFICACIONES TÉCNICAS, en el lote 25, ítem 1: (Test Rápido para Rotavirus), donde dice: “Test rápido inmunocromatográfico en un solo paso. Para la detección cualitativa in vitro de antígenos de rotavirus en materia fecal humana. Con 99,9% de sensibilidad y 99,6 % de especificidad. Con certificado de calidad del país de origen. Capacidad de realizar un mínimo de 20 determinaciones por presentación.” Se solicita a la convocante admitir una sensibilidad del 96% como mínimo ya que esta sensibilidad sigue siendo altamente efectiva para pruebas de tamizaje, y es la sensibilidad más común para estos tipos de test. Al establecer este estándar, se facilitará la participación de un mayor número de oferentes, lo que promoverá la competencia y, potencialmente, mejores condiciones en la adquisición de los productos. Por lo tanto, proponemos que dicho requerimiento se redacte de la siguiente manera: “Test rápido inmunocromatográfico en un solo paso. Para la detección cualitativa in vitro de antígenos de rotavirus en materia fecal humana. Con 96% como mínimo de sensibilidad y 99,6 % de especificidad. Con certificado de calidad del país de origen. Capacidad de realizar un mínimo de 20 determinaciones por presentación

title: Lote 4 / Item 48

answer: Ajustarse a la última versión del PBC.

description: Se podría admitir un analizador de gases en sangre que no realice los cálculos de los siguientes parámetros: QS/QT, p50, AaDO2, a/AO2, eAVDO2, AGOER, RItPHtPCO2t, AaDO2t.

title: lote 4

answer: Ajustarse al PBC. Las EETT fueron suficientemente estudiadas teniendo en cuenta la necesidad del servicio, en cuanto a la demanda de las pruebas, capacidad de RRHH e infraestructura.

description: LOTE 4, ÍTEM 48 al 49: consultar si se puede colocar un equipo auxiliar al equipo de inmunología para procesar Chagas IgG e IgM, o de preferencia proveer test rápido

title: Lote Nº 115 Ítem 16 Puntas

answer: Ajustarse a la última versión del PBC.

description: ¿Podemos ofertar Puntas AMARILLAS con volumen de 2-200 ul para pipeta automática variable y fija con volúmenes de hasta 200 ul? Aclaramos que el mismo cumple con el rango solicitado en el PBC

title: Lote 4 - Especificaciones técnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Solicitamos a la convocante, admitir un equipo de soporte para 2 o 3 determinaciones de Inmunología en caso de no cumplir con el panel completo. Esta medida ya se ha adoptado en otros llamados de la convocante para equipamientos de laboratorio clínico, como la LPN 39/2024 ID 432638.

title: Lote 1, Item 2

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: Se solicita a la convocante poder aceptar un equipo con gestor de muestras flexibles, en donde admita 240 posiciones de muestra para ensayo de química e inmunoensayo 40 posiciones para muestras de urgencia. Aspiración de 1 minuto. Carga continua de muestras mediante gradillas. Destape automático de los tubos a una velocidad de 300 tubos/hora, desinfección UV de tubos.

title: Especificaciones Tecnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: En el apartado de especificaciones técnicas, en el lote 59, ítem 1: (Test Rápido para Hepatitis A), donde dice: para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%. “Rubéola IgG. Test rápido inmunocromatográfico Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: mínimo 20 y máximo 30 determinaciones por presentación.” Solicitamos amablemente a la convocante la admisión del test inmunocromatográfico rápido que permite la detección simultánea de los anticuerpos IgG e IgM en una misma prueba. Este tipo de prueba es ampliamente utilizado y reconocido como el más adecuado para procesos de tamizaje, dada su eficiencia y confiabilidad. El propósito fundamental del tamizaje es identificar la exposición al virus de la rubéola, asegurando así el cumplimiento de los requisitos necesarios para dicho fin de manera precisa y efectiva. Por consiguiente, sugerimos que dicho requerimiento quede redactado de la siguiente forma: “Rubéola IgG o IgG/IgM. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: mínimo 20 y máximo 30 determinaciones por presentación.”

title: lote 4

answer: Ajustarse al PBC. Las EETT fueron suficientemente estudiadas teniendo en cuenta la necesidad del servicio, en cuanto a la demanda de las pruebas, capacidad de RRHH e infraestructura.

description: LOTE 4, ÍTEM 51 al 53:   requiere equipo Semiautomatizado de 4 canales de medición independiente como mínimo. Se consulta a la convocante  si aceptarían:  Semiautomatizado de 2 canales de medición independiente como mínimo.

title: Capacidad Técnica

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Se solicita amablemente a la convocante admitir la mesa entrada del expediente de Solicitud de Registro Sanitario a DINAVISA, de forma excepcional, ya que nos encontramos en una transición de LCSP a DINAVISA en cuanto a solicitud de registros. Esta medida extraordinaria ya fue admitida en otros llamados como la LPN Nº 134/2024 con ID 454374, donde se establece "En caso de que los documentos se encuentren vencidos deberá acompañar la Constancia o Certificado de Trámite emitido por DINAVISA."

title: Lote 4 – Item 48 – Especificaciones técnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Se solicita a la convocante, aclarar lo solicitado en las especificaciones técnicas adicionales de “Kit de Electrolitos y Gasómetro” donde se establece “Mínimas características: electrodos libres de mantenimiento para determinaciones de electrolitos: Na, K, Cl y Ca. Para Gasometría pH, pCO2, pO2, Sat de O2, Hemoglobina total, Hematocrito. H+, CHCO3, CTCO2, BE, BEACT, BESS, BB, pHST, CHCO3ST, pAO2, NCa, QS/QT, p50, sO2 (C), AaDO2, a/AO2, eAVDO2, AGOER, Hct (C), PAO2t, RItPHtPCO2t, H+t, AaDO2t.”, ya que, no se encuentra suficientemente claro lo solicitado en el parámetro RItPHtPCO2t. Si lo que se solicita es cálculo de Índice respiratorio (RI), solicitamos amablemente solicitar este parámetro, ya que el que se encuentra en el PBC es un parámetro que no logramos identificar.

title: Lote 1 – Item 1

answer: Ajustarse a la última versión del PBC.

description: Con relación a los años de fabricación se solicita que los equipos para el Item indicado puedan tener más de 5 años de fabricación (hasta 8 años) con mantenimiento preventivo y correctivo al día y con informe del jefe de servicio sobre el desempeño del mismo.

title: Lote 4 – Ítem 29 al 49

answer: Ajustarse a la última versión del PBC.

description: Se solicita a la convocante, establecer las especificaciones técnicas mínimas requeridas para el equipo de Inmunología automatizado, ya que se establece que el módulo de Inmunología se encuentra separado.

title: item 78

answer: Ajustarse a la última versión del PBC.-

description:
DONDE DICE: 10 a 13 litros. Descafrtable. Cartón prensadso con bolsa colectora en su interior, de polietileno resistente, con simbolo universal de riesgo biológico en ambas caras.
Consulta: Solicitamos respetuosamente a la convocante la aclaración de la solicitud es de cartones armados con bolsas en su interior o por partes para el armado posterior.

title: Lote 1, Item 30 equipo de soporte para poder realizar 5 a 6 determinaciones

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Se solicita a la convocante tener en cuenta la opción de tener un equipo de soporte para poder realizar 5 a 6 determinaciones que el módulo de inmunología no pueda realizar, esto permitiría una mejor oportunidad de competencia.

title: Especificaciones Tecnicas

answer: Ajustarse a la última versión del PBC.-

description: En el apartado de especificaciones técnicas, en el lote 22, ítem 1: (TEST MULTIDROGAS PARA ORINA - 5 DROGAS), donde dice: “TEST RÁPIDO PARA 5 DROGAS (COC, THC, BZO, MOP, AMP). Panel de test rápido inmunocromatográfico para detección cualitativa de 5 drogas de abuso. Los reactivos deben tener una sensibilidad no menor al 97%, con certificado de Calidad del país de origen.  Capacidad de realizar entre un mínimo de 20 y un máximo de 30 determinaciones por presentación.” Se solicita amablemente a la convocante la admisión de una presentación que incluya la determinación de cocaína, opiáceos, benzodiacepinas, marihuana y anfetamina. La inclusión de estas sustancias responde a la necesidad de abordar una gama más amplia de sustancias que son relevantes en el contexto actual de la salud pública y la seguridad. La prevalencia del consumo de cocaína, opiáceos, benzodiacepinas y anfetaminas ha aumentado considerablemente en los últimos años, afectando a diversas poblaciones y planteando retos significativos para los profesionales de la salud. Por lo tanto, esta modificación permitiría un análisis más completo y pertinente, facilitando la implementación de políticas adecuadas y programas de intervención. Por lo expuesto, solicitamos que dicho requerimiento quede redactado de la siguiente manera: “TEST RÁPIDO PARA 5 DROGAS (COC, THC, BZO, MOP, AMP o COC, THC, BZO, AMP, OPI). Panel de test rápido inmunocromatográfico para detección cualitativa de 5 drogas de abuso. Los reactivos deben tener una sensibilidad no menor al 97%, con certificado de Calidad del país de origen.  Capacidad de realizar entre un mínimo de 20 y un máximo de 30 determinaciones por presentación.”

title: LOTE 1 ITEM 73

answer: misma respuesta que Consulta 18

description: SE PODRIA ESPECIFICAR SI SE TRATA DE LA DETERMINACION ULTRASENSIBLE

title: Lote 6 – Item 1 – Especificaciones técnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Se solicita amablemente a la convocante, establecer como requisito mínimo que la capacidad mínima de carga de muestra por racks sea de 10 muestras, teniendo en cuenta la envergadura del llamado y la necesidad del servicio.

title: Lote 4, Item 1

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: Se solicita encarecidamente a la convocante que los contadores hematológicos de 3 partes no cuenten con racks de muestras, esto debido a que los contadores de 3 partes no cuentan con racks para muestras, de esta forma se lograría una mejor competencia entre los oferentes.

title: Lote 1, Item 1

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: Se solicita a la convocante de poder realizar la oferta del lote de contador hematológico aparte del lote de química, de esta manera se podría lograr una mejor competencia entre los oferentes.

title: Lote 1, Item 2

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: Se solicita a la convocante poder aceptar un equipo con gestor de muestras flexibles, en donde admita 240 posiciones de muestra para ensayo de química e inmunoensayo 40 posiciones para muestras de urgencia. Aspiración de 1 minuto. Carga continua de muestras mediante gradillas. Destape automático de los tubos a una velocidad de 300 tubos/hora, desinfección UV de tubos.

title: Lote 1 - Item 2

answer: Ajustarse a la última versión del PBC.

description: : Se podría ofertar equipo de química e inmunología por separado, de esta forma permitir una mayor participación de oferentes

title: Lote 4 - Item 1 - Especificaciones técnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

title: Lote 115 / Item 15

description: Se podría admitir puntas con capacidad de 20uL pero que cuenten con filtro.

title: LOTE 1 ITEM 2 ESPECIFICACIONES TECNICAS

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description: Se solicita a la convocante poder aclarar las características del Gestor de modulo de muestras que se requiera. Se solicita una aclaración al respecto, favor contestar sin colocar ajustarse al pliego como respuesta.

title: Plan de entrega de los bienes

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: En el apartado de PLAN DE ENTREGA DE LOS BIENES, en el punto REACTIVOS E INSUMOS EN GENERAL (no requieren equipos en comodato), donde menciona: “1ra ENTREGA 20% DE LA CANTIDAD MÍNIMA: El oferente tendrá un plazo de hasta 20 (veinte) días calendario para la entrega de los productos adjudicados a partir de la recepción de la Orden de Compra, la cuál será emitida dentro de los 20 (VEINTE) días de haber firmado el contrato.” Se solicita amablemente a la convocante la ampliación del plazo de entrega a 45 días calendario. Esta extensión resulta fundamental debido a los tiempos requeridos para la fabricación de los insumos, los procesos logísticos de envío y los trámites aduaneros, los cuales, en muchos casos, pueden superar el mes. Ampliar el plazo de entrega permitirá la participación de un mayor número de oferentes con capacidad de suministro, fomentando una competencia más equitativa. De esta manera, se evitaría restringir la adjudicación únicamente a proveedores con stock disponible inmediata, asegurando condiciones justas y promoviendo la adquisición de productos con los más altos estándares de calidad. Por lo expuesto, solicitamos que dicho requerimiento quede redactado de la siguiente manera: “1ra ENTREGA 20% DE LA CANTIDAD MÍNIMA: El oferente tendrá un plazo de hasta 45 (CUARENTA Y CINCO) días calendario para la entrega de los productos adjudicados a partir de la recepción de la Orden de Compra, la cuál será emitida dentro de los 20 (VEINTE) días de haber firmado el contrato.”

title: Lote 4 / Item 2:

answer: Ajustarse a la última versión del PBC.

description: Se podría admitir equipos de soporte para realizar las determinaciones que el equipo integrado no pudiese realizar

title: Lote 1 – Ítem 77

answer: Ajustarse a la última versión del PBC.

description: Se solicita a la convocante, establecer las especificaciones técnicas mínimas requeridas para el Kit para equipo de Electrolitos y Gasómetro.

title: lote 20, item 1, item 2

answer: Ajustarse a la última versión del PBC.-

description: Lote 20, permitirá la convocante método Elisa o Clia, test monotest, Presentacion de entrega kit con 24 determinaciones como mínimo. Que incluya controles y calibradores en el kit.

title: Lote 4, Ítem 50 analizador gasómetro.

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Se solicita a la convocante poder separar el lote de analizador gasómetro, en un lote distinto, debido a las especificaciones del equipo que se solicita, es una limitante en la competencia.

title: LOTE 4

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Se solicita a la convocante admitir 30 posiciones como mínimo de reactivos refrigerados a bordo para el equipo de química clinica del LOTE 4, para dar mayor participación a oferentes.

title: Abastecimiento simultáneo

answer: Ajustarse al PBC.-

description: Solicitamos a la convocante modificar el sistema de abastecimiento, a abastecimiento simultaneo para todos los lotes, debido a la envergadura del llamado, de manera a asegurar la correcta provisión de los insumos solicitados.

title: Lote 1 - Item 2

answer: Ajustarse a la última versión del PBC.

description: Se podría admitir un equipo integrado sin gestor de módulo de muestras.

title: Especificaciones Tecnicas

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description: En el apartado de especificaciones técnicas, en el lote 59, ítem 1: (Rubéola Test Rápido), donde dice: “Rubéola IgG o IgG/IgM. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.” Se solicita a la convocante la admisión de pruebas con una sensibilidad mínima del 95 %, dado que este nivel de sensibilidad sigue siendo altamente efectivo para los procesos de tamizaje. Además, este valor es ampliamente aceptado y común en este tipo de pruebas dentro de los estándares. Por consiguiente, solicitamos que dicho requerimiento quede redactado de la siguiente forma: “Rubéola IgG o IgG/IgM. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 95%. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.”

title: LOTE 144 ITEM 1 - TUBOS DE CENTRÍFUGA CONICOS DE PLÁSTICO

answer: Ajustarse al PBC .El PBC menciona lo siguiente: "de 12 mililitros de capacidad como mínimo".

description: CONSULTAR EL VOLUMEN DE CAPACIDAD DE LOS TUBOS A OFERTAR

title: Lote 4 / Item 47:

description: Se podría aceptar un equipo de soporte para las determinaciones que el equipo de inmunología no pueda realizar, que cuente con la misma tecnología por CLIA; de esta manera de podría tener una mejor oportunidad de competencia entre los oferentes. Favor de no responder ajustarse a la versión nueva del pliego. Se solicita una aclaración.

title: Capacidad técnica – Para Reactivos

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: La Convocante establece como requisito documental "En caso de que los documentos se encuentren vencidos deberá acompañar la Constancia o Certificado de Trámite emitido por DINAVISA.". Se consulta respetuosamente si con "Constancia o Certificado de Trámite" se refiere al trámite digital en sí, que corresponde a la mesa entrada del expediente de Solicitud de Registro Sanitario a DINAVISA, el cual puede ser descargado en formato PDF del portal digital de DINAVISA, ya que a la fecha, no se emite ninguna Constancia de Trámite ni Certificado de Trámite. Esta medida ya ha sido adoptada en otros llamados como el de LPN N° 07/2025 ID 458529.

title: lote 4

answer: Ajustarse al PBC. Las EETT fueron suficientemente estudiadas para responder a la necesidad del servicio y se tuvo en cuenta la calidad de los procesos para la manipulación de las muestras.

description: LOTE 4, ÍTEM 1 hemograma, CENTROS DE SALUD Y USF Se consulta si aceptarían: volumen de muestra entre 30 - 100 ul

title: Especificaciones Tecnicas

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description: En el apartado de especificaciones técnicas, en el lote 118, ítem 1 (PIPETA PASTEUR), donde dice: “DE PLASTICO, DE 5 ML DE VOLUMEN.” Se solicita a la convocante la aceptación de un volumen de 12 mililitros, con una escala de graduación que varía de 1 a 5 ml, dado que este rango incluye el volumen requerido y proporciona una mayor flexibilidad. Esto facilita el trabajo en laboratorio y permite una adaptabilidad superior en los procedimientos. Por lo expuesto, solicitamos que dicho enunciado quede redactado de la siguiente forma: “DE PLASTICO, DE 5 ML DE VOLUMEN COMO MINIMO.”

title: Lote Nº 115 Ítem 13 Puntas

answer: Ajustarse a la última versión del PBC.

description: ¿Podemos ofertar Puntas con volumen de 50-1.000 ul para pipeta automática variable y fijo con volúmenes de hasta 1.000 ul? Aclaramos que el mismo cumple con el rango solicitado en el PBC

title: Especificaciones Tecnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: En el apartado de especificaciones técnicas, en el lote 118, ítem 1 (PIPETA PASTEUR), donde dice: “DE PLASTICO, DE 5 ML DE VOLUMEN.” Se solicita amablemente a la convocante considerar la aceptación de un volumen de 12 mililitros, con una escala de graduación que varía de 1 a 5 ml, dado que este rango incluye el volumen requerido y proporciona una mayor flexibilidad. Esto facilita el trabajo en laboratorio y permite una adaptabilidad superior en los procedimientos. Por lo expuesto, solicitamos que dicho enunciado quede redactado de la siguiente forma: “DE PLASTICO, DE 5 ML COMO MÍNIMO DE VOLUMEN.” Por lo expuesto, solicitamos que dicho enunciado quede redactado de la siguiente forma: “DE PLASTICO, DE 5 ML COMO MÍNIMO DE VOLUMEN.”

title: Lote 115. Item 16. Puntas desechables para pipetas automaticas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Sería tan amable la convocante de aclarar el tipo de puntas que está precisando, ya que el "Material de polietileno de baja densidad" hace referencia a pipetas de transferencia tipo "PASTEUR", no a puntas desechables para micropipetas automáticas que están hechas con material de polipropileno (PP)

title: Lote 2 Item 1

answer:
: Ajustarse a la última versión del PBC.
Dicha determinación se realizará por el Método ELISA u otro de tecnología similar.

description: La determinación de Antigeno Dengue – NS1 podría permitirse realizar por biología molecular o por inmunocromatografia.

title: Lote 4 / Item 2

answer: Ajustarse a la última versión del PBC.

description: Se podría admitir reactivos de una marca distinta al equipo ofertado, pero que sea compatible con dicho equipo

title: Lote 4 - Item 2 al 26 - Especificaciones técnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: La convocante solicita "capacidad de carga de 40 posiciones de muestra como mínimo". Considerando que la mayoría de los equipos con capacidad 200test/h, poseen 30 posiciones de muestra. Se solicita amablemente a la convocante, aceptar como mínimo una capacidad de carga de 30 muestras, ya que la capacidad del equipo sigue siendo de 200test/h. De lo contrario, se considera que se está direccionando la oferta para oferentes que cuentan estas características. Hacemos esta solicitud amparados en los lineamientos de la Ley 7021/22 y Decreto 2264/24 en su Art.58. establece “ Especificaciones técnicas. Las especificaciones técnicas que deban contener las bases de la contratación se establecerán con la mayor amplitud de acuerdo con la naturaleza específica del contrato, con el objeto de que concurra el mayor número de oferentes. Sin embargo, deberán ser lo suficientemente claras, objetivas e imparciales, para evitar favorecer indebidamente a algún participante. Cuando los tipos conocidos de materiales, artefactos o equipos únicamente puedan ser caracterizados total o parcialmente mediante signos distintivos no universales, únicamente se hará a manera de referencia, procurando que la alusión se adecue a estándares internacionales comúnmente aceptados”

title: Lote 115 – ítem 3  Pipeta Autom. variable

answer: Ajustarse a la última versión del PBC.

description:
¿Podemos ofertas Pipeta automática con volumen variable de 20 – 200 ul?
 El mismo cumple con el rango solicitado.

title: Lote 1 item 30 ESPECIFICACIONES TECNICAS

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description: Se solicita a la convocante tener en cuenta la opción de tener un equipo de soporte para poder realizar 5 a 6 determinaciones que el módulo de inmunología no pueda realizar, con la misma tecnología del equipo ofertado; esto permitiría una mejor oportunidad de competencia entre los oferentes. Se solicita una aclaración al respecto, favor contestar sin colocar ajustarse al pliego como respuesta.

title: : Requisitos documentales para evaluar el criterio de capacidad

answer: Ajustarse al PBC. Se solicitan las documentaciones necesarias tanto certificado como constancias según corresponda. Actualmente es DINAVISA la institución, independiente del MSPBS, es una secretaria de Estado, encargada de emitir las documentaciones necesarias para la importación y comercialización de los PDIV en la República del Paraguay. El Certificado de libre venta es emitido por la autoridad sanitaria del país de origen de fabricación del producto, no es lo mismo que una aprobación de una autoridad sanitaria local como lo es DINAVISA, para poder utilizar el producto en nuestro país.

description: La convocante solicita “Constancia de inscripción del PDIV (productos para diagnóstico de uso In Vitro) expedida por la Dirección de Registros, Habilitación y Control del Laboratorio Central de Salud Pública. Y/o DINAVISA según corresponda”. Teniendo en cuenta que nos encontramos en un proceso de transición de cambio entre Laboratorio Central y Dinavisa y un retraso importante en Dinavisa en la emisión de algún tipo de documento. Solicitamos a la convocante o excluir dicho requisito o sustituir por Certificado de venta libre del país de origen de cada determinación ofertada, tal como lo está estableciendo el Instituto de Previsión Social referenciado con el LPN N° 160/24 – ID N° 454803, LPN SBE 39-23 – ID N° 432638 de la misma convocante.

title: Muestras

answer: Ajustarse a lo requerido en el PBC. Se aclara que el Comité Evaluador solicitará las muestras "si lo considera necesario". En el Pliego de Bases, página 8

description: Solicitamos a la convocante considerar aceptar Catálogos en lugar de muestras físicas debido a que no todos los proveedores contamos en stock de los insumos de laboratorio requeridos.

title: Especificaciones Tecnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: En el apartado de especificaciones técnicas, en el lote 11, ítem 1: (Chagas Test Rápido), donde dice: “Test rápido inmunocromatográfico IgG/IgM o IgG para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.” Se solicita a la entidad convocante considerar la aceptación de una sensibilidad mínima del 94% en las pruebas de tamizaje, dado que este nivel garantiza un desempeño óptimo para la detección de casos positivos y cumple con los estándares internacionales aplicables a este tipo de diagnóstico. La adopción de este criterio permitiría ampliar la participación de oferentes con productos de calidad comprobada, fomentando la competitividad en el proceso de adquisición y asegurando el acceso a soluciones diagnósticas eficientes y confiables para la población objetivo. Por lo expuesto, solicitamos que dicho requerimiento quede redactado de la siguiente manera: “Test rápido inmunocromatográfico IgG/IgM o IgG para tamizaje. Los reactivos deben tener una sensibilidad mínima del 94%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.”

title: LOTE 145  ITEM  - TUBOS DE CENTRÍFUGA CONICOS DE PLÁSTICO

answer: Ajustarse al PBC El PBC menciona lo siguiente: "de 15 mililitros de capacidad".

description: CONSULTAR EL VOLUMEN DE CAPACIDAD DE LOS TUBOS A OFERTAR

title: Lote 6 – Item 1 – Especificaciones técnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Solicitamos a la convocante, establecer como requisito mínimo que los tubos primarios sean directamente perforados por el equipo analizador HPLC, ya que esto evitaría manipulación innecesaria de la muestra por parte del personal del laboratorio, ahorraría tiempo de trabajo, además de mejorar la seguridad biológica y minimizaría la contaminación externa, ya que la muestra se encontraría con su respectiva tapa durante todo el procesamiento.

title: Confirmación de Distribución

answer: Ajustarse a la última versión del PBC.

description: Solicitamos amablemente a la convocante que aclare cada lote a que Hospital será destinado

title: LOTE 53 ITEM 1

answer: Ajustarse a la última versión del PBC.-

description: SE PODRIA ADMITIR UN TUBO DE 2,7mL

title: Lote 4 – Ítem 29 al 49

answer: Ajustarse a la última versión del PBC.

description: Se solicita a la convocante, establecer las especificaciones técnicas mínimas requeridas para el equipo de Inmunología automatizado, ya que se establece que el módulo de Inmunología se encuentra separado.

title: Especificaciones técnicas adicionales de todos los lotes

answer: Ajustarse a la última versión del PBC.

description: Se solicita a la convocante, aclarar a qué hospital, centro de salud, o unidad sanitaria corresponde cada lote. Se realiza esta solicitud de aclaración, verificándose que solamente para el HOSPITAL REGIONAL DE ENCARNACIÓN se establece de forma unívoca el lote correspondiente.

title: Lote 1 / Item 1

answer: Ajustarse a la última versión del PBC.

description: Se podría separar el lote de contador hematología del lote de equipo integrado de química-inmuno.

title: LOTE 1 ITEM 76 - Electrolitos (Na, K, Cl)

answer: Ajustarse al PBC. El Pliego menciona lo siguiente: Se permitirá un segundo equipo auxiliar con tecnología similar para un máximo de 5 o 6 test

description: Se podria admitir un equipo de soporte que realice las determinaciones de electrolitos

title: Lote 4, Ítem 48 determinación de Chagas IgM

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Se solicita a la convocante no tener en cuenta la determinación de Chagas IgM, esto debido a su nula utilidad clínica, debido a que no forma parte de los algoritmos de diagnóstico, esto porque la medición de Chagas IgM no se encuentra sugerida para el diagnóstico de enfermedad aguda. También debido a que no hay evidencia de que exista una marca de reactivo que cuente con la determinación de Chagas IgM.

title: Lote 4 / Item 48:

answer: Ajustarse a la última versión del PBC.

description: Se podría separar en un lote distinto el analizador de gases en sangre.

title: LOTE 4 ITEM 1: KIT SOLUCION CONTADOR HEMATOLÓGICO

answer: Ajustarse a la última versión del PBC.-

description: SE PODRIA OFERTAR TAMBIEN CONTADORES HEMATOLOGICOS DE 3 PARTES

title: LOTE 1 ITEM 2

answer: Ajustarse al PBC. Al utilizar reactivos de otra marca, se corren riesgos donde los resultados pueden variar, comprometiendo la exactitud (proximidad al valor verdadero) y la precisión (repetibilidad de los resultados) de los análisis, lo que puede llevar a errores clínicos en el diagnóstico.

description: Se podría admitir: ofertar reactivos compatibles con el equipo en comodato pero que sean de una marca diferente al equipo.

title: Especificaciones Tecnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: En el apartado de especificaciones técnicas, en el lote 59, ítem 1: (Rubéola Test Rápido), donde dice: “Rubéola IgG o IgG/IgM. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.” Se solicita a la convocante considerar la admisión de pruebas con una sensibilidad mínima del 95 %, dado que este nivel de sensibilidad sigue siendo altamente efectivo para los procesos de tamizaje. Además, este valor es ampliamente aceptado y común en este tipo de pruebas dentro de los estándares. La implementación de este criterio no solo garantiza la eficiencia en la detección de casos, sino que también permite una mayor participación de oferentes, asegurando competitividad en la adquisición de reactivos. Por consiguiente, sugerimos que dicho requerimiento quede redactado de la siguiente forma: “Rubéola IgG o IgG/IgM. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 95%. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.”

title: Especificaciones Tecnicas

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description: En el apartado de especificaciones técnicas, en el lote 12, ítem 1: (Test Rápido para Hepatitis A), donde dice: “Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Capacidad de realizar como mínimo 10 y máximo 20 determinaciones por presentación.” Se solicita a la convocante aceptar que la muestra sea dispensada inicialmente en un buffer de extracción antes de su aplicación directa en la tira reactiva o cassette. Este procedimiento permite optimizar la liberación y estabilidad del analito, favoreciendo su adecuada interacción con los componentes reactivos de la prueba,y asegurando una mayor confiabilidad en la detección. Por consiguiente, solicitamos que dicho requerimiento sea redactado de la siguiente forma: “Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes o en un buffer de extracción y de lectura visual. Capacidad de realizar como mínimo 10 y máximo 20 determinaciones por presentación.”

title: Lote 5 / Item 1:

answer: Ajustarse a la última versión del PBC.

description: Se podría admitir reactivos de una marca distinta al equipo ofertado, pero que sea compatible con dicho equipo.

title: Especificaciones Tecnicas

answer: Ajustarse a la última versión del PBC.-

description: En el apartado de especificaciones técnicas, en el lote 22, ítem 1: (TEST MULTIDROGAS PARA ORINA - 5 DROGAS), donde dice: “TEST RÁPIDO PARA 5 DROGAS (COC, THC, BZO, MOP, AMP). Panel de test rápido inmunocromatográfico para detección cualitativa de 5 drogas de abuso. Los reactivos deben tener una sensibilidad no menor al 97%, con certificado de Calidad del país de origen.  Capacidad de realizar entre un mínimo de 20 y un máximo de 30 determinaciones por presentación.” Solicitamos amablemente que se admita una presentación 15 determinaciones por presentación como mínimo, ya que esto no afecta la finalidad de uso del test ni compromete su efectividad. Además, se respetaría el número total de pruebas solicitadas, garantizando el cumplimiento de los requerimientos sin alterar el objetivo del proceso. Esta modificación también favorecería la participación de un mayor número de oferentes, promoviendo un proceso más justo y transparente para todos. Por lo expuesto, solicitamos que dicho requerimiento quede redactado de la siguiente manera: “TEST RÁPIDO PARA 5 DROGAS (COC, THC, BZO, MOP, AMP). Panel de test rápido inmunocromatográfico para detección cualitativa de 5 drogas de abuso. Los reactivos deben tener una sensibilidad no menor al 97%, con certificado de Calidad del país de origen.  Capacidad de realizar entre un mínimo de 15 y un máximo de 30 determinaciones por presentación.”

title: Especificaciones Tecnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: En el apartado de especificaciones técnicas, en el lote 60, ítem 1: (Test Rápido para Hepatitis A), donde dice: para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%. “Rubéola IgM. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: mínimo 20 y máximo 30 determinaciones por presentación.” Se solicita a la convocante considerar la admisión de pruebas con una sensibilidad mínima del 95 %, dado que este nivel de sensibilidad sigue siendo altamente efectivo para los procesos de tamizaje. Además, este valor es ampliamente aceptado y común en este tipo de pruebas dentro de los estándares. La implementación de este criterio no solo garantiza la eficiencia en la detección de casos, sino que también permite una mayor participación de oferentes, asegurando competitividad en la adquisición de reactivos. Por consiguiente, sugerimos que dicho requerimiento quede redactado de la siguiente forma: “Rubéola IgM. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 95%. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: mínimo 20 y máximo 30 determinaciones por presentación.”

title: Especificaciones Tecnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: En el apartado de especificaciones técnicas, en el lote 60, ítem 1: (Test Rápido para Hepatitis A), donde dice: para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%. “Rubéola IgM. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: mínimo 20 y máximo 30 determinaciones por presentación.” Solicitamos amablemente a la convocante la admisión del test inmunocromatográfico rápido que permite la detección simultánea de los anticuerpos IgG e IgM en una misma prueba. Este tipo de prueba es ampliamente utilizado y reconocido como el más adecuado para procesos de tamizaje, dada su eficiencia y confiabilidad. El propósito fundamental del tamizaje es identificar la exposición al virus de la rubéola, asegurando así el cumplimiento de los requisitos necesarios para dicho fin de manera precisa y efectiva. Por consiguiente, sugerimos que dicho requerimiento quede redactado de la siguiente forma: “Rubéola IgM o IgG/IgM. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: mínimo 20 y máximo 30 determinaciones por presentación.”

title: lote 4

answer: Ajustarse a la última versión del PBC.-

description: Lote 4 item 46 y item 47. Considerando que la pruebas comercialmente disponibles para  Chagas, es total ( IgG+IgG) se sugiere a la convocante reformular su requerimiento

title: Especificaciones Tecnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: En el apartado de especificaciones técnicas, en el lote 109, ítem 1 (LÁMINAS ESMERILADAS), donde dice: “Esmerilada, 76 x 26 mm, banda mate. Traslúcidas, libre de ralladuras, con separador de papel sulfito entre láminas. La oferta es por unidad de lámina.” Se solicita amablemente a la convocante la admisión de un rango en las dimensiones de la lámina, ya que cada marca maneja dimensiones específicas con variaciones mínimas entre ellas. Esto ayudaría a evitar que se favorezca a un solo oferente con una marca que coincida exactamente con las dimensiones descritas. Por lo expuesto, solicitamos que dicho requerimiento quede redactado de la siguiente manera: “Esmerilada, mínimas dimensiones de 76 x 26 mm ± 1 mm, banda mate. Traslúcidas, libre de ralladuras, con separador de papel sulfito entre láminas. La oferta es por unidad de lámina.”

title: Especificaciones Tecnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: En el apartado de especificaciones técnicas, en el lote 98, ítem 1 (Gradilla para tubos - 60 tubos), donde dice: “PARA TUBO DE ENSAYO GRANDE, AUTOCLAVABLE. De plástico, para 60 tubos, desmontable, con agujero redondo y aletas que sujetan al tubo.” Se solicita amablemente a la convocante la admisión de gradillas con agujero cuadrado, ya que cada marca maneja diseños distintos. Esto ayudaría a evitar que se favorezca a un solo oferente con una marca que coincida exactamente con el diseño mencionado. Por lo expuesto, solicitamos que dicho requerimiento quede redactado de la siguiente manera: “PARA TUBO DE ENSAYO GRANDE, AUTOCLAVABLE. De plástico, para 60 tubos, desmontable, con agujero redondo o cuadrado y aletas que sujetan al tubo.”

title: ESPECIFICACIONES TECNICAS..

answer: Ajustarse a la última versión del PBC.-

description:
LOTE 3 - Grupo 1:  Test Rápido SARS COV 2  ( en especificaciones tecnicas)
DICE ASI:    PRUEBA RAPIDA INMUNOCROMATOGRAFICA PARA DETECCION CUALITATIVA DE ANTIGENO ESPECIFICO DEL SARS COV2 PARA HISOPOADO NASOFARINGEO con una sensibilidad    ( NO SALE LA INFORMACION - LA DESCRIPCION COMPLETA. )

title: Especificaciones Tecnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: En el apartado de especificaciones técnicas, en el lote 108, ítem 1 (LÁMINAS PORTAOBJETO), donde dice: “De vidrio, 76 x 26 mm. Traslúcidas, libre de ralladuras y que no se peguen entre sí. La oferta es por unidad de lámina.” Se solicita amablemente a la convocante considerar la admisión de láminas lisas, ya que ofrecen una observación clara y sin distorsiones, permitiendo visualizar detalles finos y con alta resolución al no alterar la trayectoria de la luz que atraviesa la muestra. Además, son compatibles con una amplia variedad de técnicas de microscopía, incluidas la microscopía de campo claro y de contraste de fases. Por lo expuesto, solicitamos que dicho requerimiento quede redactado de la siguiente manera: “De vidrio, 76 x 26 mm. Traslúcidas o lisas, libre de ralladuras y que no se peguen entre sí. La oferta es por unidad de lámina.”

title: LOTE 1 - ITEM 76 - KIT PARA EQUIPO DE ELECTROLITOS Y GASÓMETRO

answer: Ajustarse a la última versión del PBC.

description: Con relacion a este punto Certificado FDA o CE (al menos uno de ellos). Con relación  a este punto se podría ampliar este pedido de Certificación con ISOS 9001 Y/O 13485 o Libre Venta de manera a poder ampliar mas la participacion de potenciales oferentes.

title: Plan de entrega de los bienes

answer: FAVOR AJUSTARSE AL DEL PLIEGO DE BASES Y CONDICIONES. LOS TIEMPOS FUERON ESTIPULADOS DE ACUERDO A LAS INMINENTES NCESIDADES DE LOS SERVICIOS

title: Lote 4 – Ítem 29 al 49

answer: Ajustarse a la última versión del PBC.

description: Se observa que la convocante, en las Especificaciones técnicas de los equipos para Reactivos e insumos para la sección de Drogas Terapeúticas para Marcadores de Infección y Metabólicos del Servicio de Bioquímica Clínica, solicita proporcionar gráficos de Westgard. Teniendo en cuenta que las reglas de Westgard son un conjunto de criterios aplicados a los gráficos de Levey-Jennings para identificar y evaluar posibles errores y desviaciones en los datos, y es el gráfico de Levey-Jennings el que es utilizado para representar los resultados de controles de calidad en laboratorios clínicos, mostrando los valores de las mediciones en comparación con el valor promedio esperado y sus desviaciones estándar, se solicita a la convocante modificar esta información, ya que el gráfico solicitado para ese fin es el de Levey-Jennings, ya que no existe tal gráfico de Westgard

title: Capacidad técnica – Para Reactivos

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description: Se consulta nuevamente a la convocante si con Certificado de Trámite se refiere a la mesa de entrada de solicitud de registro, ya que deseamos prepararnos correctamente para la oferta.

title: Lote 1 - Item 2

answer: Ajustarse a la última versión del PBC.

description: : Se podría admitir equipos de soporte para realizar las determinaciones que el equipo integrado no pudiese realizar

title: lote 2

answer: Ajustarse a la última versión del PBC.-

description: Lote 2 item 1 favor aclarar que tipo de equipo automatizado precisa, lector y lavador? Hacemos esta consulta con el proposito de evitar inconvenientes al momento de la evaluacion o suministro

title: ESPECIFICACIONES TECNICAS Lote 1, Ítem 77-Kit

answer: Misma respuesta que la consulta 3 y 4.-

description:
Podría la convocante especificar en el Tipo de Muestra: Controles QC (Material Acuoso) si se refiere a controles de calidad internos o externos.
Si se refiere a controles internos, podría aceptar controles de material acuoso pero integrados dentro del sistema de medición del pack.

title: Especificaciones Tecnicas

answer: Ajustarse a la última versión del PBC.-

description: En el apartado de especificaciones técnicas, en el lote 59, ítem 1: (Rubeola Test Rápido), donde dice: “Rubeola IgG. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.”  Se solicita amablemente a la convocante admitir una presentación que permita la identificación de ambos anticuerpos, tanto IgG como IgM en el mismo test, ya que esto no afectaría la finalidad de uso del test ni compromete su efectividad. Además, se respetaría el número total de pruebas solicitadas, garantizando el cumplimiento de los requerimientos sin alterar el objetivo del proceso. Esta modificación también favorecería la participación de un mayor número de oferentes, promoviendo un proceso más justo y transparente para todos. Por lo expuesto, solicitamos que dicho requerimiento quede redactado de la siguiente manera: “Rubeola IgG o IgG/IgM. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.”

title: Especificaciones Tecnicas

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description: En el apartado de PLAN DE ENTREGA DE LOS BIENES, en el punto REACTIVOS E INSUMOS EN GENERAL (no requieren equipos en comodato), donde menciona: “1ra ENTREGA 20% DE LA CANTIDAD MÍNIMA: El oferente tendrá un plazo de hasta 20 (veinte) días calendario para la entrega de los productos adjudicados a partir de la recepción de la Orden de Compra, la cuál será emitida dentro de los 20 (VEINTE) días de haber firmado el contrato.” Se solicita a la convocante la ampliación del plazo de entrega a 45 días calendario. Esta extensión resulta fundamental debido a los tiempos requeridos para la fabricación de los insumos, los procesos logísticos de envío y los trámites aduaneros, los cuales, en muchos casos, pueden superar el mes. Ampliar el plazo de entrega permitirá la participación de un mayor número de oferentes con capacidad de suministro, fomentando una competencia más equitativa. De esta manera, se evitaría restringir la adjudicación únicamente a proveedores con stock disponible inmediata, asegurando condiciones justas y promoviendo la adquisición de productos con los más altos estándares de calidad. Por lo expuesto, solicitamos que dicho requerimiento quede redactado de la siguiente manera: “1ra ENTREGA 20% DE LA CANTIDAD MÍNIMA: El oferente tendrá un plazo de hasta 45 (CUARENTA Y CINCO) días calendario para la entrega de los productos adjudicados a partir de la recepción de la Orden de Compra, la cuál será emitida dentro de los 20 (VEINTE) días de haber firmado el contrato.”

title: LOTE 4

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Se solicita separar el lote 4 donde se encuentra contador hematológico, equipo de química clínica, equipo de inmunología, equipo para gasometría y crasis para fomentar la mayor participación de oferentes.

title: Lote Nº 115 Ítem 12 Puntas

answer: Ajustarse a la última versión del PBC.

description: ¿Podemos ofertar Puntas AMARILLAS con volumen de 2-200 ul para pipeta automática variable y fija de hasta 200 ul? Aclaramos que el mismo cumple con el rango solicitado en el PBC

title: Especificaciones Tecnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: En el apartado de especificaciones técnicas, en el lote 109, ítem 1 (LÁMINAS ESMERILADAS), donde dice: “Esmerilada, 76 x 26 mm, banda mate. Traslúcidas, libre de ralladuras, con separador de papel sulfito entre láminas. La oferta es por unidad de lámina.” Se solicita respetuosamente a la entidad convocante la aceptación de láminas esmeriladas sin separador de papel sulfito entre láminas, dado que la ausencia de este componente no afecta la funcionalidad ni el propósito de uso del producto en el laboratorio. Esta consideración permitiría ampliar la participación de oferentes con productos de calidad equivalente, fomentando una mayor competitividad en el proceso de adquisición. Por lo expuesto, solicitamos que dicho requerimiento quede redactado de la siguiente manera: “Esmerilada, 76 x 26 mm, banda mate. Traslúcidas, libre de ralladuras, con o sin separador de papel sulfito entre láminas. La oferta es por unidad de lámina.”

title: Lote Nº 115 Ítem 14 Puntas

answer: Ajustarse a la última versión del PBC.

description: ¿Podemos ofertar Puntas con volumen de 500-5.000 ul para pipeta automática variable y fijo con volúmenes de hasta 5.000 ul? Aclaramos que el mismo cumple con el rango solicitado en el PBC

title: Lote 115 – ítem 2  Pipeta Autom. variable

answer: Ajustarse a la última versión del PBC.-

description:
¿Podemos ofertas Pipeta automática con volumen variable de 5 – 50 ul
 ¿ El mismo cumple con el rango solicitado.

title: Especificaciones Tecnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: En el apartado de especificaciones técnicas, en el lote 162, ítem 1 (PIPETA PASTEUR), donde dice: “DE PLASTICO, DE 3 ML DE VOLUMEN.” Se solicita amablemente a la convocante considerar la aceptación de un volumen de 7 mililitros, con una escala de graduación que varía de 0,5 a 3 ml. Esta flexibilidad en la capacidad permite cubrir un rango de volumen mayor sin comprometer los requisitos específicos de las mediciones. Además, esta capacidad es ampliamente utilizada en numerosos laboratorios y se ajusta a los protocolos estandarizados en diversas prácticas de laboratorio. Por lo expuesto, solicitamos que dicho enunciado quede redactado de la siguiente forma: “DE PLASTICO, DE 3 ML COMO MÍNIMO DE VOLUMEN.”

title: Análisis de precios ofertados

answer: Se aclara que la Resolución DNCP N° 454/2024 establece rangos de variación de precios que el Comité de Evaluación debe considerar para requerir la explicación detallada de la composición de precios, es decir, si el precio ofertado se encuentra fuera de los rangos permitidos por la DNCP, el comité de evaluación necesariamente debe solicitar la composición de los precios y analizarlos para sustentar su razonabilidad conforme lo menciona el Art. 4° "El análisis de los precios, con esta metodología, será aplicado a cada ítem, rubro o partida que componga la oferta y en cada caso deberá ser debidamente fundada la decisión adoptada por la Convocante en el ejercicio de su facultad discrecional."

description:
Solicitud de Aclaración:
Según Resolución DNCP Nº 454/2024  Art. 4to. se determina que:
(…) En contrataciones en general: cuando la diferencia entre el precio ofertado y el precio referencial sea superior al 25% para ofertas por debajo del precio referencial y del 15% para ofertas que se encuentren por encima del referencial establecido por la convocante y difundido con el llamado a contratación.

Según la evaluación de la licitación Nº 127/2024 ID Nº 453750 cuya apertura se realizó en fecha 26/11/2024 puede observarse que algunos lotes fueron adjudicados con márgenes tanto inferior como superior, del 35 %, respecto a los precios referenciales.

Consultamos al respecto si en cuanto a los porcentajes mencionados en los párrafos anteriores la Resolución mencionada arriba sigue siendo válida o si debemos considerar
que serán válidos también precios que estén un 35 % tanto por debajo como por encima
del precio referencial.

title: Lote 4 – Item 1

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: Se solicita amablemente a la convocante modificar el requisito de “25 (veinticinco) equipos automatizados de 3 partes para hemograma con gradillas en rack”, ya que los equipos con recuento diferencial de 3 partes no admiten el uso de gradillas en rack, sino que cuentan con una sonda que aspira cada muestra de manera individual. Por ello, se sugiere reformular el requisito como “25 (veinticinco) equipos automatizados de 3 partes para hemograma”. De lo contrario, se solicita amablemente modificar las especificaciones técnicas de tal modo a solicitar un Analizador de Hematología con Capacidad de realizar recuento diferencial de 5 partes, con la consecuente modificación del precio referencial.

title: Lote Nº 115 Ítem 11 Puntas

answer: Ajustarse a la última versión del PBC.

description: ¿Podemos ofertar Puntas con volumen de 2-200 ul para pipeta automática variable y fija de hasta 200 ul? Aclaramos que el mismo cumple con el rango solicitado en el PBC

title: Lote 36, Item 1

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: Se podría admitir una presentación de colorante Wright de presentación liquida de 1000mL.

title: Lote 4, Item 1 contadores hematológicos no cuenten con racks de muestras

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Se solicita a la convocante a tener en cuenta que, para los 30 contadores hematológicos requeridos para los hospitales distritales y centros de salud, considerar que los mismos no cuenten con racks de muestras, de esta manera permitiría una libre competencia entre oferentes, ya que la mayoría de equipos de 3 partes no cuentan con autosampler de muestras.

title: Lote 4, Ítem 49 equipo de soporte para poder realizar 5 a 6 determinaciones.

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Se solicita a la convocante tener en cuenta la opción de tener un equipo de soporte para poder realizar 5 a 6 determinaciones que el módulo de inmunología no pueda realizar, esto permitiría una mejor oportunidad de competencia.

title: LOTE 23 ITEM 1

answer: Ajustarse al PBC. Los equipamientos y sus características fueron ajustados dentro de las especificaciones técnicas elaboradas, conforme a los requerimientos técnicos y a las necesidades específicas del servicio. Utilizar equipos independientes para un solo test, retrasaría la capacidad técnica del servicio.

description: SE PODRIA ADMITIR PRESENTACIONES DE 1X5mL Y COMPLETAR LA CANTIDAD REQUERIDA

title: Lote 4  – Item 1

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: Se solicita amablemente a la convocante modificar el requisito de “25 (veinticinco) equipos automatizados de 3 partes para hemograma con gradillas en rack”, ya que los equipos con recuento diferencial de 3 partes no se admiten el uso de gradillas en rack, sino que cuentan con una sonda que aspira cada muestra de manera individual. Por ello, se sugiere reformular el requisito como “25 (veinticinco) equipos automatizados de 3 partes para hemograma”. De lo contrario, se considera que se está direccionando la oferta para oferentes que cuentan estas características. Hacemos esta solicitud amparados en los lineamientos de la Ley 7021/22 y Decreto 2264/24 en su Art.58. establece “ Especificaciones técnicas. Las especificaciones técnicas que deban contener las bases de la contratación se establecerán con la mayor amplitud de acuerdo con la naturaleza específica del contrato, con el objeto de que concurra el mayor número de oferentes. Sin embargo, deberán ser lo suficientemente claras, objetivas e imparciales, para evitar favorecer indebidamente a algún participante. Cuando los tipos conocidos de materiales, artefactos o equipos únicamente puedan ser caracterizados total o parcialmente mediante signos distintivos no universales, únicamente se hará a manera de referencia, procurando que la alusión se adecue a estándares internacionales comúnmente aceptados”

title: Lote 4 - Item 2 al 26 - Especificaciones técnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

title: Especificaciones Tecnicas

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description: En el apartado de especificaciones técnicas, en el lote 11, ítem 1: (Chagas Test Rápido), donde dice: “Test rápido inmunocromatográfico IgG/IgM o IgG para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.” Se solicita a la convocante aceptar una sensibilidad mínima del 94% en las pruebas de tamizaje, dado que este nivel garantiza un desempeño óptimo para la detección de casos positivos y cumple con los estándares internacionales aplicables a este tipo de diagnóstico. Por lo expuesto, solicitamos que dicho requerimiento quede redactado de la siguiente manera: “Test rápido inmunocromatográfico IgG/IgM o IgG para tamizaje. Los reactivos deben tener una sensibilidad no menor al 94%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.”

title: LOTE 1  -  Item 28 y 29

answer: Ajustarse a la última versión del PBC.

description:
C3  y C4  -   Hacer notar la diferencia en las cantidades solicitadas, ya que normalmente suelen ser las mismas cantidades, derivado de los pedidos médicos.
Producto	Cant.Min	Cant. Max.
C3	            4.000	8.000
C4	          40.000	80.000

title: Lote 4 – Item 48

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Se solicita amablemente a la convocante eliminar estos parámetros eAVDO2, AGOER, Hct (C), PAO2t, RItPHtPCO2t, H+t, AaDO2t, ya que se considera que con otros parámetros como el índice PaO₂/FiO₂, el índice de oxigenación (IO), la saturación venosa mixta de oxígeno (SvO₂), Gradiente Alveolo-Arterial de Oxígeno (A−aA-aA−a DO₂), Qsp/Qt (Shunt fisiológico), Índice respiratorio (RI) y el contenido arterial de oxígeno (CaO₂) se puede evaluar integralmente el estado de salud del paciente. Además, así también se estaría promoviendo una mayor participación de oferentes, ya que son parámetros que no todos los gasómetros son capaces de calcular.

title: lote 6 item 1

answer: Ajustarse a la última versión del PBC.-

description: Lote 6 item 1 Hemoglobina glicosilada (A1c), favor aclarar si la metodología requerida es HPLC

title: Lote 1 / Item 30:

description: Se podría aceptar un equipo de soporte para las determinaciones que el equipo de inmunología no pueda realizar, que cuente con la misma tecnología por CLIA; de esta manera de podría tener una mejor oportunidad de competencia entre los oferentes. Favor de no responder ajustarse a la versión nueva del pliego. Se solicita una aclaración.

title: Especificaciones Tecnicas

answer: Ajustarse al PBC. Las EETT fueron suficientemente estudiadas en base a la necesidad para los parámetros de sensibilidad y especificidad de los reactivos solicitados

description: En el apartado de especificaciones técnicas, en el lote 60, ítem 1: (Rubeola Test Rápido), donde dice: “Rubeola IgM. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.”  Se solicita a la convocante admitir una sensibilidad del 95% como mínimo ya que esta sensibilidad sigue siendo altamente efectiva para pruebas de tamizaje, y es la sensibilidad más común para estos tipos de test. Al establecer este estándar, se facilitará la participación de un mayor número de oferentes, lo que promoverá la competencia y, potencialmente, mejores condiciones en la adquisición de los productos. Por lo expuesto, solicitamos que dicho requerimiento quede redactado de la siguiente manera: “Rubeola IgM Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 95%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.”

title: Lugar de entrega

answer: Ajustarse a la última versión del PBC.-

description: Solicitamos amablemente a la convocante que aclare cada lote a que Hospital será destinado los insumos, solo sera para el HOSPITAL REGIONAL DE ENCARNACION

title: Especificaciones Tecnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

title: Lote 4 / Item 1

answer: Ajustarse a la última versión del PBC.

description: Se podría separar el lote de contador hematología del lote del equipo de química e inmunología -

title: Pág. 30 - Lugar de entrega

answer: Favor remitirse a la Consulta N° 5.

description: Solicitamos respetuosamente a la convocante que aclare a qué hospital se destinará cada lote, con el fin de tener claridad en la distribución.

title: Lote Nº 154 Vaso Precipitado

answer: Ajustarse al PBC. La consulta cumple con lo mínimo requerido.

description: ¿Podemos ofertar vaso precipitado de 600 ml. de capacidad?

title: Especificaciones Tecnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: En el apartado de especificaciones técnicas, en el lote 12, ítem 1: (Test Rápido para Hepatitis A), donde dice: “Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Capacidad de realizar como mínimo 10 y máximo 20 determinaciones por presentación.” Se solicita a la entidad convocante aceptar que la muestra sea dispensada inicialmente en un buffer de extracción antes de su aplicación directa en la tira reactiva o cassette. Este procedimiento permite optimizar la liberación y estabilidad del analito, favoreciendo su adecuada interacción con los componentes reactivos de la prueba. Como resultado, se incrementa la sensibilidad y precisión del diagnóstico, reduciendo la posibilidad de falsos negativos y asegurando una mayor confiabilidad en la detección. Esta metodología se encuentra respaldada por principios técnicos de diagnóstico inmunocromatográfico y es una práctica recomendada en el desarrollo de pruebas rápidas de alta calidad. Por consiguiente, solicitamos que dicho requerimiento sea redactado de la siguiente forma: “Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes o buffer de extracción y de lectura visual. Capacidad de realizar como mínimo 10 y máximo 20 determinaciones por presentación.”

title: Especificaciones Tecnicas

answer: Ajustarse a la última versión del PBC.-

description: En el apartado de especificaciones técnicas, en el lote 26, ítem 1: (Test para sangre oculta), donde dice: “Test rápido inmunocromatográfico en un solo paso. Método inmunológico para la detección de hemoglobina en las heces frescas. Capacidad de realizar entre 25 y 50 determinaciones por presentación.”  Se solicita amablemente a la convocante admitir una presentación de 20 determinaciones por caja como mínimo, ya que esto no afecta la finalidad de uso del test ni compromete su efectividad. Además, se respetaría el número total de pruebas solicitadas, garantizando el cumplimiento de los requerimientos sin alterar el objetivo del proceso. Esta modificación también favorecería la participación de un mayor número de oferentes, promoviendo un proceso más justo y transparente para todos. Por lo expuesto, solicitamos que dicho requerimiento quede redactado de la siguiente manera: “Test rápido inmunocromatográfico en un solo paso. Método inmunológico para la detección de hemoglobina en las heces frescas. Capacidad de realizar entre 20 y 50 determinaciones por presentación.”

title: Lote 4 / Item 1:

answer: Ajustarse a la última versión del PBC.

description: : Se podría admitir contadores de 3 partes sin gradillas de rack de muestras.

title: Especificaciones Tecnicas

answer: Ajustarse a la última versión del PBC.

description: En el apartado de especificaciones técnicas, en el lote 11, ítem 1: (Chagas Test Rápido), donde dice: “Test rápido inmunocromatográfico IgG/IgM para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.” Se solicita amablemente a la convocante la admisión del Test rápido inmunocromatográfico que detecta la presencia de anticuerpos IgG, ya que este tipo de prueba es la más adecuada y utilizada para procesos de tamizaje. Dado que el objetivo principal del tamizaje es identificar la exposición al parásito Trypanosoma cruzi, este test cumple con los requisitos necesarios para dicho fin de manera eficiente y efectiva. Por lo expuesto, solicitamos que dicho requerimiento quede redactado de la siguiente manera: “Test rápido inmunocromatográfico IgG/IgM o IgG para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%, con certificado de Calidad del país de origen. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.”

title: Lote 4 - Ítem 1

answer: Ajustarse a la última versión del PBC.

description: Con respecto a los contadores hematologicos para los hospitales distritales, se podría ofertar 5 contadores hematologicos de 5 partes sin gradilla de rack de muestras y 25 equipos de 3 partes para los centros de salud y USF

title: LOTE 43 ITEM 1

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description: Se consulta a la convocante si el Anticoagulante Fluoruro que solicita listo para usar, es en presentación de gotero?

title: Pág. 30 – Plan de entrega de los bienes

answer: Ajustarse al PBC. Se solicita respetar los tiempos de entrega de los productos teniendo en cuenta el stock critico de reactivos en los servicios de salud.

title: Capacidad técnica – Para Reactivos

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: Teniendo en cuenta de que la transición de LCSP a DINAVISA es reciente, y que muchos productos se registran por primera vez, no existe Constancia de Renovación de Trámite y tampoco Certificado de trámite, se solicita extraordinariamente aceptar la Mesa de Entrada de la Solicitud de Registro Sanitario, ya que es el único documento disponible a la fecha. Esta medida ya ha sido adoptada en otros llamados como el de LPN N° 07/2025 ID 458529.

title: Lote Nº 115 Ítem 14 Puntas

answer: Ajustarse a la última versión del PBC.

description: ¿Podemos ofertar en una presentación de bolsa x 200 unidades como mínimo?

title: LOTE 4 ITEM 2

answer: Ajustarse al PBC. Al utilizar reactivos de otra marca, se corren riesgos donde los resultados pueden variar, comprometiendo la exactitud (proximidad al valor verdadero) y la precisión (repetibilidad de los resultados) de los análisis, lo que puede llevar a errores clínicos en el diagnóstico.

description: Se podría admitir: ofertar reactivos compatibles con el equipo en comodato pero que sean de una marca diferente al equipo.

title: Lote 4 item 48

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description: Se solicita a la convocante no tener en cuenta la determinación de Chagas IgM, esto debido a su nula utilidad clínica, debido a que no forma parte de los algoritmos de diagnóstico, esto porque la medición de Chagas IgM no se encuentra sugerida para el diagnóstico de enfermedad aguda. Se solicita una aclaración al respecto, favor contestar sin colocar ajustarse al pliego como respuesta.

title: Lote 1 - Item 1

answer: Ajustarse a la última versión del PBC.

description: Se podría separar el lote de contador hematología del lote de equipo integrado de química-inmuno.

title: lote 1 item 76

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Lote 1 item 76 Tipo de Muestra: Sangre Total, Plasma,Suero. Permitirá la convocante “muestras: Sangre completa (arterial, venosa, mixta venosa y capilar); líquido pleural; dializado”. Considerando que Plasma o suero no es tipo de muestra obtenida para gasometria

title: Lote 2 – ítem 1

answer: Ajustarse a la última versión del PBC.

description: Se solicita a la convocante, establecer las especificaciones técnicas mínimas requeridas para el Kit para equipo de ELISA.

title: Especificaciones Tecnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: En el apartado de especificaciones técnicas, en el lote 60, ítem 1: (Rubéola Test Rápido), donde dice: “Rubéola IgM o IgG/IgM. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.” Se solicita a la convocante considerar la admisión de pruebas con una sensibilidad mínima del 95 %, dado que este nivel de sensibilidad sigue siendo altamente efectivo para los procesos de tamizaje. Además, este valor es ampliamente aceptado y común en este tipo de pruebas dentro de los estándares. La implementación de este criterio no solo garantiza la eficiencia en la detección de casos, sino que también permite una mayor participación de oferentes, asegurando competitividad en la adquisición de reactivos. Por consiguiente, sugerimos que dicho requerimiento quede redactado de la siguiente forma: “Rubéola IgM o IgG/IgM. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 95%. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.”

title: Lote 5 – Especificaciones técnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Se solicita a la convocante, para los ítems 2, 4, 6, 7, 11,16,17,18 modificar las especificaciones técnicas y redactarlas de la misma manera que el ítem 1 donde se establece “Método enzimático, colorimétrico o de punto final.” , ya que las tres son metodologías válidas, de tal forma a evitar confusiones por parte de los oferentes.

title: Pág. 4 - Abastecimiento simultáneo

answer:
No es viable la modalidad de abastecimiento simultáneo, para los Laboratorios de la DRNL, por la infraestructura, espacio físico, cantidad de RRHH, productividad, y otros, ya que se deben manejar más de un equipo, probablemente más de un software, más de una marca de reactivo, depósitos refrigerados, etc.,   además, no es posible, instalar equipos de gran porte y desinstalar de manera rápida al terminar uno de ellos y utilizar el otro, lleva días instalar y desinstalar dichos equipos de gran porte, y no se puede paralizar las prestaciones del servicio.

El artículo 43 de la Ley 7021/22 reglamenta este tipo de modalidad para aquellos casos en que ningún proveedor pueda proveer o ejecutar la totalidad de los bienes o servicios. Por experiencias previas con la modalidad de las LPN, para los lotes con equipos en COMODATO, no es viable la modalidad de abastecimiento simultáneo, para los Laboratorios de la DRNL.
Citamos inconvenientes por los cuales, no es viable esta modalidad, cuando lo que se solicita es trabajar en la modalidad de reactivos con equipos en comodato.
-	Si el llamado es Abastecimiento Simultáneo, serán adjudicados hasta 3 oferentes; implica que se tendrán mínimamente 3 equipos, de 3 diferentes proveedores.
-	No se cuenta con espacio físico suficiente para esa cantidad de equipos, ni RRHH que trabaje en simultáneo, con más cantidad de equipos.
-	No es viable ya que los reactivos de la marca de un proveedor no pueden ser utilizados en otro equipo de otra marca del otro proveedor, a fin de garantizar las Buenas prácticas de Laboratorio, la Garantía de calidad y buena procedencia de los reactivos, equipos deben ser de la misma marca.
-	Los controles, calibradores, consumibles, insumos, etc., deberían entregar 3 proveedores, acarrea la dificultad por no contar con espacio físico, ni personal necesario para el control y monitoreo.

description: Solicitamos a la convocante ajustar el sistema de suministro a un abastecimiento simultáneo para todos los lotes, con el fin de garantizar la adecuada provisión de los insumos requeridos.

title: LOTE 1 -  Item 73 y 75

answer: Ajustarse a la última versión del PBC.

description:
Hacer notar que solicitan la misma prueba  en ambos ítems, con diferentes cantidades:
Item	Producto	                                      Cant. Min	Cant. Max.
73	PROTEINA C REACTIVO (P.C.R.)	15.000	30.000
75	PCR TURBIDIMETRIA	                35.000	70.000

title: Lote 4

answer: Ajustarse a la última versión del PBC.

description: Para las EETT del gasómetro menciona lo siguiente: "Mínimas características: electrodos libres de mantenimiento para determinaciones de electrolitos: Na, K, Cl y Ca. Para Gasometría pH, pCO2, pO2, Sat de O2, Hemoglobina total, Hematocrito. H+, CHCO3, CTCO2, BE, BEACT, BESS, BB, pHST, CHCO3ST, pAO2, NCa, QS/QT, p50, sO2 (C), AaDO2, a/AO2, eAVDO2, AGOER, Hct (C), PAO2t, RItPHtPCO2t, H+t, AaDO2t", solicitamos a la convocante aceptar un gasómetro con las siguientes características "Mínimas características: electrodos libres de mantenimiento para determinaciones de electrolitos: Na, K, Cl y Ca. Para Gasometría pH, pCO₂, pO₂, Hct, pH, Cl, GLU, LAC, Hb, HCO₃, BE, BE-B, BE-ECF, TCO₂, AG, AG(K), O₂ Sat, O₂ Ct, SBC, nCa, TCa, pO₂%, A, AaDO₂ , a/A", ya que la mayoría de los gasómetros cuentan con estas características. En los ítems solicita el item 48 Chagas IgM, solicitamos a la convocante aceptar sin esta determinación ya que es una determinación que realizan muy pocos analizadores.

title: ESPECIFICACIONES TECNICAS Lote 1, Ítem 77-Kit

answer: Misma respuesta que la consulta 3 y 4.-

description: Podría la convocante especificar en el Tipo de Muestra: Controles QC (Material Acuoso) si se refiere a controles de calidad internos o externos.

title: Lote 4, Item 1

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES

description: Se solicita a la convocante poder ofertar el lote de contadores hematológicos en un lote separado del equipo de química, de esta manera se podría tener una mejor competencia entre oferentes

title: Lote 2 / Item 1:

answer:
Ajustarse a la última versión del PBC.
Dicha determinación se realizará por el Método ELISA u otro de tecnología similar.

description: La determinación de Antigeno Dengue – NS1 podría permitirse realizar por biología molecular o por inmunocromatografia

title: LOTE 23 ITEM 1 ANTIGENO FEBRILES

answer: Ajustarse al PBC. Los equipamientos y sus características fueron ajustados dentro de las especificaciones técnicas elaboradas, conforme a los requerimientos técnicos y a las necesidades específicas del servicio.

description: SE PODRIA ADMITIR PRESENTACIONES DE 1X5mL Y COMPLETAR LA CANTIDAD REQUERIDA

title: lote 4

answer: : Ajustarse al PBC. Las EETT fueron suficientemente estudiadas teniendo en cuenta la necesidad del servicio, en cuanto a la demanda de las pruebas, capacidad de RRHH e infraestructura.

description:
crasis sanguínea Centro de Salud y USF
requiere: equipo en comodato semiautomatizado; de 4- 6 canales.  Se Consulta si aceptarían: Crasis Sanguínea Reactivos para estudio de Crasis Sanguínea, con provisión de 5 (cinco) equipo en comodato semiautomatizado; de 2- 6 canales

title: LOTE 5 ITEM 1 AL 18

answer: misma respuesta que Consulta 44

description: ESPECIFICAR SI SE TRATA DE ESPECTROFOTOMETROS O EQUIPOS AUTOMATIZADOS

title: VENCIMIENTO

answer: Ajustarse a la última versión del PBC.

description:
En la página 8 bajo el punto B PARA LOS PRODUCTOS QUE NO TENGAN FECHA DE VENCIMIENTO (Drogas Puras, Productos Inertes):  se solicita Constancia de origen.
Consulta: ¿ Se puede presentar para esta exigencia el Certificado de Representación de la marca ofertada debidamente registrada y legalizada ?
Nota: Para el caso en cuestión de los productos considerados inertes, como ser: gradillas de plástico, propipetas de plástico y de goma, cámara de Neubauer de vidrio, frascos para esputo u orina, pipetas automáticas, puntas para pipetas, cronómetros, láminas cubre-objetos, morteros de porcelana, placas para tipificación, placa para VDRL,  sellador de tubos, etc., etc., (es decir: productos de metal, de vidrio, de plástico, de porcelana y algodón) existe el DICTAMEN A.J. DGGIES Nº 506/2015 que hace referencia a lo antedicho.

title: LOTE 4ITEM 49 ESPEIFICACIONS TECNICAS

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description: Se solicita a la convocante tener en cuenta la opción de tener un equipo de soporte para poder realizar 5 a 6 determinaciones que el módulo de inmunología no pueda realizar, con la misma tecnología del equipo ofertado; esto permitiría una mejor oportunidad de competencia. Se solicita una aclaración al respecto, favor contestar sin colocar ajustarse al pliego como respuesta.

title: Lote 1

answer: Ajustarse al PBC. Las EETT solicitadas fueron estudiadas y elaboradas teniendo en cuenta la necesidad de la adquisición de bienes y están ajustados dentro de los requerimientos propios de los estudios laboratoriales realizados y en base a experiencias previas. El gasómetro será solo para un Hospital Distrital.

description: En las EETT menciona "Se permitirá un segundo equipo auxiliar con tecnología similar para un máximo de 5 o 6 tests", solicitamos amablemente a la convocante aceptar un máximo de 8-10 tests. Igualmente solicitamos a la convocante aclarar cuáles serían las EETT del gasómetro ya que en las EETT solo menciona especificaciones para gasometría del hospitales distritales.

title: Lote 4 / Item 48

description: Se podría ofertar el lote de gasometría por separado, de esta manera de podría tener una mejor oportunidad de competencia entre los oferentes. Favor de no responder ajustarse a la versión nueva del pliego. Se solicita una aclaración

title: Norma de Calidad

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description:
Consultar a la convocante tener en cuenta la presentación e implementación de la norma de calidad ISO 9001:2015.
Consideramos que este documento es fundamental, ya que representa el compromiso con la calidad de los productos ofrecidos y certifica que la empresa oferente cumple con estándares internacionales, garantizando productos de vanguardia para su implementación y uso en el área de la salud.

title: Lote 1 - Item 77

answer: Ajustarse a la última versión del PBC.

description: Se podría separar en un lote distinto el equipo de crasis sanguínea del de equipos integrados de química-inmunologia

title: Lote 4 / Item 49:

answer: Ajustarse a la última versión del PBC.

description: Se podría separar en un lote distinto el equipo de crasis sanguínea, distinto del equipo de química/inmunología y hematología

title: Especificaciones Técnicas

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description: Se solicita amablemente a la convocante si puede aclarar cómo es el "Certificado de Calidad del País de Origen"?

title: Pág. 7 – Muestras

answer: Favor remitirse a la Consulta N° 8.

description: Solicitamos amablemente a la convocante que excluya la solicitud de muestras de los lotes ofertados, como se hizo en llamados anteriores.

title: lote 4

answer: Ajustarse al PBC. Las EETT fueron suficientemente estudiadas teniendo en cuenta la necesidad del servicio, en cuanto a la demanda de las pruebas, capacidad de RRHH e infraestructura.

description:
LOTE 4, ÍTEM 27, item 28. Consultar si se puede colocar equipo auxiliar al equipo de química
 para realizar ambos test

title: LOTE 143 -  ITEM 1 - TUBOS DE CENTRÍFUGA CONICOS DE PLÁSTICO

answer: Ajustarse al PBC. El PBC menciona lo siguiente: "de 12 mililitros de capacidad como mínimo".

description: CONSULTAR EL VOLUMEN DE CAPACIDAD DE LOS TUBOS A OFERTAR

title: autorizacion del fabricante

answer: El oferente deberá ajustarse a las formalidades requeridas en la Sección Requisitos de Participación y Criterios de Evaluación.-

description: Requiere:La autorización deberá ser presentada en idioma castellano o en su defecto acompañada de su traducción oficial, realizada por un traductor público matriculado en la República del Paraguay. Así también cada autorización debe indicar a que ítem corresponde. Aceptara la convocante copia de la autorización del fabricante, sin legalización por consulado paraguayo o apostillado, inscripto en el registro publico de comercio?

title: Lote 23 / Item 1:

description: : Se podría admitir presentaciones de 5mL

title: Especificaciones Tecnicas

answer: AJUSTARSE A LA ULTIMA VERSION DEL PBC.

description: En el apartado de especificaciones técnicas, en el lote 60, ítem 1: (Rubéola Test Rápido), donde dice: “Rubéola IgM o IgG/IgM. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 98%. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.” Se solicita a la convocante considerar la admisión de pruebas con una sensibilidad mínima del 95 %, dado que este nivel de sensibilidad sigue siendo altamente efectivo para los procesos de tamizaje. Además, este valor es ampliamente aceptado y común en este tipo de pruebas dentro de los estándares. Por consiguiente, sugerimos que dicho requerimiento quede redactado de la siguiente forma: “Rubéola IgM o IgG/IgM. Test rápido inmunocromatográfico para tamizaje. Los reactivos deben tener una sensibilidad no menor al 95%. Las muestras deben ser depositadas directamente en las tiras o casettes y de lectura visual. Presentación de entrega: Capacidad de realizar como mínimo 20 y máximo 30 determinaciones por presentación.”

title: Lote 1 / Item 76

description: : Se podría ofertar el lote de gasometría por separado, de esta manera de podría tener una mejor oportunidad de competencia entre los oferentes. Favor de no responder ajustarse a la versión nueva del pliego. Se solicita una aclaración

title: P.B.C.

answer: Ajustarse a la última versión del PBC.

description: Solicitamos a la convocante proveernos de informaciones completas en especial de las páginas de especificaciones técnicas debido a que no sale por completo en el pliego.

title: Lote 4 – Especificaciones técnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Se observa que la convocante solicita “6 equipos de química clínica y 6 equipos para inmunología”. Sin embargo, en especificaciones técnicas adicionales, observamos que se solicitan 5 equipos de química clínica y 5 equipos para inmunología. Solicitamos amablemente a la convocante aclarar la cantidad solicitada, de tal forma a no prestarse a malinterpretación por parte de los oferentes.

title: Lote 4 – Item 48 - Especificaciones técnicas

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: En el apartado donde se solicita RItPHtPCO2t, se solicita aclarar lo que la convocante desea como requisito para esta determinación, de tal forma a que el oferente pueda preparar la oferta de sus productos de manera adecuada.

title: LOTE 1 ITEM 2 AL 76

answer: Ajustarse al PBC. Las EETT fueron suficientemente estudiadas y planificadas en el contexto de la necesidad y espacio físico del servicio. Desglosar el ítem con equipos independientes para un solo test, retrasaría la capacidad técnica del servicio.

description: SE PODRIA OFERTAR TAMBIEN EQUIPOS POR SEPARADO?

title: Lote 4 / Item 1:

answer: Ajustarse a la última versión del PBC.

description: Se podría admitir contadores de 3 partes sin gradillas de rack de muestras.

title: Lote 138 Cronometro

answer: Ajustarse a la última versión del PBC.

description: Favor aclarar si están precisando TIMER para LABORATORIO ó Cronometro, debido a que en las especificaciones técnicas se refieren más a Timer a que Cronometro.

title: Lote 23, Ítem 1 presentación de caja de 5 viales por 1mL.

answer: FAVOR AJUSTARSE A LA ULTIMA VERSION DEL PLIEGO DE BASES Y CONDICIONES.

description: Lote 23, Ítem 1: Se podría admitir una presentación de caja de 5 viales por 1mL.

title: lote 5

answer: Ajustarse a la última versión del PBC

description: LOTE 5, ÍTEM 1 al 18:   química clínica Centro de Salud y USF requiere  equipos automatizados para química clínica, con gradillas en rack. Se consulta si aceptarían:   equipos semiautomatizados para química clínica

title: Especificaciones Técnicas - Lote 4 ITEM 50

answer: misma respuesta que consulta 4

description: Mínimas características: electrodos libres de mantenimiento para determinaciones de electrolitos: Na, K, Cl y Ca. Para Gasometría pH, pCO2, pO2, Sat de O2, Hemoglobina total, Hematocrito. H+, CHCO3, CTCO2, BE, BEACT, BESS, BB, pHST, CHCO3ST, pAO2, NCa, QS/QT, p50, sO2 (C), AaDO2, a/AO2, eAVDO2, AGOER, Hct (C), PAO2t, RItPHtPCO2t, H+t, AaDO2t", solicitamos a la convocante aceptar un gasómetro con las siguientes características "Mínimas características: electrodos libres de mantenimiento para determinaciones de electrolitos: Na, K, Cl y Ca. Para Gasometría pH, pCO₂, pO₂, Hct, pH, Cl, GLU, LAC, Hb, HCO₃, BE, BE-B, BE-ECF, TCO₂, AG, AG(K), O₂ Sat, O₂ Ct, SBC, nCa, TCa, pO₂%, A, AaDO₂ , a/A", ya que según investigaciones realizadas, los gasómetros que se encuentran en el mercado no cumplen con las especificaciones requeridas  y consideramos que las especificaciones técnicas de los gasómetros comerciales son suficientes para satisfacer la necesidad del servicio.

streetAddress: DEPARTAMENTO DE LICITACIONES - UOC N° 01 - NIVEL CENTRAL - DGAF/DOC

submissionMethodDetails: Lugar entrega ofertas: DEPARTAMENTO DE LICITACIONES - UOC N° 01 - NIVEL CENTRAL - DGAF/DOC

mainProcurementCategoryDetails: Bienes - Equipo de Laboratorio Medida, Observacion y Comprobacion

legalName: Ministerio de Salud Pública y Bienestar Social (MSPBS)

legalName: Ministerio de Salud Pública y Bienestar Social

BIOTEC DEL PARAGUAY S.A.

streetAddress: TUYUTI N°1545 ENTRE AVDA.JOSE FELIX BOGADO Y GRAL.FRANCISCO MIRANDA

name: Equipos, Productos e instrumentales Médicos y de Laboratorio. Servicios asistenciales de salud

name: GONZALO IGNACIO SAMANIEGO ZUBIZARRETA- CARLOS AYALA- LORENA SAMANIEGO

DYSA HEALTHCARE S.A.

name: RICARDO HELLMERS FONSECA- Ricardo Hellmers Llano- Juan Gabriel Appleyard- Luis Pettengil- Ivan Maciel- Benjamin Diaz de Bedoya

INFOMED SOCIEDAD ANONIMA

streetAddress: CALLE CAÑADA DEL CARMEN CASI AMERICA 2565

name: Materiales e insumos eléctricos, metálicos y no metálicos, Plásticos, cauchos. Repuestos, herramientas, cámaras y cubiertas.

name: Maquinarias, Equipos y herramientas mayores - Equipos de transporte

name: Servicios de Limpiezas, Mantenimientos y reparaciones menores y mayores de Instalaciones, Maquinarias y Vehículos

name: Consultorías, Asesorías e Investigaciones. Estudios y Proyectos de inversión

name: Equipos Militares y de Seguridad. Servicio de Seguridad y Vigilancia

name: Utiles de oficina, Productos de papel y cartón e impresos

name: Adquisición y Locación de inmuebles. Alquiler de muebles

name: Bienes e insumos agropecuario y forestal

name: Construcción, Restauración, Reconstrucción o Remodelación y Reparación de Inmuebles

name: GABRIEL AVILA CORVALAN, FRANCESCA SCAVONE, CHRISTIAN WENTZENZEN, GLORIA RIVAS

DISTRIBUIDORA LA POLICLINICA S. A.

streetAddress: TTE. ROJAS SILVA N° 1044 C/ TTE. FARIÑA.

name: NORMAN HARRISON PALEARI - SUSY BEATRIZ BOGADO DE HARRISON - MARÍA CRISTINA DUARTE OZUNA - MARÍA GLORIA VELÁZQUEZ.

CHACO INTERNACIONAL SA

streetAddress: San Juan María Vianney N° 1684 c/ Bruno Guggiari e/ Juan Pablo II

name: CARMEN STELLA ESCOBAR BARRIOCANAL- SILVANA LOPEZ- EFREN LOPEZ

AO POTI S.A.

name: DIEGO DAUD, LUIS FELIPE SOUSA MUJICA- Cristhian Nunes

PROSALUD FARMA S.A.

name: Juan Domingo Rienzi Grassi, Camila Rienzi, Christian Rienzi- Magdalena Anahi Castro

Medical Supply Sociedad Anonima

name: Ramón Eduardo Doria Zubizarreta- Sebastian Rojas- Maria Lorena Gimenez- Jorge alberto Perez

CONSORCIO NEFROTEX

streetAddress: CALLE, NUESTRA SEÑORA DE LA ASUNCION CASI RODRIGUEZ DE FRANCIA

name: Frida E. Gerszkoviez de Singer / Gabriel Maria Avila Corvalan

name: Consorcio de Insumos Quirúrgicos del Paraguay

streetAddress: Avda Santa Teresa c/ Aviadores del Chaco Paseo Galería Torre 2 piso 26

legalName: Consorcio de Insumos Quirúrgicos del Paraguay

name: Ricardo Hellmers Fonseca/ Marcial Nuñez Gimenez

CONSORCIO DYRMEC

streetAddress: Av. Santa Teresa c/ Av. Aviadores del Chaco - Torre 2, Piso 26

name: Ricardo Hellmers Fonseca, Juan Gabriel Appleyard Pettengill, Javier Eugenio Moreno Iribas, Santiago José Moreno Iribas, Hugo Codas Thompson Benza, Oscar Antonio Codas Thompson Benza

CONSORCIO DYLAV

streetAddress: Avda Santa teresa c/ Avda Aviadores Chaco

name: Sistema de Información de las Contrataciones Públicas - DNCP